- About this Journal ·
- Abstracting and Indexing ·
- Aims and Scope ·
- Article Processing Charges ·
- Articles in Press ·
- Author Guidelines ·
- Bibliographic Information ·
- Citations to this Journal ·
- Contact Information ·
- Editorial Board ·
- Editorial Workflow ·
- Free eTOC Alerts ·
- Publication Ethics ·
- Submit a Manuscript ·
- Subscription Information ·
- Table of Contents
Volume 2012 (2012), Article ID 597124, 7 pages
Changing Trends of Imaging in Angle Closure Evaluation
1Hamilton Glaucoma Center, Shiley Eye Center and Department of Ophthalmology, University of California, La Jolla, CA 92093, USA
2Department of Ophthalmology and Visual Science, Yale University School of Medicine, New Haven, CT 06510, USA
Received 6 January 2012; Accepted 23 February 2012
Academic Editors: I. G. Pallikaris and Á. Szél
Copyright © 2012 Syril Dorairaj et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Primary angle closure glaucoma (PACG) is a significant cause of visual disability worldwide. It predominantly affects the Eastern and South Asian population of the world. Early detection of anatomically narrow angles is important, and the subsequent prevention of visual loss from PACG depends on an accurate assessment of the anterior chamber angle (ACA). Gonioscopy has given way to modern day imaging technologies such as ultrasound biomicroscopy (UBM) and more recently, anterior segment optical coherence tomography (AS-OCT). Ultrasound biomicroscopy provides objective, high-resolution images of anterior segment anatomy, including the cornea, iris, anterior chamber, anterior chamber angle, and ciliary body. Optical coherence tomography (OCT) is a noncontact optical signal acquisition and processing device that provides magnified, high-resolution cross-sectional images of ocular tissues. Recent technological advances towards three-dimensional visualization broadened the scope of AS-OCT in ophthalmologic evaluation. Optical coherence tomography systems use low-coherence, near-infrared light to provide detailed images of anterior segment structures at resolutions exceeding that of UBM. This paper summarizes the clinical application of UBM and OCT for assessment of anterior segment in glaucoma.
Primary angle closure glaucoma (PACG) is a leading cause of blindness worldwide . It is potentially preventable if diagnosed early in the course of the disease, before irreversible damage has occurred to the optic nerve or trabecular meshwork (TM). Primary angle closure glaucoma comprises about 10% of glaucoma patients in the USA, and its prevalence varies amongst ethnic and racial groups . Narrow angles are found in about 2% of Caucasians, with 0.1% having acute angle closure glaucoma [3–5]. The ethnic group in which PACG is most common is Eskimos . Angle-closure is less common in blacks but more likely to develop chronic ACG when they do develop the disease [7–9]. Asians are prone to chronic angle-closure and often do not reach clinical attention until severe ocular damage has already occurred. In Asians, specifically, the incidence of angle-closure glaucoma outnumbers open-angle glaucoma . In all racial and ethnic populations, ACG is found 3-4 times more often in females than males . Finally, ACG is most prevalent in hyperopic and elderly patients, peaking between ages 55 and 70, since the anterior chamber depth and volume decrease with age due to nuclear sclerosis and in patients with a family history of angle-closure glaucoma .
Angle-closure glaucoma is three times as likely as open angle glaucoma (OAG) to cause blindness. Angle-closure glaucoma studies conducted in Asian countries estimate 4.3 million blind from ACG and 3.3 million from OAG and that bilateral blindness affects fewer than 10% of those with OAG but 25–30% of ACG sufferers [5, 12]. Needless to say, ACG is a worldwide problem to which attention should be paid with special regards to prevention and diagnosis.
Anterior segment pathologies, resulting from anatomic, structural, or mechanical abnormalities which cause apposition of the iris to the trabecular meshwork can contribute to angle-closure and thus the risk of progressive trabecular damage, elevated intraocular pressure (IOP), peripheral anterior synechiae (PAS), and acute angle-closure. Primary angle-closure glaucoma can, like other types of glaucoma, ultimately result in blindness if not diagnosed and treated timely.
The pathophysiology of angle closure can be divided into primary and secondary causes. Primary angle closure is known as pupillary block, and it accounts for more than 90% of cases. In pupillary block, the flow of aqueous from the posterior chamber, where it is produced by nonpigmented ciliary epithelium, to the anterior chamber is limited because of resistance to aqueous flow through the pupil in the region of iridolenticular contact. This limitation of flow creates an increased pressure gradient between the anterior and posterior chambers, which in turn forces the iris anteriorly and causes anterior iris bowing, narrowing of the angle, and acute/chronic or acute on chronic iridotrabecular apposition or angle closure glaucoma. In relative pupillary block, all the anatomical structures are usually normal. Secondary causes of angle closure, by contrast, can result from structural or anatomic abnormalities in the anterior or posterior segments, such as plateau iris configuration, lens subluxation, or malignant glaucoma (ciliary block or aqueous misdirection).
Clinically, the gold standard for diagnosis of narrow angles is dark-room gonioscopy, in which the iridocorneal angle and aqueous outflow through the trabecular meshwork can be assessed; however, this technique is subjective, and there are currently no standards related to gonioscopy to determine which angles require treatment.
Qualitative studies of the anterior segment structures can provide only limited information for diagnosis and subsequently, fail to provide a single, worldwide standard of care for narrow angles. Ideally, quantitative studies of the anterior chamber and the relationships of structures therein could provide objective measurements, which will standardize the anterior chamber (AC) parameters requiring interventions and treatments. The best way to achieve these quantitative measurements is through anterior chamber imaging devices. Ultrasound Biomicroscopy (UBM) has been used for this very purpose for more than 15 years [13–20]. UBM provides objective, high-resolution images of anterior segment anatomy, with tissue resolution of approximately 50 microns and penetration depth of 5 mm, providing a useful diagnostic tool for narrow angles and other anterior chamber pathologies. UBM is capable of imaging the cornea, iris, anterior chamber, anterior chamber angle, posterior chamber, and ciliary body (please see Figure 1). All UBM images shown in this manuscript were taken with a UBM model P40 from Paradigm Medical Industries, Salk Lake City, Utah. The various forms of angle closure glaucoma, such as pupillary block and plateau iris configuration can be differentiated using a UBM (Please see Figures 2(a), 2(b) and 3). Please see Figures 4, 5, 6, 7 for UBM image examples of peripheral anterior synechia, plateau iris configuration after argon laser peripheral iridoplasty (ALPI), phacomorphic angle closure, and malignant glaucoma.
Imaging technologies have proven extremely useful for explaining the nature of various pathologies and in determining a rationale for treatment in patients who may be confused concerning open-angle and angle-closure glaucoma and the laser treatment modality which best suits their condition.
In a UBM image, the scleral spur can be seen as the innermost point of the line separating the ciliary body and the sclera at its point of contact with the anterior chamber (please see Figure 1). The trabecular meshwork is located directly anterior to this structure and posterior to Schwalbe’s line, which is the most peripheral portion of Descemet’s membrane (please see Figure 1). Thus, the essential structures for the diagnosis of angle-closure glaucoma are clearly visible in a UBM image .
Despite the advantages of UBM, there are some disadvantages as well [22–24]. First, this procedure can be uncomfortable for the patient, requiring placement of an eyecup between the lids while lying in the supine position. The eyecup is filled with saline, and a 35–50 MHz transducer is placed in the saline as the patient looks down, right, up, and left to record the superior, nasal, inferior, and temporal angles. The exam occurs once in room lighting and once with the lights turned off, known as the light and dark provocative test, to look for appositional closure and angle occludability. Many patients have difficulty tolerating this procedure. Imaging with the UBM carries some risks for the patient, such as scratching the cornea and requires a skilled and trained operator. In addition, positional changes, such as the supine position used during UBM testing, may alter relationships between chamber structures, particularly in eyes with narrow angles. In the supine position, at least in some patients, the lens might move posteriorly, thus missing some occludable angles during UBM testing.
The risks of performing UBM, the requirement for a skilled operator, and the incomplete or nonideal information provided by the supine position leave room for improvement in anterior segment imaging. The next generation of imaging would provide images taken in the seated position, preferably without any contact, and, with less invasiveness, less risk and less discomfort for the patient (please see Table 1 for details of UBM as well as ASOCT devices available in the market).
With new anterior segment optical coherence tomography (ASOCT) imaging techniques, detailed spatial relationships of the anterior segment structures can be visualized and objective anterior chamber angle (ACA) measurements can be performed in a noncontact manner (please see Figure 8). In addition, the use of infrared laser and real time eye position monitor during examination permits the precise capture of angle morphology in the dark. With higher scan speed, slit lamp optical coherence tomography (SLOCT) has the potential to provide valuable quantitative and spatial information regarding dynamic changes of the angle configuration, which cannot be provided by standard gonioscopy and UBM. SLOCT provides information that is comparable to an intermediate standard between ASOCT and UBM in terms of ACA measurement . Please see Figures 8, 9, 10, 11 for examples of AS-OCT images of relative pupillary block, plateau iris, phacomorphic glaucoma, and malignant glaucoma. All ASOCT images shown in this paper were taken with a slit-lamp-adapted anterior segment OCT from Heidelberg Engineering, Heidelberg, Germany.
Enhancing the clinical applicability of ASOCT, Prata et al., described a novel dynamic technique to differentiate appositional from synechial angle closure and to understand the underlying mechanisms of angle closure using indentation ASOCT . Identification of the causes of angle closure is of utmost importance, as each case may have a different course therefore requiring a different treatment approach. Differentiation of appositional and synechial angle closure in eyes with iridotrabecular contact during indention ASOCT adds to the clinical utility of ASOCT in the evaluation of patients with angle closure.
Asrani et al. recently reported the successful visualization of details of anterior chamber drainage angle (Schlemm’s canal, trabecular meshwork and configuration details of the iris with respect to the angle) using a swept source Fourier-domain OCT system .
2. The Reliability of ACA Measurement
The scleral spur is a protrusion of sclera anchoring the trabecular meshwork anteriorly and the longitudinal muscle of the ciliary body posteriorly. It represents an anatomical landmark for the trabecular meshwork which is located approximately 250 to 500 μm anterior to the scleral spur along the angle wall. Because of the different tissue reflectivity between the sclera and cornea, the scleral spur can be visualized in UBM or OCT as a distinct anatomical landmark for measurement of the ACA [21, 28, 29]. Because most of the important parameters for ACA quantitative measurements are based on the identification of the point of scleral spur, reliable documentation of the angle dimensions is therefore dependent on precise and repeatable localization of the scleral spur. Sakata et al. found that, on the same Visante OCT images, the intraobserver agreement in detecting the scleral spur (132 quadrants) was moderate to substantial with . They also reported that, in the assessment of the exact scleral spur location, the distance between the scleral spur localized in the same image across 2 sessions was within 10 μm in 83% of the 78 quadrants assessed and within 20 μm in 90%. The location of the scleral spur on ASOCT images was less detectable in quadrants with a closed angle on gonioscopy (odds ratio = 0.54, ) and also in images obtained in the superior and inferior compared with the nasal and temporal quadrants (64%, 67%, 75%, and 80%, respectively; ). Using the same images for measurement, the intraobserver coefficient of variation (CVw) of angle opening distance (AOD) ranged between 4.9% and 7.8% by ASOCT and was up to 16.97% by UBM () which indirectly indicates that identification of scleral spurs might be more repeatable by ASOCT than UBM [25–31].
The differences in anterior chamber angle measurements in different lighting conditions have been investigated with UBM and ASOCT . Leung et al. also described the dynamic ACA changes induced by dark-light changes by Visante OCT through real-time video recording . They found that the AOD and trabecular iris space area (TISA) decreased linearly with increasing pupil size in most cases (85.5% in AOD and 90.9% in TISA). It was estimated that for each mm change in pupil size, there was an average of 94 μm change in the AOD and 0.035 mm2 change in the TISA. Although significant differences of angle measurements were found between light and dark conditions, good repeatability and reproducibility were achieved as long as the lighting condition had been standardized by ASOCT [33, 34]. The intersession CVw for angle measurements by Visante OCT was less than that of UBM (Visante OCT: 6~11%, UBM: 16~18%) [31, 33].
Significant correlations were found among ACA measurements by ASOCT, gonioscopy, and UBM [29, 35]. In general, the correlation in detecting a closed ACA quadrant using ASOCT and gonioscopy was fair with a of 0.4. But ASOCT tended to detect more closed angles than gonioscopy, particularly in the superior and inferior quadrants . There was no significant difference in angle measurements between ASOCT and UBM in either nasal or temporal quadrants, but a significant higher AOD measurement was observed by ASOCT in the superior and inferior angles compared with UBM [29, 36]. Of note, although slit-lamp OCT (SLOCT) and Visante OCT generally had no significant difference in angle measurements, the two available ASOCT models had poor correlations in ACA measurement despite comparable pupil diameters obtained, with the spans of 95% limits of agreement (LOA) of the nasal/temporal angle measurements between them being 437 μm/531 μm, 0.174 mm2/0.186 mm2, and 25.3°/28.0° for AOD, TISA, and trabecular-iris angle (TIA), respectively . The poor correlation is likely related to differences in the choice of refractive indexes in the calculation of anterior segment dimensions, algorithms for image dewarping, the exact scan locations, and the state of accommodation.
Furthermore, UBM offers a better view of the ciliary body, which is rarely visible during ASOCT, since attenuated light from the overlying sclera obscures the view of the ciliary body. However, recent studies have confirmed the ability of ASOCT to evaluate and confirm a clinical suspicion of plateau iris configuration and syndrome .
Finally, different factors can influence the appearance of the angle, including background illumination, blinking, patient posture, contact with the eye, and image processing software.
In most cases the software to analyze OCT images is based on manual labeling of the scleral spur, cornea, and iris which is not only a tedious process but sometimes not possible to accomplish as in about 20–30% of the cases; these landmarks cannot be identified. Sakata et al. found that the sclera spur could not be detected in approximately 30% of the ACA quadrants, this problem being worse in the superior and inferior quadrants . In order to overcome this limitation Jing et al. suggested a new algorithm capable of automatically detecting Schwalbe’s line in HD-OCT scans .
Clinically, ASOCT has been applied to the observation of ACA change after glaucoma surgeries, such as laser peripheral iridotomy (LPI), argon laser peripheral iridoplasty (ALPI), trabeculectomy combined with cataract extraction, and intraocular lens implantation.
In a retrospective study involving 71 Caucasian eyes, Ang and Wells compared AS-OCT parameters before and after laser iridotomy . The authors found that this procedure resulted in significant angle widening as shown on increased TIA, AOD, TISA, and iris profile flattening. In an Asian population of 46 patients, Lee et al.  found that when assessed by measurement variability criteria, the percentage of eyes that showed no significant change in ACA parameters ranged from 23.9% to 45.7% after LPI. In 15 patients with primary angle closure Lei et al.  found that after LPI the peripheral anterior chamber depth and anterior chamber volume increased as well as the central anterior chamber depth increased as seen on AS-OCT.
In most of the priory reported studies where LPI was done, the authors found a proportion of patients in whom the angle did not widen but did not specify the possible etiologies for it, and this is consistent with the limitation of current OCT technologies to evaluate the structures behind the iris like an anterior insertion of the ciliary body causing plateau iris configuration/syndrome. Despite this limitation in a small case series some authors have suggested the presence of signs that are suspicious for plateau iris syndrome in AS-OCT . Both OCT and UBM showed excellent performance in identifying eyes with plateau iris. The UBM confirmed the plateau iris diagnosis by showing the iris root indentation caused by the ciliary body. The OCT can detect indirect signs of plateau iris syndrome after iridotomy .
3. Summary and Conclusions
It is envisioned that the new anterior segment imaging devices would have as significant impact as the new posterior segment imaging devices. The new imaging devices do not aim to replace conventional slit-lamp biomicroscopy. They would act to supplement and augment clinical practice and become invaluable tools for ophthalmic research. The major advantages of the newer devices are the noncontact nature of examination, high scan speed, good repeatability and reproducibility for quantitative and qualitative measurements, and cross-sectional visualization of anterior segment structures. Since ASOCT can visualize the entire anterior chamber, all the essential parameters for detection of angle closure/narrow angle can be examined in a single scan. The ASOCT would become an essential tool for screening PAC, making screening programs for PACG more feasible and less doctor dependent. The application of ASOCT has led to a better understanding of anterior segment diseases. It can now be readily quantified making longitudinal followups and assessments possible. The outcome of treatment can be monitored without discomfort or risks of inflammation. The new devices may improve our understanding of current limitations of surgery. The potential clinical applications of these methods are only starting to be explored and the range of information they may yield has yet to be determined. Therefore, the use of the newer anterior segment imaging devices could well be the start of a new era for ophthalmic diagnosis.
The authors acknowledge Ocular Imaging Laboratory at the Shelley and Steven Einhorn Clinical Research Center at New York Eye and Ear Infirmary for providing images. The authors also thank Vishal Jhanji, MD from Department of Ophthalmology and Visual sciences Chinese University of Hong Kong, for helping with critical review of the paper. T. M. Grippo is supported by a departmental challenge grant from research to prevent blindness, Inc.
- H. Quigley and A. T. Broman, “The number of people with glaucoma worldwide in 2010 and 2020,” British Journal of Ophthalmology, vol. 90, no. 3, pp. 262–267, 2006.
- American Academy of Ophthalmology, Primary Angle Closure, Preferred Practice Pattern, American Academy of Ophthalmology, San Francisco, Calif, USA, 2005.
- L. Dandona, R. Dandona, P. Mandal et al., “Angle-closure glaucoma in an urban population in Southern India: the andhra pradesh eye disease study,” Ophthalmology, vol. 107, no. 9, pp. 1710–1716, 2000.
- A. Jacob, R. Thomas, S. P. Koshi, A. Braganza, and J. Muliyil, “Prevalence of primary glaucoma in an urban South Indian population,” Indian Journal of Ophthalmology, vol. 46, no. 2, pp. 81–86, 1998.
- P. J. Foster and G. J. Johnson, “Glaucoma in china: how big is the problem?” British Journal of Ophthalmology, vol. 85, no. 11, pp. 1277–1282, 2001.
- G. H. M. B. van Rens, S. M. Arkell, W. Charlton, and W. Doesburg, “Primary angle-closure glaucoma among Alaskan Eskimos,” Documenta Ophthalmologica, vol. 70, no. 2-3, pp. 265–276, 1988.
- V. Clemmesen and M. H. Luntz, “Lens thickness and angle closure glaucoma: a comparative oculometric study in South African Negroes and Danes,” Acta Ophthalmologica, vol. 54, no. 2, pp. 193–197, 1976.
- J. T. Wilensky, N. Gandhi, and T. Pan, “Racial influences in open-angle glaucoma,” Annals of Ophthalmology, vol. 10, no. 10, pp. 1398–1402, 1978.
- J. F. Salmon, “Presenting features of primary angle-closure glaucoma in patients of mixed ethnic background,” South African Medical Journal, vol. 83, no. 8, pp. 594–597, 1993.
- Y. Liang, D. S. Friedman, Q. Zhou et al., “Prevalence and characteristics of primary angle-closure diseases in a rural adult Chinese population: the handan eye study,” Investigative Ophthalmology & Visual Science, vol. 52, no. 12, pp. 8672–8679, 2011.
- N. G. Congdon and D. S. Friedman, “Angle-closure glaucoma: impact, etiology, diagnosis, and treatment,” Current Opinion in Ophthalmology, vol. 14, no. 2, pp. 70–73, 2003.
- P. J. Foster, F. T. S. Oen, D. Machin et al., “The prevalence of glaucoma in chinese residents of singapore: a cross-sectional population survey of the tanjong pagar district,” Archives of Ophthalmology, vol. 118, no. 8, pp. 1105–1111, 2000.
- C. J. Pavlin, K. Harasiewicz, M. D. Sherar, and F. S. Foster, “Clinical use of ultrasound biomicroscopy,” Ophthalmology, vol. 98, no. 3, pp. 287–295, 1991.
- C. J. Pavlin, M. D. Sherar, and F. S. Foster, “Subsurface ultrasound microscopic imaging of the intact eye,” Ophthalmology, vol. 97, no. 2, pp. 244–250, 1990.
- R. Ritch and J. M. Liebmann, “Role of ultrasound biomicroscopy in the differentiation of block glaucomas,” Current Opinion in Ophthalmology, vol. 9, no. 2, pp. 39–45, 1998.
- S. Kaushik, R. Jain, S. Pandav, and A. Gupta, “Evaluation of the anterior chamber angle in Asian Indian eyes by ultrasound biomicroscopy and gonioscopy,” Indian Journal of Ophthalmology, vol. 54, no. 3, pp. 159–163, 2006.
- Y. Barkana, S. K. Dorairaj, Y. Gerber, J. M. Liebmann, and R. Ritch, “Agreement between gonioscopy and ultrasound biomicroscopy in detecting iridotrabecular apposition,” Archives of Ophthalmology, vol. 125, no. 10, pp. 1331–1335, 2007.
- A. Narayanaswamy, L. Vijaya, B. Shantha, M. Baskaran, A. V. Sathidevi, and S. Baluswamy, “Anterior chamber angle assessment using gonioscopy and ultrasound biomicroscopy,” Japanese Journal of Ophthalmology, vol. 48, no. 1, pp. 44–49, 2004.
- P. Mora, C. Sangermani, S. Ghirardini, A. Carta, N. Ungaro, and S. A. Gandolfi, “Ultrasound biomicroscopy and iris pigment dispersion: a case-control study,” British Journal of Ophthalmology, vol. 94, no. 4, pp. 428–432, 2010.
- Z. Sbeity, S. K. Dorairaj, S. Reddy, C. Tello, J. M. Liebmann, and R. Ritch, “Ultrasound biomicroscopy of zonular anatomy in clinically unilateral exfoliation syndrome,” Acta Ophthalmologica, vol. 86, no. 5, pp. 565–568, 2008.
- C. J. Pavlin, K. Harasiewicz, and F. S. Foster, “Ultrasound biomicroscopy of anterior segment structures in normal and glaucomatous eyes,” American Journal of Ophthalmology, vol. 113, no. 4, pp. 381–389, 1992.
- R. Ursea and R. H. Silverman, “Anterior segment imaging for assessment of glaucoma,” Expert Review Of Ophthalmology, vol. 5, no. 1, pp. 59–74, 2010.
- T. Dada, R. Gadia, A. Sharma et al., “Ultrasound Biomicroscopy in Glaucoma,” Survey of Ophthalmology, vol. 56, no. 5, pp. 433–450, 2011.
- D. Quek, M. Nongpiur, S. Perera, and T. Aung, “Angle imaging: advances and challenges,” Indian Journal of Ophthalmology, vol. 59, no. 1, pp. S69–S75, 2011.
- C. K. Leung, H. Li, R. N. Weinreb et al., “Anterior chamber angle measurement with anterior segment optical coherence tomography: a comparison between slit lamp OCT and VisanteOCT,” Investigative Ophthalmology & Visual Science, vol. 49, no. 8, pp. 3469–3474, 2008.
- T. S. Prata, S. Dorairaj, C. G. V. De Moraes, C. Tello, J. M. Liebmann, and R. Ritch, “Indentation slitlamp-adapted optical coherence tomography technique for anterior chamber angle assessment,” Archives of Ophthalmology, vol. 128, no. 5, pp. 646–647, 2010.
- S. Asrani, M. Sarunic, C. Santiago, and J. Izatt, “Detailed visualization of the anterior segment using fourier-domain optical coherence tomography,” Archives of Ophthalmology, vol. 126, no. 6, pp. 765–771, 2008.
- C. J. Pavlin and F. S. Foster, “Ultrasound biomicroscopy in glaucoma,” Acta Ophthalmologica. Supplementum, no. 204, pp. 7–9, 1992.
- C. Wirbelauer, A. Karandish, H. Häberle, and T. P. Duy, “Noncontact goniometry with optical coherence tomography,” Archives of Ophthalmology, vol. 123, no. 2, pp. 179–185, 2005.
- L. M. Sakata, R. Lavanya, D. S. Friedman et al., “Assessment of the scleral spur in anterior segment optical coherence tomography images,” Archives of Ophthalmology, vol. 126, no. 2, pp. 181–185, 2008.
- H. Ishikawa, J. M. Liebmann, and R. Ritch, “Quantitative assessment of the anterior segment using ultrasound biomicroscopy,” Current Opinion in Ophthalmology, vol. 11, no. 2, pp. 133–139, 2000.
- S. F. Urbak, “Ultrasound biomicroscopy. I. Precision of measurements,” Acta Ophthalmologica Scandinavica, vol. 76, no. 4, pp. 447–455, 1998.
- C. K. S. Leung, D. W. F. Yick, Y. Y. Y. Kwong et al., “Analysis of bleb morphology after trabeculectomy with Visante anterior segment optical coherence tomography,” British Journal of Ophthalmology, vol. 91, no. 3, pp. 340–344, 2007.
- M. Müller, G. Dahmen, E. Pörksen et al., “Anterior chamber angle measurement with optical coherence tomography: intraobserver andinterobserver variability,” Journal of Cataract and Refractive Surgery, vol. 32, no. 11, pp. 1803–1808, 2006.
- T. Dada, R. Sihota, R. Gadia, A. Aggarwal, S. Mandal, and V. Gupta, “Comparison of anterior segment optical coherence tomography and ultrasound biomicroscopy for assessment of the anterior segment,” Journal of Cataract and Refractive Surgery, vol. 33, no. 5, pp. 837–840, 2007.
- S. Dorairaj, J. M. Liebmann, and R. Ritch, “Quantitative evaluation of anterior segment parameters in the era of imaging,” Transactions of the American Ophthalmological Society, vol. 105, pp. 99–108, 2007, discussion 108–110.
- S. Ulaganathan, S. B. Ganeshrao, M. Baskaran, S. Srinivasan, B. Shantha, and L. Vijaya, “Plateau Iris configuration and dark-light changes in anterior segment optical coherence tomography,” Ophthalmic Surg Lasers Imaging, vol. 41, pp. e1–e4, 2010.
- T. Jing, P. Marziliano, and H. T. Wong, “Automatic detection of Schwalbe's line in the anterior chamber angle of the eye using HD-OCT images.,” Conference Proceedings: IEEE Engineering in Medicine and Biology Society, vol. 2010, pp. 3013–3016, 2010.
- G. S. Ang and A. P. Wells, “Factors influencing laser peripheral iridotomy outcomes in white eyes: an anterior segment optical coherence tomography study,” Journal of Glaucoma, vol. 20, no. 9, pp. 577–583, 2011.
- K. S. Lee, K. R. Sung, S. Y. Kang, J. W. Cho, D. Y. Kim, and M. S. Kook, “Residual anterior chamber angle closure in narrow-angle eyes following laser peripheral iridotomy: anterior segment optical coherence tomography quantitative study,” Japanese Journal of Ophthalmology, vol. 55, no. 3, pp. 213–219, 2011.
- K. Lei, N. Wang, L. Wang, and B. Wang, “Morphological changes of the anterior segment after laser peripheral iridotomy in primary angle closure,” Eye, vol. 23, no. 2, pp. 345–350, 2009.
- C. Parc, J. Laloum, and O. Bergès, “Comparison of optical coherence tomography and ultrasound biomicroscopy for detection of plateau iris,” Journal Francais d'Ophtalmologie, vol. 33, no. 4, pp. 266–e1, 2010.