Research Article  Open Access
Minyu Zeng, Carl Schaschke, "High Pressure Falling Sinker Liquid Viscosity Determination without Supplementary Density Data: A New Approach", International Journal of Chemical Engineering, vol. 2009, Article ID 747592, 8 pages, 2009. https://doi.org/10.1155/2009/747592
High Pressure Falling Sinker Liquid Viscosity Determination without Supplementary Density Data: A New Approach
Abstract
Accurate measurement and determination of liquid viscosity data under high pressure conditions requires knowledge of liquid density data. In this study, a high pressure falling sinkertype viscometer was used to determine the viscosity of ndodecane at elevated pressures up to 132 MPa without supplementary knowledge of liquid density. The viscometer, which involves the downward movement of a cylindrical sinker under the influence of gravity through the liquid contained within a closed tube, avoided the need for density data by repeating the sinkertiming experiments with two geometrically similar but differentsized sinkers thereby allowing the liquid density in the associated formulae to be eliminated. Furthermore, it was possible to subsequently derive liquid density. Both viscosity and density data were compared to published data for which good correlation was found for viscosity. To minimize errors, it is suggested that the two sinkers for such an approach should be of sufficiently differing densities.
1. Introduction
The viscosity of liquids is an important property in fluid flow and the estimation of other physical properties such as in the estimation of diffusion in liquid systems [1]. Accurate data is therefore required for many applications in engineering design. The high pressure falling body viscometer is a simple and effective device for the accurate measurement of viscosity for liquids over a wide range of viscosities [2–12]. Consisting of a vertically positioned closed tube, a cylindrical sinker is allowed to descend at terminal velocity under the influence of gravity through a liquid under investigation. The viscosity is readily determined from the time for the sinker to descend a known distance and the physical details of the viscometer. However, under very high pressure conditions, the calculations are not so straightforward as liquids may not be assumed to be incompressible since density data is required to complete the viscosity calculations. Where experimental data may not be available, published experimental data may be available; predictive approaches based on critical point data or multiparameter corresponding states methods may be used for many pure fluids and some mixtures [13–16]. For many complex fluids and mixtures such as automotive fluids, liquid foods and mixtures such as biodiesels, density data may, however, not be available [17]. Although the critical point data can be predicted in some cases, this can lead to complication and mathematical uncertainty which can significantly influence the determined viscosity data. The resulting error in the data may be correspondingly very significant thus rendering it unreliable for good scientific and engineering purposes. To overcome the need for compressibility data for liquids at high pressure, we examined the use of two geometrically similar sinkers to determine liquid viscosity as a way of overcoming the need for supporting density data.
2. Falling Body Viscometer
The highpressure viscometer used in this work was based on the falling sinker design. Designed and fabricated by the National Engineering Laboratory (NEL) in East Kilbride, UK, the viscometer consists of a vertical tube containing a test liquid through which the sinker falls freely under the influence of gravity (Figure 1). The rate of descent of the sinker which falls concentrically is related to the displacement of liquid and frictional effects in the annular region between the tube and the sinker. As the sinker descends, the liquid is displaced upwards forcing it to pass through the annular gap between the walls of the tube and the sinker.
The sinker and the tube are fabricated from a solid bar of nonmagnetic stainless steel to minimise compressibility and thermal expansion effects. Designed to be selfcentring, the sinker used has a cylindrical body with a hemispherical nose embedded with a ferrite core (Figure 2). Ratios of sinker to inner tube radii in excess of 0.95 are required for the sinker to descend concentrically within the tube. Sinkers which fall eccentrically are well known to cause significant errors [18, 19].
The viscosity is determined directly from the time taken for the sinker to descend a fixed distance. In this case, this is the distance between two detection coils wrapped around the outside of the tube with the location of the sinker being detected by the change in inductance in the coils. The coils are set a distance of 12 mm apart and consist of around 200 turns of lacquered copper wire, each with approximately equal electrical resistance, and both forming the active arm of a balanced bridge circuit. The outofbalance signal from the bridge is amplified and passed through an AC/DC converter to give a DC signal which increases to a maximum when the ferrite core of the sinker is positioned at the centre of the coil (Figure 3). From the dimensions of the viscometer tube and sinker, and the time for the sinker to fall freely with terminal velocity between two points, the viscosity can be determined.
Samples of ndodecane oil were sealed into the tube using a shrinkable PTFE expansion sheath. It was essential that neither air nor particles remained entrapped with the sample which could influence the rate of sinker descent. The tube and associated assembly was housed in a highpressure vessel and maintained at a constant temperature by air circulation (Figure 4). The pressure is transmitted in the oil through the flexible PTFE sheath using a paraffin/Shell Tellus oil mixture as the hydraulic medium by pressure amplification to a maximum possible operating pressure of 500 MPa. This mixture offers good lubricating properties with a low freezing point. The entire high pressure vessel itself is mounted on bearings and rotated to return the sinker to its original starting position between each measurement. A calibrated Kistler piezoresistive pressure gauge, type 46180A and calibrated to 500 MPa, was used to measure the high pressure within the viscometer tube.
3. Analysis
The rate of descent of a sinker in a closed tube is based on a force balance of gravity, buoyancy and resistive shear forces with fully developed laminar flow and has been previously shown [20, 21] in which the viscosity can be expressed in the form
where
The thermal expansion coefficient and compressibility of the sinker are small (7.6 10^{6} K^{1} and 3.075 10^{6} MPa^{1}, resp.) and can thus be ignored. In contrast, small measurement errors in radii and result in significant errors; the effects of which have been studied by Wehbeh et al. [22]. Ideally, the coefficient A is based on the physical dimensions of the sinker as shown. In practice, however, there is often a deviation in viscosity determined experimentally due to wall and end effects [23–26]. The coefficient A is therefore adjusted using a calibration liquid of known properties under high pressure [27]. To confirm the cause of these deviations 2D and 3D CFD studies have previously shown the presence of vortices shedding from the trailing edge of the sinker. The simulations have shown that fully developed laminar flow is not met within the annual gap [21, 28]. It has, however, been confirmed that coefficient A tends towards the theoretical value for a modified Reynolds numbers below 60 [18], where
For two geometrically similar sinkers tested in the same liquid at the same temperature and pressure, the density of the liquid can be deduced to be
Likewise, for the two sinkers, the viscosity is obtained from
To determine both the viscosity and density, this may be possible for two measurable and distinguishable values of A and sinker descent times.
4. Materials and Methods
In this work, ndodecane with a purity greater than 99% purchased from VWR International Ltd, UK was used as the test liquid. The properties of ndodecane have been previously reported [29].
4.1. Density
The density data was based on using a Taittype equation of the form
in which
4.2. Viscosity
The viscosity was found from
for which the dimensionless viscosity was calculated from
where the core volume is dependent on the absolute temperature in the form
The parameters are given in Table 1.

5. Sinker Selection
To determine the viscosity without supplementary density data, it is important to select sufficiently different sized but geometrically similar sinkers. There are certain other criteria that must also be fulfilled in this selection process. In particular, the sinker is required to be selfcentering during descent. The time for descent for the two sinkers is also required to be appreciably different, for which the time for descent is
It is worth noting that since the density of most liquids does not increase significantly with pressure then as the pressure increases large values of A contributes to longer descent time. Conversely, similar values for A imply the need for appreciably different sinker densities. The details of the two sinkers used in this study are given in Table 2.

For both sinkers used in this study, readings for the time for sinker descent were recorded over a pressure range of 0.1 MPa to 132 MPa. Above this applied pressure, the sinker was not found to descend. This may have been due to a pressureinduced phase change. This phenomenon is found in certain liquids including hydrocarbons and lipids. All readings were repeated in triplicate.
6. Results
The sinker fall times for both sinkers are shown in Figure 5. This shows a rising increase in time with increasing applied pressure up to 132 MPa. Beyond this pressure, the sinker failed to descend. The modified Reynolds number for both sinkers at these pressures is shown in Figure 6. For both sinkers, the experimental coefficients A tends towards a constant value for which the average values were found to be 16122.5 Pa^{1} (σ_{n1 }= 320.2 Pa^{1}) for sinker 1 while for sinker 2, A was 4062.6 Pa^{1} (σ_{n1 }= 56.0 Pa^{1}). For comparison, the values for A calculated from the dimensions of sinker and tube are 16575.5 Pa^{1} and 3890.7 Pa^{1}, respectively (Table 2), and are equivalent to a zero modified Reynolds number. While the two experimental and theoretical values are reasonably close, it should be noted that differences exist due to inaccuracies in measurement as well as end effects during sinker descent described earlier. Table 3 presents the sinker fall data and coefficients for the same applied pressures.

The deviation of evaluated viscosity and density data is presented in Table 4 and Figures 7 and 8.

Using this new approach, the determined viscosity data is comparable to that of Caudwell et al. [29]. The source of the errors obtained can be attributed to the variation of the coefficient A for which a summary of the errors is presented in Table 5.

7. Error Transfer Analysis
Although the error transfer analysis can be used to identify parameters other than A which are able to reduce the error caused by the coefficient in density determination, it is worth noting that the errors in both calibration coefficients may be of significance. In this analysis, the errors are assumed to be associated with both coefficients A_{1 }and A_{2.} In general, by assuming an error ∆R on the final parameter R for which
then using Taylor’s series expansion with only first order terms, the error can be given in the form:
for which the relative error term is thus
7.1. Density
Based on the above, the relative error in density evaluation can be written as
in which
Using x to represent the relative error for the corresponding coefficient value of sinker, the error in the calculated density is, after simplification
The relative error is therefore proportional to the difference of the two relative errors in A. However, the magnitude of the errors shown in Table 6 do not conclude that the errors were attributed to the coefficients A_{1} and A_{2}.

In terms of fall times t_{1} and t_{2}, the approach to take the relative error as
rewritten as
with further simplification in which
gives a relative error of
In this case, to minimize the error in liquid density with respect to the liquid density in which
gives suggesting that the liquid density should be the geometric mean of the two sinkers. Care should be taken with this result since it implies that one density should be less than that of the liquid thereby violating the phenomenon of buoyancy.
7.2. Viscosity
Again, it is worth noting that the errors between average value of A and experimental data are of great importance. The errors for A have been shown previously (Table 6) and are not large. The error transfer analysis was carried out to show how the errors between the average values of the two calibration coefficients and the corresponding experimental data which can affect the final error in viscosity determination. The basic condition is that the errors between the average and experimental data of A are not big.
The error analysis for viscosity determination is based on
in which the Taylor series expansion is
where
The largest possible relative error in the calculated density is, after simplification
Writing in the form of fluid density gives
To minimize the error in which
gives . This suggests that the widest possible difference in density is required.
8. Conclusions
It is possible to use the high pressure falling body sinker to determine the viscosity of liquid without the need for supplementary knowledge of liquid density data. Good agreement is found with determining the viscosity data although less so for the derived density data. There are inherent errors in this new approach which have been addressed. However, in order to ensure that the errors remain as small as possible, it is recommended that sinkers should be used, which exhibit good self centering properties. The radius ratio of the sinker to inner tube should be greater than 0.95. Additionally, it is necessary to use two sinkers with a large difference in value of A to avoid inaccuracies. The density of the sinkers should be obviously greater than that of the liquid tested so that it is capable of descening although the bulk densities of the two sinkers should be sufficiently dissimilar.
Nomenclature
:  Constant 
:  Calibration coefficient (Pa^{1}) 
:  Constant (Pa) 
:  Constant (−) 
:  Gravitational acceleration (ms^{2}) 
:  Length of sinker (m) 
:  Distance between the centre of two detecting coils (m) 
:  Original length of the sinker (m) 
:  Distance between two detecting coils (m) 
:  Mass of sinker (kg) 
:  Molecular weight (kgmol^{1}) 
:  Pressure (Pa) 
:  Reference pressure (Pa) 
:  Inner radius of tube (m) 
:  Radius of sinker (m) 
:  Parameter 
Re_{m}:  Modified Reynolds number (−) 
:  Falling duration of the sinker (s) 
:  Temperature (K) 
:  Reference temperature (K) 
:  Parameter 
:  Terminal velocity (ms^{1}) 
:  Molar volume 
:  Molar core volume 
:  Relative error (%). 
Greek
:  Thermal expansion coefficient (K^{1}) 
:  Compressibility of the sinker (MPa^{1}) 
:  Viscosity of fluid (Pas) 
:  Dimensionless viscosity (−) 
:  Density of fluid (kgm^{3}) 
:  Reference density (kgm^{3}) 
:  Density of sinker (kgm^{3}). 
Subscript
1:  Sinker 1 
2:  Sinker 2. 
References
 M. R. Riazi and G. N. AlOtaibi, “Estimation of viscosity of liquid hydrocarbon systems,” Fuel, vol. 80, no. 1, pp. 27–32, 2001. View at: Publisher Site  Google Scholar
 N. A. Park and T. F. Irvine Jr., “Measurements of rheological fluid properties with the falling needle viscometer,” Review of Scientific Instruments, vol. 59, no. 9, pp. 2051–2058, 1988. View at: Publisher Site  Google Scholar
 N. A. Park and T. F. Irvine Jr., “The falling needle viscometer a new technique for viscosity measurements,” Wärmeund Stoffübertragung, vol. 18, no. 4, pp. 201–206, 1984. View at: Publisher Site  Google Scholar
 M. Brizard, M. Megharfi, E. Mahé, and C. Verdier, “Design of a high precision fallingball viscometer,” Review of Scientific Instruments, vol. 76, no. 2, Article ID 025109, 6 pages, 2005. View at: Publisher Site  Google Scholar
 J. H. Dymond, K. J. Young, and J. D. Isdale, “Transport properties of nonelectrolyte liquid mixturesII. Viscosity coefficients for the nhexane + nhexadecane system at temperatures from 25 to 100${}^{\circ}\text{C}$ at pressures up to the freezing pressure or 500 MPa,” International Journal of Thermophysics, vol. 1, no. 4, pp. 345–373, 1980. View at: Publisher Site  Google Scholar
 J. B. Irving and A. J. Barlow, “An automatic high pressure viscometer,” Journal of Physics E, vol. 4, no. 3, pp. 232–236, 1971. View at: Publisher Site  Google Scholar
 H.P. Mattischek and R. Sobczak, “New cell for measurement of viscosity under high pressure,” Measurement Science and Technology, vol. 5, no. 7, pp. 782–785, 1994. View at: Google Scholar
 S. C. Vant, Investigation of fluid properties at nonambient conditions, Ph.D. thesis, Department of Chemical and Process Engineering, University of Strathclyde, Glasgow, UK, 2003.
 Z.S. Sha, “The improvement on the falling needle viscometer,” Review of Scientific Instruments, vol. 68, no. 4, pp. 1809–1811, 1997. View at: Google Scholar
 A. M. J. Davis and H. Brenner, “The fallingneedle viscometer,” Physics of Fluids, vol. 13, no. 10, pp. 3086–3088, 2001. View at: Publisher Site  Google Scholar
 A. Kumagai, Y. Kawase, and C. Yokoyama, “Falling capillary tube viscometer suitable for liquids at high pressure,” Review of Scientific Instruments, vol. 69, no. 3, pp. 1441–1445, 1998. View at: Google Scholar
 C. Boned, C. K. ZébergMikkelsen, A. Baylaucq, and P. Daugé, “Highpressure dynamic viscosity and density of two synthetic hydrocarbon mixtures representative of some heavy petroleum distillation cuts,” Fluid Phase Equilibria, vol. 212, no. 12, pp. 143–164, 2003. View at: Publisher Site  Google Scholar
 A. Allal, M. MohaOuchane, and C. Boned, “A new free volume model for dynamic viscosity and density of dense fluids versus pressure and temperature,” Physics and Chemistry of Liquids, vol. 39, no. 1, pp. 1–30, 2001. View at: Google Scholar
 M. M. Aalto and K. I. Keskinen, “Liquid densities at high pressures,” Fluid Phase Equilibria, vol. 166, no. 2, pp. 183–205, 1999. View at: Publisher Site  Google Scholar
 V. N. Belonenko, V. M. Troitsky, Y. E. Belyaev, J. H. Dymond, and N. F. Glen, “Application of a micro(p, V, T) apparatus for measurement of liquid densities at pressures up to 500 MPa,” Journal of Chemical Thermodynamics, vol. 32, no. 9, pp. 1203–1219, 2000. View at: Publisher Site  Google Scholar
 E. Kiran and Y. L. Sen, “Highpressure viscosity and density of nalkanes,” International Journal of Thermophysics, vol. 13, no. 3, pp. 411–442, 1992. View at: Publisher Site  Google Scholar
 H. Korsten, “Critical properties of hydrocarbon systems,” Chemical Engineering and Technology, vol. 21, no. 3, pp. 229–244, 1998. View at: Google Scholar
 C. J. Schaschke, S. Allio, and E. Holmberg, “Viscosity measurement of vegetable oil at high pressure,” Food and Bioproducts Processing, vol. 84, no. 3, pp. 173–178, 2006. View at: Publisher Site  Google Scholar
 J. B. Irving, “The effect of nonvertical alignment on the performance of a fallingcylinder viscometer,” Journal of Physics D, vol. 5, no. 1, pp. 214–224, 1972. View at: Publisher Site  Google Scholar
 N. D. Cristescu, B. P. Conrad, and R. TranSonTay, “A closed form solution for falling cylinder viscometers,” International Journal of Engineering Science, vol. 40, no. 6, pp. 605–620, 2002. View at: Publisher Site  Google Scholar
 C. J. Schaschke, S. Abid, I. Fletcher, and M. J. Heslop, “Evaluation of a falling sinkertype viscometer at high pressure using edible oil,” Journal of Food Engineering, vol. 87, no. 1, pp. 51–58, 2008. View at: Publisher Site  Google Scholar
 E. G. Wehbeh, T. J. Ui, and R. G. Hussey, “End effects for the falling cylinder viscometer,” Physics of Fluids A, vol. 5, no. 1, pp. 25–33, 1992. View at: Google Scholar
 J. F. Stalnaker and R. G. Hussey, “Wall effects on cylinder drag at low Reynolds number,” Physics of Fluids, vol. 22, no. 4, pp. 603–613, 1979. View at: Google Scholar
 T. Lommatzsch, M. Megharfi, E. Mahe, and E. Devin, “Conceptual study of an absolute fallingball viscometer,” Metrologia, vol. 38, no. 6, pp. 531–534, 2001. View at: Publisher Site  Google Scholar
 P. Y. Huang and J. Feng, “Wall effects on the flow of viscoelastic fluids around a circular cylinder,” Journal of NonNewtonian Fluid Mechanics, vol. 60, no. 23, pp. 179–198, 1995. View at: Google Scholar
 G. H. Ristow, “Wall correction factor for sinking cylinders in fluids,” Physical Review E, vol. 55, no. 3, pp. 2808–2813, 1997. View at: Google Scholar
 N. A. Park and T. F. Irvine Jr., “Falling cylinder viscometer end correction factor,” Review of Scientific Instruments, vol. 66, no. 7, pp. 3982–3984, 1995. View at: Publisher Site  Google Scholar
 F. Gui and T. F. Irvine Jr., “Theoretical and experimental study of the falling cylinder viscometer,” International Journal of Heat and Mass Transfer, vol. 37, supplement 1, pp. 41–50, 1994. View at: Google Scholar
 D. R. Caudwell, J. P. M. Trusler, V. Vesovic, and W. A. Wakeham, “The viscosity and density of ndodecane and noctadecane at pressures up to 200 MPa and temperatures up to 473 K,” International Journal of Thermophysics, vol. 25, no. 5, pp. 1339–1352, 2004. View at: Publisher Site  Google Scholar
Copyright
Copyright © 2009 Minyu Zeng and Carl Schaschke. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.