- About this Journal ·
- Abstracting and Indexing ·
- Advance Access ·
- Aims and Scope ·
- Article Processing Charges ·
- Articles in Press ·
- Author Guidelines ·
- Bibliographic Information ·
- Citations to this Journal ·
- Contact Information ·
- Editorial Board ·
- Editorial Workflow ·
- Free eTOC Alerts ·
- Publication Ethics ·
- Reviewers Acknowledgment ·
- Submit a Manuscript ·
- Subscription Information ·
- Table of Contents

Computational Intelligence and Neuroscience

Volume 2010 (2010), Article ID 397272, 7 pages

http://dx.doi.org/10.1155/2010/397272

## The Influence of Age and Skull Conductivity on Surface and Subdermal Bipolar EEG Leads

^{1}Department of Biomedical Engineering, Tampere University of Technology, Korkeakoulunkatu 3, P.O. Box 692, 33101 Tampere, Finland^{2}Institute of Biomedical Engineering, Karlsruhe Institute of Technology, 76131 Karlsruhe, Germany

Received 1 July 2009; Accepted 8 October 2009

Academic Editor: Fabrizio De Vico Fallani

Copyright © 2010 Katrina Wendel et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

#### Abstract

Bioelectric source measurements are influenced by the measurement location as well as the conductive properties of the tissues. Volume conductor effects such as the poorly conducting bones or the moderately conducting skin are known to affect the measurement precision and accuracy of the surface electroencephalography (EEG) measurements. This paper investigates the influence of age via skull conductivity upon surface and subdermal bipolar EEG measurement sensitivity conducted on two realistic head models from the Visible Human Project. Subdermal electrodes (a.k.a. subcutaneous electrodes) are implanted on the skull beneath the skin, fat, and muscles. We studied the effect of age upon these two electrode types according to the scalp-to-skull conductivity ratios of 5, 8, 15, and 30 : 1. The effects on the measurement sensitivity were studied by means of the half-sensitivity volume (HSV) and the region of interest sensitivity ratio (ROISR). The results indicate that the subdermal implantation notably enhances the precision and accuracy of EEG measurements by a factor of eight compared to the scalp surface measurements. In summary, the evidence indicates that both surface and subdermal EEG measurements benefit better recordings in terms of precision and accuracy on younger patients.

#### 1. Introduction

Clinical electroencephalography (EEG) and evoked potential (EP) recordings such as the visually evoked potentials (VEPs) demand high signal-to-noise ratios (SNRs), minimization of skin artifacts, and high accuracy, to name a few important criteria. Subdermal needle electrodes (a.k.a. subcutaneous needle electrodes) are commonly used in clinical electromyography (EMG), which are inserted into the muscles of interest. It is less commonly known that these subdermal needle electrodes also record continuous EEGs and EPs in intensive care units (ICU) [1–4]. The measurement setup is achieved by inserting the needle nearly tangentially to the skin so that it is stabilized and the recording tip touches the skull. Furthermore, these recordings offer higher SNRs with lower proclivity of standard measurement artifacts when compared with traditional surface measurements and are more suitable for long-term EEG monitoring in the ICU. Higher SNR requires less averaging, thus yielding faster and more accurate diagnostic measurements. We believe that clinical EEGs and EPs such as the VEP could adopt the subdermal measurement setup, thus placing the lead on the skull bypassing the artifact-prone skin.

Previously, we correlated skull conductivity with age (Figure 1) [6]. In that former study, we analyzed the reported skull conductivities of living skull fragments temporarily excised during epilepsy surgery with the age of the patient [5]. We reported a decreasing trend that stabilized in early adulthood. According to medical texts, physiologists explain that the calvarial bone completes the ossification process between the ages of 18 and 20 [7]; therefore, the skull conductivity should nearly approach steady state after adolescence. From the study of Hoekema et al. [5], we extrapolate that the scalp-to-skull conductivity ratio of 5 represents children and a small percentage of adolescents, the ratio of 8 represents adolescents and some adults, the ratio of 15 represents most adults, and lastly the ratio of 30 represents cadavers suffering from postcellular death. Ages that overlap scalp-to-skull conductivity ratios accommodate inter- and intrasubject variability [1, 8]. Taking standard skin conductivity values [9, 10] divided by the adult skull conductivity values yields a ratio of 8.5, and then scaled by the living to postmortem factor [6, 11, 12] yields approximately 20 to 26. These ratios fit accordingly with [13], which reported a ratio of 15 for post mortem skulls beyond cellular death.

In the present study we apply the concepts of the half-sensitivity volume (HSV) [14] and region of interest sensitivity ratio (ROISR) [15]. We use these metrics to analyze the effects of EEG electrode implantation on the measurement sensitivity distribution within the brain. Specifically, we aim to compare the sensitivity distributions of the bipolar subdermal EEG measurement with the well-documented surface electrode according to a patient's age [6, 14, 16–19].

#### 2. Methods

##### 2.1. Sensitivity Distribution

The sensitivity distributions of measurement leads in an inhomogeneous volume conductor can be illustrated with lead current fields as defined by [20–22]. The lead vectors define the relationship between the measured signal in the lead and the current sources in the volume conductor such that where is the voltage, for example, measured EEG voltage, in the volume conductor . The reciprocal current field is the lead field, is the impressed current density vector in the volume conductor, and is the conductivity (S/m) [17].

The sensitivity distribution in the volume conductor can be established by applying the reciprocity theorem of Helmholtz with Poisson’ equation (2) applied to describe quasistatic bioelectric source-field problems [23, 24]. A source distribution, , containing only reciprocal source currents at the measurement electrodes raises a gradient potential distribution, , that is, measurement sensitivity, according to the linear Poisson equation setting the Neumann boundary conditions equal to zero on the scalp where is the electrical conductivity tensor, is the electrical potential, is the current source density, is a vector normal to the surface, is the volume of the head, and is the surface of the head [25].

##### 2.2. The Half-Sensitivity Volume

In Malmivuo et al. [14], the concept of the half-sensitivity volume (HSV) was applied to define the volume in which the sensitivity of the measurement lead is concentrated. The HSV is the size of the volume within the source region of the volume conductor, where the magnitude of the sensitivity is at least half of its maximum value. The size of the HSV reflects how focused the region is from which the lead measures bioelectric activity, that is, smaller volumes have a higher measurement resolution and, conversely, larger volumes have a lower measurement resolution. The half-sensitivity volume is thus applied to evaluate the ability of the lead to concentrate the measurement sensitivity.

##### 2.3. The Region of Interest Sensitivity Ratio

Väisänen et al. [15] introduced the concept of the region of interest sensitivity ratio (ROISR), which provides a parameter to analyze the specificity of a measurement system. Equation (4) defines ROISR as a ratio between the average sensitivity of a predefined region-of-interest (ROI) volume (5) and the average sensitivity in the rest of the source volume, hereafter called a nonROI volume. The ratio is formulated such that where is the ROI source volume () and is the nonROI source volume ().

In the case of EEG, the nonROI volume consists of the entire brain source volume excluding the ROI volume. ROISR thus defines how well the measurement sensitivity is concentrated within the selected ROI, that is, how specific the measurement is to the signals generated within the ROI. We define the ROI volume as where is the brain source volume containing the gray and white matters, and is a sphere with a 20 mm radius from the cortical electrode located on the occipital cortex surface (10/20 location, , Figure 2). Consequently, our ROI contains both gray and white matters. We selected this location due to its relevance in visually evoked studies by Sörnmo and Laguna [26].

##### 2.4. Model and Computations

We calculate the sensitivity distributions in a realistically shaped male and female heads model based on the U.S. National Library of Medicine's Visible Human Project digital male and female anatomical dataset [27–29], VHP. Calculation of the sensitivity distributions is based on the principle of reciprocity and the numerical finite difference method (FDM) solution of EEG electrode sensitivity. In the FDM model, the segmented head data from a magnetic resonance image (MRI) data set is divided into cubic elements forming a resistive network [30]. The conductivities, of the elements correspond to the tissue conductivities and the dimensions of the elements correspond to the resolution of the dataset. The FDM is based on Poisson's equation that can be used to describe the bioelectric quasistatic source field problems [24]. A potential distribution within the model for a specific source configuration is solved with linear equations and iterative methods [31, 32].

EEG source localization and head model simulations significantly depend on the conductivities used in the models. In literature many studies apply a brain-to-skull conductivity ratio between 15 and 80 [33]; however, these two parameters vary widely in their conductivity values. The brain tissue conductivity value ranges from 0.12 S/m to 0.48 S/m [1, 8, 34–40], whereas the skull conductivity value ranges from 0.0042 S/m to 0.3 S/m [5, 8, 11, 13, 34–36, 41]. The scalp (skin) conductivity value varies less in literature from 0.33 S/m to 0.45 S/m [8, 9, 34, 35, 42]. Therefore, in the present study we apply the scalp-to-skull conductivity ratios of 5, 8, 15, and [1, 6, 13, 38–40, 43]. The tissues and their corresponding conductivity values that we used in this study are listed in Table 1 [10].

We calculate the sensitivity distributions of the brain for each bipolar electrode pair located on the scalp and the skull. The surface electrodes (a.k.a. scalp electrodes) and the subdermal electrodes measure 1 mm 1 mm 1 mm, which reflects the size of one pixel. These dimensions represent one type of subdermal recording electrodes that are insulated up to the tip. Our bipolar leads reflect a visually evoked measurement over the occipital cortex (10/20 location ) referenced against an apex electrode (10/20 location ). The sagittal views of the models (Figure 2) show the two bipolar EEG locations: surface electrode on the scalp and the subdermal electrode on the skull.

#### 3. Results

Figures 3 and 4 present the sensitivity distributions of both the scalp and subdermal leads solved with different conductivity ratios. Clearly, the conductivity ratio has a significant impact on the sensitivity distribution when we consider only one type of electrodes. However, the comparison of both types of electrodes diminishes the influence of the conductivity correlated with age, thus indicating the improved measurement resolution of the needle electrodes irrespective of the patient's age.

Optimally placed subdermal electrodes nearly outperform surface electrodes at every age. The smearing effect of the scalp disappears with the subdermal leads because the recording locations are closer to the target region, thus bypassing the skin (Figures 3 and 4). Tables 2 and 3 show that the subdermal lead's HSV decreases to nearly one-seventh, one-nineth, one-eighth, and one-fourth the size of the scalp lead's HSV. Similarly, we find a 35% to 37% improvement in the subdermal lead's ROISR over the surface lead's ROISR. Figures 3 and 4 illustrate that the subdermal measurement distributions visibly concentrate the measurement sensitivity more efficiently to the target region on the cortex of the younger patient's skull (i.e., lower conductivity values). Moreover, the smearing effect of the skull is reduced with the subdermal leads, and nearly the entire scalp and skull smearing is eliminated when the patient is the youngest (i.e., the skull conducting value is at its peak). Conversely, the older the patient, namely, the higher the scalp-to-skull conductivity ratio, the more the skull conductivity smears the lead field formation. Precisely, the subdermal leads measure neuroelectric activity on or near the gyral cortical surface rather than sulcal or deep sources.

#### 4. Discussion

The present study compares two variables influencing EEG source localization studies: age and electrode location. This study shows that the ratio between the scalp and subdermal measurements regarding the HSV is smallest with the lowest skull conductivity ratio. The correlation between the HSV ratios indicates that measurements will be more localized, that is, increased sensitivity, with higher specificity (ROISR). The subdermal measurement distributions visibly concentrate the measurement sensitivity more efficiently to the target region on the cortex as the skull conductivity increases. The smearing effect of the scalp is reduced with the subdermal leads, and nearly the entire scalp and skull smearing is eliminated when the skull has its highest conducting value [8]. Precisely, the subdermal leads measure neuroelectric activity on or near the gyral cortical surface rather than sulcal or deep sources.

Tissue conductivities such as skin, cortical bone, and brain conductivities change with age [41, 44–50]. Their results indicate a decrease in conductivity between 40.7% and 75.4% from newborn to maturity stages. Furthermore, their results show that the aging process slows during childhood before adolescence after the rapid growth phases of the body have been completed. This is due to the reduction of water content in tissue as a function of age [47, 50]. We believe that the conductivity of the skin changes again in late adulthood, that is, the elderly, particularly decreasing in conductivity. Therefore, the skin conductivity from adolescence onwards should minimally affect this study.

We selected our scalp-to-skull conductivity ratios to span from early childhood through adulthood. Our skull values reflect an 83.9% decrease in the human skull conductivity value compared with the 75.4% change in rats, whereas we kept a fixed conductivity for the brain and skin. When we compare similar sets of measurements such as the surface measurements we obtain an improvement in measurement, resolution between 10.4% and 51.1% for the HSV and an improvement in the measurement accuracy between 25.5% and 38.2% for the ROISR. When we include the subdermal needle measurements, we yield improvements between 75% and 89% in the measurement resolution over the surface electrodes. If we had factored in growth from youth through adolescence to adulthood, then the change in HSV and ROISR would have increased the variation in the results. The additional variables would have plausibly enhanced the measurement precision in childern due to the high water content of their tissues [47].

#### 5. Conclusion

The implantation of EEG electrode on the skull notably increases the measurement sensitivity and accuracy over traditional surface electrodes. These measurements known as subdermal or subcutaneous measurements bypass the artifact prone skin to obtain relatively artifact-free, high-resolution EEG recordings. The measurement sensitivity of the needle electrodes concentrates the subdermal EEG measurements. Consequently, the subdermal electrode reduces the need for the extremely invasive electrocorticogram (ECoG) and minimizes the influence of age on EEG source localization. We found that the scalp-to-skull conductivity ratio influenced the subdermal EEG measurement less than the surface EEG measurements. From our correlative study we can definitively claim that children, specifically preadolescent children, would benefit the most from the increased resolution of the subdermal electrodes.

The age plays an important role in the surface electrode measurements, but the change in measurement location to subdermal electrodes irrefutably improves the measurement sensitivity distributions. Succinctly, the subdermal electrodes outperform surface electrodes because they minimize the effect due to the intersubject variability in the scalp-to-skull conductivity ratio associated with the change in age.

#### References

- J. Latikka, T. Kuurne, and H. Eskola, “Conductivity of living intracranial tissues,”
*Physics in Medicine and Biology*, vol. 46, no. 6, pp. 1611–1616, 2001. View at Publisher · View at Google Scholar - J. R. Ives, “New chronic EEG electrode for critical/intensive care unit monitoring,”
*Journal of Clinical Neurophysiology*, vol. 22, no. 2, pp. 119–123, 2005. View at Publisher · View at Google Scholar - S. Fossi, A. Amantini, A. Grippo, et al., “Continuous EEG-SEP monitoring of severely brain injured patients in NICU: methods and feasibility,”
*Neurophysiologie Clinique*, vol. 36, no. 4, pp. 195–205, 2006. View at Publisher · View at Google Scholar · View at PubMed - G. Bryan Young, J. R. Ives, M. G. Chapman, and S. M. Mirsattari, “A comparison of subdermal wire electrodes with collodion-applied disk electrodes in long-term EEG recordings in ICU,”
*Clinical Neurophysiology*, vol. 117, no. 6, pp. 1376–1379, 2006. View at Publisher · View at Google Scholar · View at PubMed - R. Hoekema, G. H. Wieneke, F. S. S. Leijten, et al., “Measurement of the conductivity of skull, temporarily removed during epilepsy surgery,”
*Brain Topography*, vol. 16, no. 1, pp. 29–38, 2003. View at Publisher · View at Google Scholar - K. Wendel and J. Malmivuo, “Correlation between live and post mortem skull conductivity measurements,” in
*Proceedings of the 28th Annual International Conference of the IEEE Engineering in Medicine and Biology Society*, pp. 4285–4288, August 2006. View at Publisher · View at Google Scholar · View at PubMed - K. L. Moore and A. F. Dalley,
*Clinically Oriented Anatomy*, Lippincott Williams & Wilkins, Philadelphia, Pa, USA, 5th edition, 2005. - C. H. Wolters, A. Anwander, X. Tricoche, D. Weinstein, M. A. Koch, and R. S. MacLeod, “Influence of tissue conductivity anisotropy on EEG/MEG field and return current computation in a realistic head model: a simulation and visualization study using high-resolution finite element modeling,”
*NeuroImage*, vol. 30, no. 3, pp. 813–826, 2006. View at Publisher · View at Google Scholar · View at PubMed - T. C. Ferree, K. J. Eriksen, and D. M. Tucker, “Regional head tissue conductivity estimation for improved EEG analysis,”
*IEEE Transactions on Biomedical Engineering*, vol. 47, no. 12, pp. 1584–1592, 2000. View at Publisher · View at Google Scholar · View at PubMed - C. Ramon, P. H. Schimpf, and J. Haueisen, “Influence of head models on EEG simulations and inverse source localizations,”
*BioMedical Engineering Online*, vol. 5, article 10, 2006. View at Publisher · View at Google Scholar · View at PubMed - J. D. Kosterich, K. R. Foster, and S. R. Pollack, “Dielectric permittivity and electrical conductivity of fluid saturated bone,”
*IEEE Transactions on Biomedical Engineering*, vol. 30, no. 2, pp. 81–86, 1983. View at Google Scholar - J. D. Kosterich, K. R. Foster, and S. R. Pollack, “Dielectric properties of fluid-saturated bone—the effect of variation in conductivity of immersion fluid,”
*IEEE Transactions on Biomedical Engineering*, vol. 31, no. 4, pp. 369–374, 1984. View at Google Scholar - T. F. Oostendorp, J. Delbeke, and D. F. Stegeman, “The conductivity of the human skull: results of in vivo and in vitro measurements,”
*IEEE Transactions on Biomedical Engineering*, vol. 47, no. 11, pp. 1487–1492, 2000. View at Google Scholar - J. Malmivuo, V. Suihko, and H. Eskola, “Sensitivity distributions of EEG and MEG measurements,”
*IEEE Transactions on Biomedical Engineering*, vol. 44, no. 3, pp. 196–208, 1997. View at Publisher · View at Google Scholar · View at PubMed - J. Väisänen, O. Väisänen, J. Malmivuo, and J. Hyttinen, “New method for analysing sensitivity distributions of electroencephalography measurements,”
*Medical and Biological Engineering and Computing*, vol. 46, no. 2, pp. 101–108, 2008. View at Publisher · View at Google Scholar · View at PubMed - M. F. Suesserman, F. A. Spelman, and J. T. Rubinstein, “In vitro measurement and characterization of current density profiles produced by nonrecesed, simple recessed, and radially varying recessed stimulating electrodes,”
*IEEE Transactions on Biomedical Engineering*, vol. 38, no. 5, pp. 401–408, 1991. View at Google Scholar - J. Malmivuo and R. Plonsey,
*Bioelectromagnetism: Principles and Applications of Bioelectric and Biomagnetic Fields*, Oxford University Press, New York, NY, USA, 1995. - S. Grimnes and O. G. Martinsen,
*Bioimpedance and Bioelectricity Basics*, Academic Press, San Diego, Calif, USA, 2000. - J. O. Ollikainen, M. Vauhkonen, P. A. Karjalainen, and J. P. Kaipio, “Effects of electrode properties on EEG measurements and a related inverse problem,”
*Medical Engineering and Physics*, vol. 22, no. 8, pp. 535–545, 2000. View at Publisher · View at Google Scholar - R. McFee and F. D. Johnston, “Electrocardiographic leads: I. Introduction,”
*Circulation*, vol. 8, no. 10, pp. 554–568, 1953. View at Google Scholar - R. McFee and F. D. Johnston, “Electrocardiographic leads: II. Analysis,”
*Circulation*, vol. 9, no. 2, pp. 255–266, 1954. View at Google Scholar - R. McFee and F. D. Johnston, “Electrocardiographic leads: III. Synthesis,”
*Circulation*, vol. 9, no. 6, pp. 868–880, 1954. View at Google Scholar - H. L. F. Helmholtz, “Über einige Gesetze der Vertheilung elektrischer Ströme in körperlichen Leitern mit Anwendung auf die thierisch-elektrischen Versuche,”
*Annalen der Physik und Chemie*, vol. 89, pp. 211–233, 354–377, 1853. View at Google Scholar - J. Sarvas, “Basic mathematical and electromagnetic concepts of the biomagnetic inverse problem,”
*Physics in Medicine and Biology*, vol. 32, no. 1, pp. 11–22, 1987. View at Publisher · View at Google Scholar - C. Johnson, M. Mohr, U. Rude, A. Samsonov, and K. Zyp, “Multilevel methods for inverse bioelectric field problems,” in
*Multiscale and Multiresolution Methods in Computational Science and Engineering*, Springer, Berlin, Germany, 2003. View at Google Scholar - L. Sörnmo and P. Laguna,
*Bioelectrical Signal Processing in Cardiac and Neurological Applications*, Academic Press, San Diego, Calif, USA, 2005. - M. J. Ackerman, “The visible human project,”
*The Journal of Biocommunication*, vol. 18, no. 2, p. 14, 1991. View at Google Scholar - National Institutes of Health, “Visible human project,” U.S. National Library of Medicine, http://www.nlm.nih.gov/research/visible/visible_human.html.
- F. B. Sachse, C. D. Werner, K. Meyer-Waarden, and O. Dössel, “Applications of the visible man dataset in electrocardiology: calculation and visualization of body surface potential maps of a complete heart cycle,” in
*Proceedings of the 2nd Users Conference of the National Library of Medicine's Visible Human Project*, pp. 47–48, 1998. - X. Franceries, B. Doyon, N. Chauveau, B. Rigaud, P. Celsis, and J.-P. Morucci, “Solution of Poisson's equation in a volume conductor using resistor mesh models: application to event related potential imaging,”
*Journal of Applied Physics*, vol. 93, no. 6, pp. 3578–3588, 2003. View at Publisher · View at Google Scholar - C. R. Johnson, “Computational and numerical methods for bioelectric field problems,”
*Critical Reviews in Biomedical Engineering*, vol. 25, no. 1, pp. 1–81, 1997. View at Google Scholar - P. Kauppinen, J. Hyttinen, P. Laarne, and J. Malmivuo, “A software implementation for detailed volume conductor modelling in electrophysiology using finite difference method,”
*Computer Methods and Programs in Biomedicine*, vol. 58, no. 2, pp. 191–203, 1999. View at Publisher · View at Google Scholar - S. Rush and D. A. Driscoll, “EEG electrode sensitivity—an application of reciprocity.,”
*IEEE Transactions on Biomedical Engineering*, vol. 16, no. 1, pp. 15–22, 1969. View at Google Scholar - C. Gabriel, S. Gabriel, and E. Corthout, “The dielectric properties of biological tissues: I. Literature survey,”
*Physics in Medicine and Biology*, vol. 41, no. 11, pp. 2231–2249, 1996. View at Publisher · View at Google Scholar - S. Gabriel, R. W. Lau, and C. Gabriel, “The dielectric properties of biological tissues: II. Measurements in the frequency range 10 Hz to 20 GHz,”
*Physics in Medicine and Biology*, vol. 41, no. 11, pp. 2251–2269, 1996. View at Publisher · View at Google Scholar - S. Gabriel, R. W. Lau, and C. Gabriel, “The dielectric properties of biological tissues: III. Parametric models for the dielectric spectrum of tissues,”
*Physics in Medicine and Biology*, vol. 41, no. 11, pp. 2271–2293, 1996. View at Publisher · View at Google Scholar - J. A. Latikka, J. A. Hyttinen, T. A. Kuurne, H. J. Eskola, and J. A. Malmivuo, “The conductivity of brain tissues: comparison of results in vivo and in vitro measurements,” in
*Proceedings of the 23rd Annual International Conference of the IEEE Engineering in Medicine and Biology Society*, vol. 1, pp. 910–912, Istanbul, Turkey, October 2001. - S. Goncalves, J. de Munck, J. P. A. Verbunt, F. Bijma, R. M. Heethaar, and F. H. Lopes da Silva, “In vivo measurement of the brain and skull resistivites using an EIT-based method and realistic models for the head,”
*IEEE Transactions on Biomedical Engineering*, vol. 50, pp. 754–767, 2003. View at Google Scholar - S. Gonçalves, J. de Munck, J. P.A. Verbunt, R. M. Heethaar, and F. H. Lopes da Silva, “In vivo measurement of the brain and skull resistivities using an EIT-based method and the combined analysis of SEF/SEP data,”
*IEEE Transactions on Biomedical Engineering*, vol. 50, no. 9, pp. 1124–1128, 2003. View at Publisher · View at Google Scholar · View at PubMed - Y. Lai, W. van Drongelen, L. Ding, et al., “Estimation of in vivo human brain-to-skull conductivity ratio from simultaneous extra- and intra-cranial electrical potential recordings,”
*Clinical Neurophysiology*, vol. 116, no. 2, pp. 456–465, 2005. View at Publisher · View at Google Scholar · View at PubMed - C. Gabriel, A. Peyman, and E. H. Grant, “Electrical conductivity of tissue at frequencies below 1 MHz,”
*Physics in Medicine and Biology*, vol. 54, no. 16, pp. 4863–4878, 2009. View at Publisher · View at Google Scholar · View at PubMed - V. Raicu, N. Kitagawa, and A. Irimajiri, “A quantitative approach to the dielectric properties of the skin,”
*Physics in Medicine and Biology*, vol. 45, no. 2, pp. L1–L4, 2000. View at Publisher · View at Google Scholar - T. C. Ferree and D. M. Tucker, “Development of high-resolution EEG devices,”
*International Journal of Bioelectromagnetism*, vol. 1, pp. 4–10, 1999. View at Google Scholar - A. Peyman, A. A. Rezazadeh, and C. Gabriel, “Chnages in the dielectric properties of rat tissue as a function of age at microwave frequencies,”
*Physics in Medicine and Biology*, vol. 46, pp. 1617–1629, 2001. View at Google Scholar - A. Peyman, A. A. Rezazadeh, and C. Gabriel, “Erratum: changes in the dielectric properties of rat tissue as a function of age at microwave frequencies,”
*Physics in Medicine and Biology*, vol. 47, no. 12, pp. 2187–2188, 2002. View at Google Scholar - C. Gabriel, “Dielectric properties of biological tissue: variation with age,”
*Bioelectromagnetics*, vol. 26, supplement 7, pp. S12–S18, 2005. View at Publisher · View at Google Scholar · View at PubMed - J. Keshvari, R. Keshvari, and S. Lang, “The effect of increase in dielectric values on specific absorption rate (SAR) in eye and head tissues following 900, 1800 and 2450 MHz radio frequency (RF) exposure,”
*Physics in Medicine and Biology*, vol. 51, no. 6, pp. 1463–1477, 2006. View at Publisher · View at Google Scholar · View at PubMed - A. Peyman, S. J. Holden, S. Watts, R. Perrott, and C. Gabriel, “Dielectric properties of porcine cerebrospinal tissues at microwave frequencies: in vivo, in vitro and systematic variation with age,”
*Physics in Medicine and Biology*, vol. 52, no. 8, pp. 2229–2245, 2007. View at Publisher · View at Google Scholar · View at PubMed - J. Wiart, A. Hadjem, M. F. Wong, and I. Bloch, “Analysis of RF exposure in the head tissues of children and adults,”
*Physics in Medicine and Biology*, vol. 53, no. 13, pp. 3681–3695, 2008. View at Publisher · View at Google Scholar · View at PubMed · View at Scopus - A. Peyman, C. Gabriel, E. H. Grant, G. Vermeeren, and L. Martens, “Variation of the dielectric properties of tissues with age: the effect on the values of SAR in children when exposed to walkie-talkie devices,”
*Physics in Medicine and Biology*, vol. 54, no. 2, pp. 227–241, 2009. View at Publisher · View at Google Scholar · View at PubMed