Robotic AstronomyView this Special Issue
Research Article | Open Access
R. Hudec, M. Spurny, M. Krizek, P. Pata, R. Slosiar, M. Rerabek, M. Klima, "Detection of GRBs and OTs by All-Sky Optical and SID Monitors", Advances in Astronomy, vol. 2010, Article ID 428943, 8 pages, 2010. https://doi.org/10.1155/2010/428943
Detection of GRBs and OTs by All-Sky Optical and SID Monitors
We report on two alternative simple methods to detect counterparts of cosmic gamma-ray bursts (GRBs) and optical transients (OTs). We report on the development and tests of an alternative optical all-sky monitor recently tested at the Karlovy Vary Observatory. The monitor is based on a Peleng 8 mm fish-eye lens (1 : 3,5–1 : 16) and CANON EOS 350D digital CCD camera. This type of monitor represents a low-cost device suitable for easy replication and still able to detect brighter optical transients simultaneously to GRB triggers. Such OTs have been observed for some of the GRBs such as GRB990123, GRB060117, or recently GRB080319 indicating that some fraction of GRBs can generate optical transient emission accessible by simple small aperture instrumentation as described here. These efforts are accompanied by development of dedicated programmes to access and to evaluate all-sky images; these efforts will be also briefly described. The All-Sky Monitor is a space variant optical system and its point spread function (PSF) has not uniform shape in the field of view. The processing and measuring of image data is complicated, and sophisticated deconvolution algorithms are used for image restoration. The second method is the GRB detection based on their ionospheric response.
1. Bright Prompt Optical Emission of GRBs
The fast response is required to investigate the optical prompt emission of GRBs. However, even the fastest optical follow-up telescopes cannot access the times close or identical to times of GRBs, and the time domain before GRB remains completely hidden. These time domains can be accessed only by optical wide-field monitors (as the position of the GRB is unpredictable). The all-sky monitors offer the best sky coverage. An alternative approach is to monitor the FOVs of recent GRB satellites with optical WF cameras. Some of the all-sky monitors operated on daily basis are based on the use of photographic emulsion (allowing long exposures and fine spatial resolution of few microns). However, the photographic emulsion is not very sensitive to short optical flares and has some additional disadvantages. The alternative digital all-sky monitoring is provided by the CONCAM system. However, the limiting magnitudes are not very deep, so one can hardly expect that such system will be able to detect optical transients (OTs) of GRBs. The brightest OTs related to GRBs observed so far were observed at magnitudes 6–10 (e.g., GRB990123, GRB060117, GRB080319) [1, 2]. For GRB060117, the optical transient was followed due to technical reasons from 2 minutes after the GRB trigger and already declining, so one can deduce that the peak brightness probably exceeded magnitude 8 . We hence need monitors able to detect short OTs with duration of about 1 minute and fainter than magnitude 8. Some optical all-sky monitors are in operation already such as the EN photographic network managed by the Ondrejov Observatory (Figures 5, 6, and 7) and CCD-based sky monitor was tested at the Sonneberg Observatory (Figure 4).
2. The Optical All-Sky Monitor
We report on an alternative low cost all-sky monitor. The used instrumentation is simple and low-cost. The camera has two parts, namely, the Peleng 8 mm fish-eye lens (1 : 3,5–1 : 16, Figure 2) that provides a 24 mm circular 180 field of view (Figure 3), and a CANON EOS 350D digital CCD camera (Figure 1). The total cost of the hardware is around 1500 USD, that is, one order less than the CONCAM system. One can hence expect the system to be easily and cheaply replicated to numerous sites. One can consider the alternative type of the digital camera, such as Canon EOS 5D with a larger CCD chip (but somewhat larger pixel size) hence covering the whole FOV of the fish-eye lens. We plan to improve the performance of the system by designing, in collaboration with the Czech Technical University in Prague, especially dedicated mount and miniature dome for the camera. Use of another lens can provide deeper magnitudes but smaller FOV (Figure 8).
3. Simulation and Evaluation of Images from Optical All-Sky Systems
As shown in Figure 2, the all-sky fish-eye lenses consist of 10 glass elements. This type of optics influences the focal images. Any reliable scientific use and evaluation of images from these systems must take this into account. There are two related problems in WFC and UWFC optical systems: () optical aberrations, and () space variant systems. The motivation for the simulation is as follows: () modelling of optical system used in all-sky monitors and their transfer characteristics aberrations and distortion, () UPSF image restoration and deconvolution of acquired image data removal, and () enhancement of measurement precision.
Real UWFC image data and their evaluation are related with the following problems: () objects on ultra-wide-field images are very small (a few pixels per object dimension). () Optical aberrations and distortions in UWFC systems, see Figure 11, () influence of optical aberrations increases at the edges of FOV and, () these aberrations distort the PSF of optical system and rapidly cut the accuracy of measurements. () Objects on the frontier of field of view are not used for analysis in astrometry because a lot of aberrations. Above we show several preliminary results obtained in this direction (Figures 9, 10, and 12).
4. The VLF Detections of GRBs
The method of indirect detections of GRBs at Very Low Frequency (VLF) has been described already before , however only very few positive records are known. Previously reported VLF detections of GRBs are as follows. (i) GRB830801: a first observation of an ionospheric disturbance from a gamma-ray burst reported . The burst, occurred at 22 : 14 : 18 UT on August 1, 1983, and was one of the strongest ever observed at that time. The total fluency was 0.002 erg/sq cm, most of which occurred in the first 4 seconds of the burst. Simultaneously, a change was observed in the amplitude of a VLF radio signal from a transmitter in Rugby, England indicative of an ionospheric disturbance. Weaker disturbances were also recorded at the same receiving site on signals from VLF stations in Annapolis, Maryland and Lualualei, Hawaii. The times of the burst and the disturbances are coincident within the 10-second resolution of the VLF recording system. No similar disturbances were observed within 60 hours around the time of the burst. (ii) SGR1806: detection of a Sudden Ionospheric Disturbance , (iii) GRB030329 observed as a Sudden Ionospheric Disturbance (SID) . Although there were numerous efforts to detect GRBs by VLF and despite the fact that the necessary instrumentation is inexpensive, this field still remains little exploited .
The instrumentation used in obtaining the results discussed below consists of a loop radio antenna and a 2-channel VLF radio receiver (Figure 13). The motivation is an alternative and inexpensive approach to detection and investigation of GRBs.
5. Physics of Ionospheric Detection
The solar particle stream, solar wind, shapes and controls the Earths magnetic envelope—the magnetosphere—and increases heat in the aurora zones. But not all ionospheric variability is caused by solar or geomagnetic disturbances. The ionosphere is not a constant “mirror in the sky.” The E layer (100–200 km aboveground) and the F1 layer (170–200 km) usually behave in regular, solar-controlled way, but the F2 layer (250–350 km) does not. It is the F2 layer, which has the greatest density of free electrons, and is potentially the most effective reflector of radio waves (see ).
The ionospheric D layer plays in the GRB detections an important role, as the detection of X-ray and gamma-ray triggers is based on the measurement (monitoring) of reflected radio waves from this layer. The ionospheric D layer is not transparent for radio VLF waves (frequencies 3 kHz to 30 kHz) and behaves like a mirror. If the transmitter is at large distance (800 to 2000 km) then the radio waves are guided like in a waveguide consisting of the D layer and the earth surface. Any change in the quality of this waveguide results then in the signal change in the SID monitor. The change can be positive but in some cases such as the sudden phase anomaly also negative.
6. VLF Detections of GRBs
We present below examples of our VLF/SID detection of GRBs in three cases: GRB 060124A (Figure 14), GRB080319D and GRB080320A (Figure 15), as well as an indication for detection of GRB induced propagating ionospheric waves (Figure 15).
An alternative low-cost (1.5 kUSD) optical digital all-sky monitoring system has been assembled and it is tested recently. The preliminary results indicate the limiting magnitude even for nonguided system and for one image with exposure of 30 seconds amounts to mag 8. Deeper magnitudes are expected for guided system and longer or cumulated exposures, then the expected limiting magnitude can reach the magnitude 9-10 range. This makes the system suitable for wide-field monitoring in the sky for brighter optical transients. Using nonfish-eye lenses, the limiting magnitude can reach 15 over a smaller (10 10 degrees) FOV. The system can be very easily duplicated to numerous sites. Future improvements are planned such as design of a miniature camera mount and dome to allow guided images and development of dedicated control and evaluation software. The algorithms for evaluation of images from all-sky monitors were developed and tested.
The independent and indirect detection of GRBs by their ionospheric response (SID) observed at VLF is feasible. We presented and discussed examples of such VLF/SID detection in three cases: GRB 060124A, GRB080319D, and GRB080320A. In addition, these measurements are in agreement with the scenario of propagating ionospheric waves triggered by the relevant GRBs. Although few such detections have been already reported in the past, the capability of such alternative and indirect investigations of GRBs, as well as the possible contribution to analyses of GRBs, still remains to be investigated in more details.
The GRB analyses described here are linked to the GRB analyses within the ESA PECS INTEGRAL Project 98023. Some parts are related to the grant of the Grant Agency of the Czech Republic 205/08/1207 and 102/09/0997. The analyses of GRBs in optical range are newly supported by MSMT KONTAKT Project ME09027.
- C. Akerlof, R. Balsano, S. Barthelmy et al., “Observation of contemporaneous optical radiation from a -ray burst,” Nature, vol. 398, no. 6726, pp. 400–402, 1999.
- J. L. Racusin, S. V. Karpov, M. Sokolowski et al., “Broadband observations of the naked-eye -ray burst GRB 080319B,” Nature, vol. 455, no. 7210, pp. 183–188, 2008.
- M. Jelínek, M. Prouza, P. Kubánek et al., “The bright optical flash from GRB 060117,” Astronomy and Astrophysics, vol. 454, no. 3, pp. L119–L122, 2006.
- G. J. Fishman and U. S. Inan, “Observation of an ionospheric disturbance caused by a -ray burst,” Nature, vol. 331, no. 6155, pp. 418–420, 1988.
- P. Campbell, M. Hill, R. Howe et al., “SGR1806: detection of a sudden ionospheric disturbance,” GCN GRB Observation Report 2932, GRB Coordinates Network, AAVSO, Cambridge, Mass, USA, 2003.
- P. W. Schnoor, D. L. Welch, G. J. Fishman, and A. Price, “GRB030329 observed as a sudden ionospheric disturbance (SID),” GCN GRB Observation Report 2176, GRB Coordinates Network, AAVSO, Cambridge, Mass, USA, 2003.
- H. Rishbeth, I. C. F. Müller-Wodarg, L. Zou et al., “Annual and semiannual variations in the ionospheric F2-layer. II. Physical discussion,” Annales Geophysicae, vol. 18, no. 8, pp. 945–956, 2000.
Copyright © 2010 R. Hudec et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.