Research Article | Open Access
Ken Ohata, Kuangi Fu, Eiji Sakai, Kouichi Nonaka, Tomoaki Tashima, Yohei Minato, Akiko Ohno, Takafumi Ito, Yosuke Tsuji, Hideyuki Chiba, Makoto Yamawaki, Hideyuki Hemmi, Teruo Nakaya, Junichi Fukushima, Nobuyuki Matsuhashi, "Esophageal Endoscopic Submucosal Dissection Assisted by an Overtube with a Traction Forceps: An Animal Study", Gastroenterology Research and Practice, vol. 2016, Article ID 3186168, 7 pages, 2016. https://doi.org/10.1155/2016/3186168
Esophageal Endoscopic Submucosal Dissection Assisted by an Overtube with a Traction Forceps: An Animal Study
Esophageal endoscopic submucosal dissection (ESD) is technically difficult. To make it safer, we developed a novel method using overtube with a traction forceps (OTF) for countertraction during submucosal dissection. We conducted an ex vivo animal study and compared the clinical outcomes between OTF-ESD and conventional method (C-ESD). A total of 32 esophageal ESD procedures were performed by four beginner and expert endoscopists. After circumferential mucosal incision for the target lesion, structured as the isolated pig esophagus 3 cm long, either C-ESD or OTF-ESD was randomly selected for submucosal dissection. All the ESD procedures were completed as en bloc resections, while perforation only occurred in a beginner’s C-ESD procedure. The dissection time for OTF-ESD was significantly shorter than that for C-ESD for both the beginner and expert endoscopists ( min versus min, , and min versus min, , resp.). The frequency and volume of the submucosal injections were significantly smaller for OTF-ESD than for C-ESD ( times versus times, , and mL versus mL, , resp.). Histologically, muscular injury was more common among the C-ESD procedures (80% versus 13%, ). Our results indicated that the OTF-ESD technique is useful for the safe and easy completion of esophageal ESD.
The incidence of esophageal cancer has increased rapidly in both Western and Asian countries , although the predominant pathologic types are quite different [2, 3]. Recent technical innovations in endoscopy (e.g., narrow-band imaging) have enabled improvements in the detectability of superficial esophageal neoplasms [4, 5]. Because of the higher complication rates and lower quality of life in patients undergoing an esophagectomy , endoscopic resection is now often used for the treatment of such lesions. To reduce the morbidity and mortality rates associated with advanced esophageal cancer, the appropriate management of early cancer is necessary.
Recently, endoscopic submucosal dissection (ESD) has been applied as an alternative to EMR for the removal of superficial esophageal neoplasms, as it provides a much higher en bloc resection rate and reduced rates of local residual tumor and/or recurrence [7, 8]. Although gastric ESD has now spread worldwide, esophageal ESD remains challenging because of the thin and narrow luminal wall in the esophagus. Moreover, the relatively lower incidence of superficial esophageal neoplasms has made it difficult for surgeons, especially Western endoscopists, to become proficient at performing esophageal ESD given the lack of training opportunities and supervision by experts . Submucosal dissection after the completion of the circumferential mucosal incision is considered to be the most challenging step in ESD, as it is often difficult to obtain a good view of the dissection plane in the narrow esophageal lumen because of respiratory and cardiac pulsations. Underestimating the depth of the submucosal layer and inaccurate identification of the cutting line may cause perforation and/or residual tumor. Therefore, the development of safer and more reliable devices as well as modifications of presently used techniques is required.
To facilitate submucosal dissection, adequate tissue traction and better visualization of the lesion are important. Although such conditions can sometimes be achieved with a postural change, it is difficult to maintain a good countertraction constantly. Previously, we reported a case of esophageal ESD in which a safer and faster submucosal dissection was achieved using a novel overtube with traction forceps (OTF-ESD) . Therefore, we conducted an animal model study comparing the safety and clinical outcomes of OTF-ESD and the conventional method (C-ESD). The objective of the present study was to evaluate the feasibility and potential effectiveness of OTF-ESD for possible application in humans.
2. Materials and Methods
Using an animal model, we conducted a prospective, randomized controlled trial comparing two different esophageal ESD techniques: the conventional method (C-ESD) and a newly developed method using a novel overtube with traction forceps (OTF-ESD) for submucosal dissection. The present study was planned according to the guidelines of our institutional review board and the Animal Use Committee.
2.1. Animal Models
Isolated pig organs (esophagus, stomach, and duodenum), which were commercially available at a Tokyo meat market, were irrigated with water and Pronase MS (400 U/mL; Kaken Pharma, Tokyo, Japan) before ESD. First, we dug a trench (20 mm in width and 25 cm in length) in a Styrofoam box in which we then placed the pig esophagus; the box was then fixed to a plastic stand (Johnson & Johnson, Tokyo, Japan). We inserted an overtube (Flexible Overtube MD-48518; Sumitomo Bakelite, Tokyo, Japan) into the oral edge of the pig esophagus and tied it firmly with a string. The gastric outlet was also closed tightly with a string to enable air insufflation during ESD. A counter electrode board was placed on top of the bottom wall of the trench. The porcine esophagus was dipped in saline to enable the conduction of electrical current and was then placed inside the trench. Finally, the oral side of the overtube was fixed with gum tape (Figure 1).
2.2. ESD Operators
A total of eight endoscopists (T. N., M. Y., H. H., A. O., T. I., H. C., Y. T., and K. O.) were enrolled in this study. The endoscopists were classified into two groups: four beginners who had limited experience performing esophageal ESD procedures and four experts with experience performing more than 30 esophageal ESD procedures. The profile of each operator is shown in Table 1. Each endoscopist resected four lesions: two using C-ESD and two using OTF-ESD.
|Beginners were defined as operators with limited experience performing esophageal ESDs. Experts were defined as operators who had performed more than 30 esophageal ESDs and more than 150 gastric ESDs. |
2.3. Devices and Settings
In this study, an upper GI endoscope (GIF-Q260J; Olympus Optical Co., Tokyo, Japan) with a transparent attachment cap (D201-10704, 4 mm; Olympus Optical Co.) was used for all the procedures. A dual knife (KD-650L; Olympus Optical Co.) was used for the circumferential mucosal resection, and the submucosal dissection was mainly performed using an IT knife nano (KD-612L; Olympus Optical Co.). A VIO generator (VIO300D; Erbe Elektromedizin, Tubingen, Germany) was used for all the ESD procedures as a high-frequency generator unit. The settings were as follows: Endocut I Effect 3 (duration 2, interval 2), 60 W for all ESD procedures, and Swift Coagulation, 45 W for submucosal dissection. Submucosal injection was performed using Glyceol mixed with indigo carmine using a 23-gauge, 4 mm long needle (Olympus, Optical Co).
2.4. Design and Concept of OTF Method
To create an accessory work channel, we mounted a long straw tube with red vinyl tape on an overtube (Figure 2(a)). Next, we modified a pair of grasping forceps (FB-230K; Olympus, Optical Co.) in a way such that they become straight when closed and curved when opened, enabling the edge of the target lesion to be grasped during the submucosal dissection (Figure 2(b)). Importantly, the modified overtube can be rotated around the endoscope within the esophagus, thereby enabling the target lesion to be regrasped from 360 degrees, if needed. Using the OTF method, adequate tissue traction and better visualization were expected to be maintained constantly.
2.5. ESD Procedure
The target lesion was created by endoscopic marking using an electrosurgical knife; each target lesion covered half the circumference of the esophageal lumen and measured 3 cm in length. After first ESD procedure, the next target lesion was created at 5 cm proximal to the previous resected margin. A total of two target lesions were removed from each porcine ex vivo animal model. After the circumferential mucosal resection, each of the two lesions was randomized into either a C-ESD group or an OTF-ESD group for submucosal dissection.
The C-ESD procedure was conducted as previously described . OTF-ESD was performed according to the following steps. First, a solution was injected into the submucosal layer, similar to the step used in the C-ESD procedure (Figure 3(a)). Then, an assistant operator inserted the modified grasping forceps into the accessory channel on the overtube and grasped the proximal edge of the target lesion; the assistant then straightened the forceps and lifted the submucosal layer. Adjusting the rotation angle of the overtube, the mucosal flap was retracted opposite to the surgical bed, enabling the submucosal layer to be clearly visualized without additional injections (Figure 3(b)). While the assistant continued to lift the mucosa with the forceps, the submucosal dissection was performed (Figure 3(c)). During the procedure, the assistant regulated and adjusted the direction of the lifted layer as appropriate, and the target lesion was finally removed (Figure 3(d)).
2.6. Outcomes and Data Analysis
The main outcome of the present study was the dissection time, which was defined as the time between the first submucosal injection of solution until the complete removal of the target lesion. In addition, the completion and en bloc resection rates, specimen size, perforation rate, and volume and frequency of injection needed for submucosal dissection were also evaluated as secondary outcomes. These measurements were compared between the OTF-ESD and C-ESD groups. The muscular layer of the dissected areas was fixed in 10% formalin and embedded in paraffin. After staining with hematoxylin and eosin, the depth of the muscular layer injury was histologically evaluated by a pathologist who was blinded to the ESD technique. When injury to the outer muscular layer (deeper half of the muscularis propria) was observed, the ESD procedure was evaluated as being positive for muscular layer injury.
2.7. Statistical Analysis
Continuous data are shown as the mean ± SD. Categorical variables were compared using the Fisher exact test. Continuous variables were compared using the Student -test. Unless otherwise specified, values of <0.05 were considered to denote statistical significance. All the statistical analyses were performed using PASW Statistics 18 for Windows (SPSS Japan, Tokyo, Japan).
A total of 32 ESD procedures (16 C-ESD and 16 OTF-ESD) were performed using 16 ex vivo porcine models. The outcomes were summarized in Table 2.
|C-ESD, conventional endoscopic submucosal dissection; OTF-ESD, overtube with forceps endoscopic submucosal dissection.|
Continuous data are shown as the mean SD.
P values were compared between the C-ESD and OTF-ESD groups.
3.1. Performance of ESD Procedure
All the ESD procedures were completed as en bloc resections. Although perforation occurred in one beginner’s C-ESD procedure, no perforations occurred during the OTF-ESD procedures. No significant difference in the measured area of the resected lesions was seen between the C-ESD and OTF-ESD groups ( cm2 versus cm2, ).
3.2. Dissection Time Needed for Submucosal Dissection
The dissection time for OTF-ESD was significantly shorter than that for C-ESD ( min versus min, ). Interestingly, the same tendency was observed in both the expert and the beginner group ( min versus min, , and min versus min, , for the expert and beginner groups, resp.) (Figure 4).
3.3. Volume and Frequency of Submucosal Injections
The frequency of submucosal injection was significantly lower for the OTF-ESD group than for the C-ESD group ( times versus times, ), and the volume of the submucosal injections was also significantly smaller for the OTF-ESD group ( mL versus mL, ). Significant differences were also seen in both the beginner and expert groups (Table 2).
3.4. Histological Evaluation
A total of 12 thermal muscular layer injuries were confirmed histologically: 10 occurred during C-ESD, and 2 occurred during OTF-ESD (Table 3). Thermal muscular layer injury to the ulcer floor was significantly less common in the OTF-ESD group (80% versus 13%, ).
|Abbreviations: C-ESD, conventional endoscopic submucosal dissection; OTF-ESD, overtube with forceps endoscopic submucosal dissection.|
NOTE: When injury of the outer muscular layer was observed, the ESD procedure was evaluated as positive for muscular layer injury.
P = 0.009, calculated using the Fisher exact test.
Using an ex vivo animal model, we demonstrated the effectiveness of OTF-ESD. Our results indicated that adequate tissue traction using a newly developed overtube enabled both beginner and expert endoscopists to complete esophageal ESD safely and easily.
ESD has been an emerging technique for the treatment of superficial esophageal neoplasms, while the EMR technique is mainly used in Western countries. Although EMR can be effective for removing small localized superficial esophageal neoplasms, the en bloc resection of lesions larger than 20 mm can be difficult; such difficulties can lead to incorrect pathological examinations and a relatively high frequency of local recurrence . The main advantage of ESD is the higher rate of complete resection, regardless of tumor size. Until now, the efficacy and safety of esophageal ESD have been evaluated mainly in Asian countries . Recently, Probst et al. reported that ESD is effective for the treatment of early esophageal adenocarcinoma, as well as squamous cell carcinoma . Meanwhile, Neuhaus et al. reported a relatively low complete en bloc resection rate for early esophageal adenocarcinoma (38.5%) , indicating that surgeons in Western countries are still acquiring proficiency in ESD. To overcome the technical difficulties associated with esophageal ESD and to establish this technique as a standard treatment worldwide, technical innovations are required.
Several reports have attempted to make the ESD procedure easier and safer using various traction devices [15–23]. Tsao et al. reported the usefulness of a modified fish-line traction for the ESD of large superficial esophageal cancer . Chen et al. conducted an animal study and demonstrated the efficacy and safety of percutaneous transgastric traction-assisted esophageal ESD, which was designed to offer esophageal mucosal traction directed to the stomach . Our method is different from those used by other devices from the following viewpoints. First, the traction is totally independent of the endoscope’s movement. Therefore, our method is especially useful for the tumors located at the left side wall of the esophagus, where the fluid and blood tend to be pooled. Second, the target tissue can be repeatedly grasped easily and effectively, if needed. Third, various traction directions, such as a left/right direction and an oral/anal direction, can be obtained by rotating the overtube and pulling/pushing the grasping forceps, respectively. The length of the flexible overtube is approximately 20 cm. Although operators should pull the overtube slightly during ESD procedure, the lesions located at cervical/upper esophagus could be also treated using OTF method. These advantages contribute to maintaining an adequate traction and stable visibility of the submucosal layer to facilitate a safer and faster submucosal dissection in various situations.
Recently, Hirota et al. conducted an in vivo animal study and evaluated the efficacy of a full-traction esophageal ESD technique using a similar overtube . They reported that the procedure time did not differ between the conventional technique and full-traction ESD, while the frequency of submucosal injection was significantly reduced by the full-traction technique. In contrast to their results, we found that the dissection time was significantly reduced by the OTF method. In addition, we also demonstrated that muscular layer injury to the ulcer floor was significantly less common in the OTF-ESD group. These discrepancies can probably be explained by differences in lesion size, animal model, cutting device, and setting. At our institution, submucosal dissection was performed using the “cut mode” once an appropriate dissection line had been confirmed; this factor may be associated with our improved dissection time. Importantly, our results indicated that OTF-ESD can enable a faster and safer ESD for beginners as well as experts. The performance of ESD is thought to be highly dependent on the expertise of the operator. Tsou et al. reported that at least 30 esophageal ESD procedures are needed for a novice endoscopist to gain proficiency in this technique . Technical difficulties such as uncontrollable hemorrhage frequently occur during submucosal dissection and can result in unwanted perforation. As the OTF method helps to maintain a good view of the dissection plane and may reduce such situations, it can offer opportunities for novice ESD operators to achieve higher self-completion rates.
The present study had several limitations. First, this study was a single-centered, preliminary study with a relatively small sample size. Second, an operator should consider the risk of overtube-associated complications, such as perforation, whereas the prevalence is reportedly quite low . Third, the study used an ex vivo animal model in which hemostasis was not required during the procedure, in addition to the lack of esophageal movement induced by peristalsis, breathing, and heartbeat. However, as the OTF method provides better fixation and visualization of the submucosal layer during submucosal dissection, we strongly believe that it might also facilitate dissection in living animals and/or humans. Finally, no data is available on esophageal ESD performed in humans by endoscopists who have received training using this model. Further prospective human studies are needed to demonstrate the efficacy and safety of the OTF method for esophageal ESD.
We revealed the potential advantages of the OTF method for performing safe and easy esophageal ESD. The OTF method is useful not only for experts but also for beginners, thus potentially contributing to the achievement of preferable outcomes for esophageal ESD worldwide and offering an opportunity to overcome the present learning curve.
None of the authors have any potential conflict of interests to declare.
- J. H. Rubenstein and N. J. Shaheen, “Epidemiology, diagnosis, and management of esophageal adenocarcinoma,” Gastroenterology, vol. 149, no. 2, pp. 302–317.e1, 2015.
- C. R. Chien, C. Y. Lin, and C. Y. Chen, “Incidence of adenocarcinoma of the esophagus among white Americans by sex, stage, and age,” Journal of the National Cancer Institute, vol. 100, no. 16, pp. 1184–1187, 2008.
- A. Shibata, T. Matsuda, W. Ajiki, and T. Sobue, “Trend in incidence of adenocarcinoma of the esophagus in Japan, 1993–2001,” Japanese Journal of Clinical Oncology, vol. 38, no. 7, pp. 464–468, 2008.
- M. Muto, K. Minashi, T. Yano et al., “Early detection of superficial squamous cell carcinoma in the head and neck region and esophagus by narrow band imaging: a multicenter randomized controlled trial,” Journal of Clinical Oncology, vol. 28, no. 9, pp. 1566–1572, 2010.
- B. J. Qumseya, H. Wang, N. Badie et al., “Advanced imaging technologies increase detection of dysplasia and neoplasia in patients with Barrett's esophagus: a meta-analysis and systematic review,” Clinical Gastroenterology and Hepatology, vol. 11, no. 12, pp. 1562–1570.e2, 2013.
- C. Katada, M. Muto, T. Manabe, N. Boku, A. Ohtsu, and S. Yoshida, “Esophageal stenosis after endoscopic mucosal resection of superficial esophageal lesions,” Gastrointestinal Endoscopy, vol. 57, no. 2, pp. 165–169, 2003.
- H. Takahashi, Y. Arimura, H. Masao et al., “Endoscopic submucosal dissection is superior to conventional endoscopic resection as a curative treatment for early squamous cell carcinoma of the esophagus (with video),” Gastrointestinal Endoscopy, vol. 72, no. 2, pp. 255–264.e2, 2010.
- S. Ono, M. Fujishiro, and K. Koike, “Endoscopic submucosal dissection for superficial esophageal neoplasms,” World Journal of Gastrointestinal Endoscopy, vol. 4, no. 5, pp. 162–166, 2012.
- S. Farhat, S. Chaussade, T. Ponchon et al., “Endoscopic submucosal dissection in a European setting. A multi-institutional report of a technique in development,” Endoscopy, vol. 43, no. 8, pp. 664–670, 2011.
- K. Ohata, K. Fu, M. Shouzushima et al., “A novel traction system for esophageal endoscopic submucosal dissection,” Endoscopy, vol. 44, no. 2, pp. E410–E411, 2012.
- M. Esaki, T. Matsumoto, K. Hirakawa et al., “Risk factors for local recurrence of superficial esophageal cancer after treatment by endoscopic mucosal resection,” Endoscopy, vol. 39, no. 1, pp. 41–45, 2007.
- J. S. Kim, B.-W. Kim, and I.-S. Shin, “Efficacy and safety of endoscopic submucosal dissection for superficial squamous esophageal neoplasia: a meta-analysis,” Digestive Diseases and Sciences, vol. 59, no. 8, pp. 1862–1869, 2014.
- A. Probst, D. Aust, B. Märkl, M. Anthuber, and H. Messmann, “Early esophageal cancer in Europe: endoscopic treatment by endoscopic submucosal dissection,” Endoscopy, vol. 47, no. 2, pp. 113–121, 2015.
- H. Neuhaus, G. Terheggen, E. M. Rutz, M. Vieth, and B. Schumacher, “Endoscopic submucosal dissection plus radiofrequency ablation of neoplastic Barrett's esophagus,” Endoscopy, vol. 44, no. 12, pp. 1105–1113, 2012.
- S. K. K. Tsao, T. Toyonaga, Y. Morita, T. Fujita, T. Hayakumo, and T. Azuma, “Modified fishing-line traction system in endoscopic submucosal dissection of large esophageal tumors,” Endoscopy, vol. 43, no. 2, p. E119, 2011.
- T. Uraoka, J. Kato, S. Ishikawa et al., “Thin endoscope-assisted endoscopic submucosal dissection for large colorectal tumors (with videos),” Gastrointestinal Endoscopy, vol. 66, no. 4, pp. 836–839, 2007.
- P.-J. Chen, H.-C. Chu, W.-K. Chang, T.-Y. Hsieh, and Y.-C. Chao, “Endoscopic submucosal dissection with internal traction for early gastric cancer (with video),” Gastrointestinal Endoscopy, vol. 67, no. 1, pp. 128–132, 2008.
- H. Imaeda, N. Hosoe, Y. Ida et al., “Novel technique of endoscopic submucosal dissection using an external grasping forceps for superficial gastric neoplasia,” Digestive Endoscopy, vol. 21, no. 2, pp. 122–127, 2009.
- Y. Saito, F. Emura, T. Matsuda et al., “A new sinker-assisted endoscopic submucosal dissection for colorectal cancer,” Gastrointestinal Endoscopy, vol. 62, no. 2, pp. 297–301, 2005.
- N. Sakamoto, T. Osada, T. Shibuya et al., “Endoscopic submucosal dissection of large colorectal tumors by using a novel spring-action S-O clip for traction (with video),” Gastrointestinal Endoscopy, vol. 69, no. 7, pp. 1370–1374, 2009.
- J. Y. Ahn, K. D. Choi, J. Y. Choi et al., “Transnasal endoscope-assisted endoscopic submucosal dissection for gastric adenoma and early gastric cancer in the pyloric area: a case series,” Endoscopy, vol. 43, no. 3, pp. 233–235, 2011.
- O. Motohashi, K. Nishimura, N. Nakayama, S. Takagi, and N. Yanagida, “Endoscopic submucosal dissection (two-point fixed ESD) for early esophageal cancer,” Digestive Endoscopy, vol. 21, no. 3, pp. 176–179, 2009.
- A. Parra-Blanco, D. Nicolas, M. R. Arnau, A. Z. Gimeno-Garcia, L. Rodrigo, and E. Quintero, “Gastric endoscopic submucosal dissection assisted by a new traction method: the clip-band technique. A feasibility study in a porcine model (with video),” Gastrointestinal Endoscopy, vol. 74, no. 5, pp. 1137–1141, 2011.
- P.-J. Chen, W.-C. Huang, H.-P. Wang et al., “Percutaneous transgastric traction-assisted esophageal endoscopic submucosal dissection: a randomized controlled trial in a porcine model,” Scandinavian Journal of Gastroenterology, vol. 47, no. 11, pp. 1386–1393, 2012.
- M. Hirota, M. Kato, M. Yamasaki et al., “A novel endoscopic submucosal dissection technique with robust and adjustable tissue traction,” Endoscopy, vol. 46, no. 6, pp. 499–502, 2014.
- Y.-K. Tsou, W.-Y. Chuang, C.-Y. Liu et al., “Learning curve for endoscopic submucosal dissection of esophageal neoplasms,” Diseases of the Esophagus, 2015.
- W. H. Holderman, K. P. Etzkorn, S. A. Patel, J. M. Harig, and J. L. Watkins, “Endoscopic findings and overtube-related complications associated with esophageal variceal ligation,” Journal of Clinical Gastroenterology, vol. 21, no. 2, pp. 91–94, 1995.
Copyright © 2016 Ken Ohata et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.