Application of 19F NMR Spectroscopy for Content Determination of Fluorinated Pharmaceuticals
A simple, rapid, and selective quantitative nuclear magnetic resonance spectroscopic method was evaluated for the determination of the content of fluorinated pharmaceuticals. 19F NMR spectra were either obtained in dimethylsulfoxide-d6 or aqueous buffer, using trifluoroacetic acid as internal standard. Quantification of 13 fluorine-containing pharmaceuticals spanning various pharmacological classes was accomplished using the proposed method. The method was found to be fit for purpose (interday precision 1.2% relative standard deviation) and may thus be applied for routine analysis and quality control of fluorine-containing pharmaceuticals due to its simplicity, nondestructive sample measurement, reliability, and high specificity. Therefore, 19F NMR may serve as a suitable analytical tool for the identification and selective determination of fluorinated pharmaceuticals used as reference materials and bulk samples.
There is growing need for modern analytical techniques for analysis of fluorine-containing pharmaceutical substances including active pharmaceutical ingredients, brand and generic finished products, and also for the detection of potentially counterfeited pharmaceuticals. These new techniques are aimed not only at maximizing analytical throughput but also minimizing expenses while preserving acceptable method performance characteristics. Nuclear magnetic resonance spectroscopy (NMR) is such a technique whose scope of applications continues to expand. Indeed with the marked improvements on the sensitivity of NMR spectrometers, quantitative analytical applications have tremendously sprung up across many fields including foods, beverages [1–5], and pharmaceuticals . NMR spectroscopy offers unparalleled rich information on samples, and, with rapid validated methods, high throughput can be achieved without destroying the sample . Unlike the current chromatographic techniques that require reference standards, often expensive and unavailable in many laboratories, NMR spectroscopy enables quantification without necessity of a primary reference standard, thus lowering the cost of analysis [8, 9].
Traditionally, most of the NMR spectroscopic applications are based on 13C and 1H nuclides due to the relatively high abundance of carbon and hydrogen in natural compounds compared to phosphorus and fluorine. As an alternative nuclide, 19F offers the advantages of higher natural abundance compared to 13C and less risk of signal overlap compared to 1H since proton NMR shows a narrow range of the chemical shift (typically 0–10 ppm) and shows an increased spectral complexity due to coupling of neighbouring protons. Furthermore, 19F NMR has a broader chemical shift range (approx. 500 ppm) , which helps to avoid signal overlap and shows less interference from homonuclear coupling [11–13] and more importantly the solvent (mostly containing no fluorine) has no effect on the 19F NMR signal and as such solvent suppression is not necessary. This makes the technique uniquely suitable for quantification of fluorine-containing compounds in various matrices including finished pharmaceutical products. Currently over 200 medicines and one-third of the top-performing drugs on the market  contain fluorine atoms in their structure. This gives credence to the plausibility of using 19F NMR spectroscopy as an alternative analytical technique for assay of fluorinated drugs.
Currently, various instrumental analytical methods based on chromatography, spectroscopy, and electrochemistry are used to determine some of these fluorine-containing compounds. Chromatographic techniques such as thin layer chromatography [15, 16], HPLC [17–20], gas chromatography or capillary electrophoresis , and spectrophotometric methods such as UV spectroscopy [17, 22] and electroanalytical techniques  have been utilized for purity determination of fluorinated compounds. These methods generally require the use of reference substances, expensive columns, and large volumes of organic solvents and in some cases laborious and time-consuming sample preparative steps. More often, the analysis takes a long time. NMR methods have been proven to be simple, reasonably rapid, and cost-effective in the long term compared to HPLC [24, 25]. Except for the determination of decomposition products of flurazepam dihydrochloride in various media at different temperatures by 19F NMR [26, 27], to the best of our knowledge, no single study has been published where 19F NMR spectroscopy has been applied for the content determination of fluorinated pharmaceuticals spanning across many pharmacological classes including anticancer drugs (fludarabine and flucytosine), antipsychotic agents (fluphenazine), anxiolytics (fluoxetine), sedative hypnotics (flumazenil and flunitrazepam), antiandrogens (flutamide), nonsteroidal anti-inflammatory agents (flurbiprofen), antifungal agents (fluconazole and flucytosine), and glucocorticoids (fluprednidene and fluorometholone) (Figure 1). The aim of this study is therefore to validate and apply a 19F NMR spectroscopic method for the determination of content of twelve fluorine-containing pharmaceuticals.
2. Experimental Section
2.1. Samples and Chemicals
The samples for analysis were fluorinated pharmaceuticals consisting of fluorometholone, flutamide, sodium flurbiprofen (USP-Standard, Rockville, MD, USA), flumazenil, flunitrazepam (Hoffmann-La Roche, Grenzach-Wyhlen, Germany), fluvastatin sodium hydrate (Hemofarm, Vrsac, Serbia), fluprednidene acetate (Alcon-Pharma, Freiburg, Germany), fluphenazine dihydrochloride (from the former pharmaceutical company Byk-Gulden, Konstanz, Germany), fluoxetine hydrochloride (Teva, Ulm, Germany), fludarabine phosphate (CRS-standard from EDQM, Strasbourg, France), flurazepam hydrochloride (Roche Diagnostics, Mannheim, Germany), flucytosine (Sigma-Aldrich, Taufkirchen, Germany), and fluconazole (Dr. Reddy’s, Hyderabad, India). Proanalysis quality reagents, trifluoroacetic acid (TFA), and dimethylsulfoxide-d6 were obtained from Sigma-Aldrich (Taufkirchen, Germany).
2.2. Confirmation of Purity of Trifluoroacetic Acid (TFA) Using Ion Chromatography
The purity of TFA intended to be used as internal standard for 19F NMR spectroscopy was determined using a validated in-house ion chromatographic method. The separation of TFA was performed on a Metrohm system (Deutsche Metrohm GmbH & Co. KG, Filderstadt, Germany), containing an autosampler (Professional Sample Processor Mod. 858) connected to a stirrer Mod. 801 and Dosino Mod. 800 (10 mL) for degassing and stirring. The liquid TFA standard solution was transferred via a loop to the anionic column (Metrosep A Supp 5 150/4.0, Metrohm), maintained at 35°C in the chromatography Compact IC system (Mod. 881, Metrohm). The eluent consisted of 3.2 mM Na2CO3 and 2.4 mM NaHCO3 delivered at a flow rate of 0.7 mL·min−1. The injection volume was 20 μL. All steps of ion chromatographic measurements were performed using Metrohm software MagIC NET Professional 3.1. The purity was found to be 99.83% by ion chromatography. For NMR measurements, 26.62 mg TFA was dissolved in 2 ml dimethylsulfoxide-d6 as internal standard solution.
2.3. NMR Analysis
Sample powder equivalent to 5 mg of fluorometholone, flumazenil, flunitrazepam, flutamide, fluvastatin sodium hydrate, and fluprednidene acetate were each dissolved in 1.0 ml dimethylsulfoxide-d6. Then to 500 μl of the resultant solution, 100 μl of TFA internal standard solution for referencing and quantitation (see Section 2.2) was added before NMR analysis. Similarly, sample powder equivalent to 5 mg of fluphenazine, fluoxetine, fludarabine, flurazepam, flucytosine, fluconazole, and flurbiprofen were prepared in 1.0 ml of buffer (pH 9.0) consisting of 1000 mg Li2CO3, 50 ml distilled water, 10.2 ml 1 M HCl, and 4 mg NaN3. For NMR measurements, 500 μl of the resultant solution, 50 μl of TFA, and 50 μl TSP (sodium salt of 3-(trimethylsilyl)-propionate acid-d4)/D2O (for locking) were vortexed until the sample and internal standard were completely dissolved. The clear solution was transferred into an analytical NMR tube followed by acquisition of NMR spectra on a Bruker Avance III HD 400 NanoBay spectrometer (Bruker BioSpin, Rheinstetten, Germany) equipped with a 5 mm BBO probe using a Bruker Automatic Sample Changer (SampleXpress). The spectra were automatically acquired under the control of ICON-NMR (Bruker BioSpin, Rheinstetten, Germany). All 19F NMR spectra were recorded in the range of F −20.0 to −210.0 ppm (CF3COOH as reference, ppm) relative to TFA ( ppm). The NMR probe was maintained at 300 K during the whole experiment. A delay time () of 20 s was adopted to ensure full relaxation, and pulse angle of 90° was used to provide maximum signal-to-noise ratio (S/N) and minimize the influence of off-resonance effect on the accuracy of 19F NMR measurement . To maximize sensitivity, 512 scans were collected into 65536 data points over a spectral width of typically 95–125 ppm (for details see Tables 1 and 2). The Bruker pulse program, zgfhigqn, was used with a receiver gain (RG) of 203. All 19F NMR spectra were automatically phased and baseline corrected for accurate quantitative measurements using the Topspin 3.2 software package (Bruker BioSpin, Rheinstetten, Germany). Peak areas were obtained by electronic integration of the expanded regions around diagnostic resonances, using an integral limit of ±20 Hz around the corresponding signals. The percentage of error (reported in % relative standard deviation) was found to be less than 1%.
The purity of the analyte was calculated using the following:where and are the areas of selected signals for sample of fluorinated pharmaceutical and internal standard; and are the number of fluorine atoms in the fluorinated pharmaceutical and internal standard; and are the molar mass of sample of fluorinated pharmaceutical and internal standard; and are amounts of fluorinated pharmaceutical and internal standard, respectively; is the purity of the internal standard as determined by ion chromatography.
To achieve acceptable accuracy and precision necessary for quantitative determination, the influence of the relaxation delay time, , on the S/N ratio of selected signals was investigated in the range of 5–35 s. Saturation of S/N was found in the range of 20–35 s. Therefore, for a short spectra acquisition run time, = 20 s was preferred.
2.4. NMR Method Validation and Comparison
The proposed 19F NMR method was validated for selectivity, trueness, precision, and sensitivity according to the Eurachem criteria . The trueness of the method was assessed in comparison with certifications of the manufacturers as well as with an independent method for some of the fluorinated compounds. HPLC analysis was conducted for fluorometholone, flumazenil, flunitrazepam, fluvastatin, and fluprednidene acetate (by using the specific methods for these substances from the admission dossier). The results were compared with the certificates of analysis from USP for flutamide or fluorometholone.
Selectivity of the proposed method was evaluated by determining the resolution between fluorine signals and other nuclei while precision was investigated by consecutive NMR measurements of a fluvastatin sample (including sample preparation) over 5 different days. The limit of detection (LOD) was manually estimated based on a small, but still integrable, signal in the region of the target resonance .
3. Results and Discussion
3.1. Method Optimization
Dimethylsulfoxide-6 was selected as a solvent for the samples on account of its good dissolving properties while TFA was used as the internal standard. It is important to ascertain the purity of the reference substances before use in analysis since high purity is a prerequisite for a substance to be considered suitable for use as a standard in NMR spectroscopy. The purity of TFA used as an internal standard was established by ion chromatography to be 99.83%, which is considered adequate for use in our analytical purpose. TFA was selected as the internal standard for the purity assessment of the fluorine-containing pharmaceuticals on account of its good solubility in dimethylsulfoxide-d6 and also its sharp distinct signal at −75.0 ppm (3F, s), which is well separated from the compounds being tested. Normally, the number of neighbouring protons influences the multiplicity of the 19F signal observed. However, to avoid this effect, proton decoupling was applied, so that each 19F signal appears as a singlet, which simplifies the spectra.
Experimental parameters for 19F NMR analysis of fluorinated pharmaceuticals were selected to optimize the NMR method with respect to accuracy, precision, and analysis time. The optimized measurement parameters were established as TD = 65536; number of scans (NS) = 512, s; . Longitudinal relaxation delay of the fluorine compounds was determined by the inversion recovery pulse sequence method, using Relaxation Bruker program which fitted the data to the exponential equation:where I is the intensity of the compound of interest resonance at the inversion delay timeτand at the equilibrium state and P is a constant. The 19F NMR signals selected for quantitative purposes and the selected spectral widths (SW) are shown in Tables 1 and 2. 19F NMR spectra showed distinct, well separated, nonoverlapping signals that were selected for quantitation.
3.2. NMR Validation and Applicability
The selectivity of the NMR method was demonstrated since there was no signal overlap with other components. Acceptable precision was established for a fluvastatin sample with intraday and interday relative standard deviations (coefficient of variation) of 0.9 and 1.2%, respectively. The limit of detection was less than 0.1 g/100 g in 27 evaluated spectra (average limit of detection 0.06 g/100 g). Trueness was similarly demonstrated since the assay values as determined using NMR spectroscopy and HPLC were <5%.
The purity of 13 fluorine-containing pharmaceuticals was measured using the proposed 19F NMR method (Tables 3 and 4). Supplementary Figures S1–S13 show the 19F NMR spectra of all investigated compounds (see Supplementary Material available online at https://doi.org/10.1155/2017/9206297). A potential shortcoming of 19F NMR is that, due to the wide chemical shift dispersion, a single excitation pulse cannot produce a uniform excitation over the entire range of the chemical shift region of interest . To overcome this problem usually two measurements have to be performed: one overview experiment over the whole range to identify the optimal sweep width and then a second experiment for quantification in this range. Additionally, the internal standard should have a similar chemical shift as the signal of interest.
The reliability of the results obtained by the 19F NMR spectroscopic method was tested by comparing them to the data obtained using HPLC method and certifications of manufacturers. The results are presented in Tables 3 and 4. In all samples, small differences were noted in the content of the fluorinated pharmaceuticals determined through the NMR method compared with HPLC method and as such the two methods can be equally applied in the analysis of fluorinated pharmaceuticals. Only one inconsistency was observed during our experiments: the spectrum of flurazepam (Figure S13) contains two distinct signals, which could be due to hydrolytic decomposition in acidic water solution previously observed in 19F NMR leading to a 44 : 56 ratio between flurazepam and its ring-opened form after 24 h equilibrium . In the current study, the combined integral of both signals was used for quantification, but further analysis such as using LC/MS/MS is required for clarification.
Although proton NMR has been used for quantitative determination of pharmaceutical compounds in different matrices, there are few official applications . The USP, for example, prescribes the use of NMR for assay of amylnitrite isomers. The European Pharmacopoeia uses NMR for the characterization of impurities in hydroxypropylcyclodextrin and poloxamer. Generally, the pharmacopoeia uses NMR spectroscopy mainly for the identification of drugs such as low molecular weight (LMW) heparins, tobramycin, and hydrocortisone sodium phosphate . Additionally, reagents such as adenosine and butoxycaine are identified using both 1H NMR and 13C NMR spectroscopy. However, no compendium prescribes the use of 19F NMR for any assay. Therefore, our research expands the applicability of 19F NMR spectroscopy beyond purity determination to routine assays with appreciable sensitivity. However, to be used in practice, it would be necessary to extend the domain of application of the method for fluorine-containing drugs to expand the database. To the best of our knowledge, this study is the largest evaluation of fluorinated pharmaceuticals samples by 19F NMR spectroscopy.
The proposed 19F NMR spectroscopic method demonstrated adequate accuracy, simplicity, and reproducibility for a rapid routine analysis and quality control of important compounds of fluorine-containing pharmaceuticals. The 19F NMR method was successfully applied to determine the purity of fluorinated pharmaceuticals. The direct relationship between the peak area and number of fluorine nuclides was exploited. Due to the lack of interferences or signal overlap, 19F NMR offers adequate specificity for quantification of pharmaceuticals without preceding extensive sample preparation. Additionally, 19F NMR spectroscopy combines the speed, simplicity, and low cost per analysis. No reference standard of the analyte is required for quantification and also the method is nondestructive.
The views expressed in this paper do not necessarily reflect those of the employers of the authors.
Conflicts of Interest
The authors declare that there are no conflicts of interest.
The authors are grateful to Jürgen Geisser, Klaus Müller, and Andreas Scharinger for excellent technical assistance.
Supplementary Figures S1-S13 (19F NMR spectra of all investigated compounds).
M. F. Marcone, S. Wang, W. Albabish, S. Nie, D. Somnarain, and A. Hill, “Diverse food-based applications of nuclear magnetic resonance (NMR) technology,” Food Research International, vol. 51, no. 2, pp. 729–747, 2013.View at: Publisher Site | Google Scholar
L. Laghi, G. Picone, and F. Capozzi, “Nuclear magnetic resonance for foodomics beyond food analysis,” TrAC Trends in Analytical Chemistry, vol. 59, pp. 93–102, 2014.View at: Publisher Site | Google Scholar
Y. B. Monakhova, T. Kuballa, and D. W. Lachenmeier, “Chemometric methods in NMR spectroscopic analysis of food products,” Journal of Analytical Chemistry, vol. 68, no. 9, pp. 755–766, 2013.View at: Publisher Site | Google Scholar
D. W. Lachenmeier, E. Humpfer, F. Fang et al., “NMR-spectroscopy for nontargeted screening and simultaneous quantification of health-relevant compounds in foods: the example of melamine,” Journal of Agricultural and Food Chemistry, vol. 57, no. 16, pp. 7194–7199, 2009.View at: Publisher Site | Google Scholar
G. Le Gall and I. J. Colquhoun, “NMR spectroscopy in food authentication,” in Authenticity, Traceability, M. Lees and B. Popping, Eds., CRC Press LLC, Washington, DC, USA, 1st edition, 2003.View at: Google Scholar
U. Holzgrabe, R. Deubner, C. Schollmayer, and B. Waibel, “Quantitative NMR spectroscopy - Applications in drug analysis,” Journal of Pharmaceutical and Biomedical Analysis, vol. 38, no. 5, pp. 806–812, 2005.View at: Publisher Site | Google Scholar
C. Szńtay Jr., “High-field NMR spectroscopy as an analytical tool for quantitative determinations: Pitfalls, limitations and possible solutions,” Trends in Analytical Chemistry, vol. 11, no. 9, pp. 332–344, 1992.View at: Publisher Site | Google Scholar
Y. B. Monakhova, M. Kohl-Himmelseher, T. Kuballa, and D. W. Lachenmeier, “Determination of the purity of pharmaceutical reference materials by 1H NMR using the standardless PULCON methodology,” Journal of Pharmaceutical and Biomedical Analysis, vol. 100, pp. 381–386, 2014.View at: Publisher Site | Google Scholar
R. D. Farrant, J. C. Hollerton, S. M. Lynn, S. Provera, P. J. Sidebottom, and R. J. Upton, “NMR quantification using an artificial signal,” Magnetic Resonance in Chemistry, vol. 48, no. 10, pp. 753–762, 2010.View at: Publisher Site | Google Scholar
R. Martino, V. Gilard, F. Desmoulin, and M. Malet-Martino, “Interest of fluorine-19 nuclear magnetic resonance spectroscopy in the detection, identification and quantification of metabolites of anticancer and antifungal fluoropyrimidine drugs in human biofluids,” Chemotherapy, vol. 52, no. 5, pp. 215–219, 2006.View at: Publisher Site | Google Scholar
C. Dalvit, P. E. Fagerness, D. T. A. Hadden, R. W. Sarver, and B. J. Stockman, “Fluorine-NMR experiments for high-throughput screening: Theoretical aspects, practical considerations, and range of applicability,” Journal of the American Chemical Society, vol. 125, no. 25, pp. 7696–7703, 2003.View at: Publisher Site | Google Scholar
J. T. Gerig, “Fluorine NMR of proteins,” Progress in Nuclear Magnetic Resonance Spectroscopy, vol. 26, no. 4, pp. 293–370, 1994.View at: Publisher Site | Google Scholar
M. C. Malet-Martino and R. Martino, “Magnetic resonance spectroscopy: a powerful tool for drug metabolism studies,” Biochimie, vol. 74, no. 9-10, pp. 785–800, 1992.View at: Publisher Site | Google Scholar
J. Wang, M. Sánchez-Roselló, J. L. Aceña et al., “Fluorine in pharmaceutical industry: Fluorine-containing drugs introduced to the market in the last decade (2001-2011),” Chemical Reviews, vol. 114, no. 4, pp. 2432–2506, 2014.View at: Publisher Site | Google Scholar
S. S. Saleh, H. M. Lotfy, N. Y. Hassan, and S. M. Elgizawy, “A comparative study of validated spectrophotometric and TLC- spectrodensitometric methods for the determination of sodium cromoglicate and fluorometholone in ophthalmic solution,” Saudi Pharmaceutical Journal, vol. 21, no. 4, pp. 411–421, 2013.View at: Publisher Site | Google Scholar
M. A. Tantawy, N. Y. Hassan, N. A. Elragehy, and M. Abdelkawy, “Simultaneous determination of olanzapine and fluoxetine hydrochloride in capsules by spectrophotometry, TLC-spectrodensitometry and HPLC,” Journal of Advanced Research, vol. 4, no. 2, pp. 173–180, 2013.View at: Publisher Site | Google Scholar
C. Sajeev, P. R. Jadhav, D. RaviShankar, and R. N. Saha, “Determination of flurbiprofen in pharmaceutical formulations by UV spectrophotometry and liquid chromatography,” Analytica Chimica Acta, vol. 463, no. 2, pp. 207–217, 2002.View at: Publisher Site | Google Scholar
H. Y. Aboul-Enein and C. Thiffault, “Rapid liquid chromatographic analysis of flurazepam in pharmaceutical preparation,” Toxicological & Environmental Chemistry, vol. 34, no. 2-4, pp. 73–76, 1992.View at: Publisher Site | Google Scholar
R. I. El-Bagary, M. A. Fouad, M. A. El-Shal, and E. H. Tolba, “Stability-indicating RP-HPLC methods for the determination of fluorometholone in its mixtures with sodium cromoglycate and tetrahydrozoline hydrochloride,” Journal of Chromatographic Science, vol. 54, no. 6, pp. 923–933, 2016.View at: Publisher Site | Google Scholar
S. Esmaeilzadeh, H. Valizadeh, and P. Zakeri-Milani, “A simple, fast, low cost, HPLC/UV validated method for determination of flutamide: Application to protein binding studies,” Advanced Pharmaceutical Bulletin, vol. 6, no. 2, pp. 251–256, 2016.View at: Publisher Site | Google Scholar
M. Tomita and T. Okuyama, “Application of capillary electrophoresis to the simultaneous screening and quantitation of benzodiazepines,” Journal of Chromatography B: Biomedical Applications, vol. 678, no. 2, pp. 331–337, 1996.View at: Publisher Site | Google Scholar
R. J. Ekiert and J. Krzek, “Determination of azole antifungal medicines using zero-order and derivative UV spectrophotometry,” Acta Poloniae Pharmaceutica - Drug Research, vol. 66, no. 1, pp. 19–24, 2009.View at: Google Scholar
A. Álvarez-Lueje, C. Peña, L. J. Núñez-Vergara, and J. A. Squella, “Electrochemical study of flutamide, an anticancer drug, and its polarographic, UV spectrophotometric and HPLC determination in tablets,” Electroanalysis, vol. 10, no. 15, pp. 1043–1051, 1998.View at: Publisher Site | Google Scholar
M. Jakubcova, A. Meddour, J.-M. Péchiné, A. Baklouti, and J. Courtieu, “Measurement of the optical purity of fluorinated compounds using proton decoupled 19F NMR spectroscopy in a chiral liquid crystal solvent,” Journal of Fluorine Chemistry, vol. 86, no. 2, pp. 149–153, 1997.View at: Publisher Site | Google Scholar
U. Holzgrabe, I. Wawer, and B. Diehl, NMR Spectroscopy in Drug Development and Analysis, Wiley Press, Weinheim, Germany, 1999.View at: Publisher Site
G. A. Neville, H. D. Beckstead, D. B. Black, B. A. Dawson, and H. F. Shurvell, “Vibrational and NMR spectroscopic study of aged flurazepam mono‐ and dihydrochloride salts for content identity,” Journal of Pharmaceutical Sciences, vol. 83, no. 9, pp. 1274–1279, 1994.View at: Publisher Site | Google Scholar
B. A. Dawson, D. B. Black, and G. A. Neville, “Multinuclear NMR (1H, 13C and 19F) spectroscopic re-examination of the solvolytic behaviour of flurazepam dihydrochloride,” Journal of Pharmaceutical and Biomedical Analysis, vol. 13, no. 4-5, pp. 395–407, 1995.View at: Publisher Site | Google Scholar
T. Yamazaki, T. Saito, and T. Ihara, “A new approach for accurate quantitative determination using fluorine nuclear magnetic resonance spectroscopy,” Journal of Chemical Metrology, vol. 11, no. 1, pp. 16–22, 2017.View at: Publisher Site | Google Scholar
T. Schönberger, Y. B. Monakhova, D. W. Lachenmeier, and T. Kuballa, “Guide to NMR method development and validation - Part I: Identification and quantification,” Eurolab Technical Report 01/2014, EUROLAB General Secretariat, Brussels, Belgium, 2014.View at: Google Scholar
D. W. Lachenmeier, T. Schönberger, S. Ehni et al., “A discussion about the potentials and pitfalls of quantitative nuclear magnetic resonance (qNMR) spectroscopy in food science and beyond,” in Proceedings of the XIII International Conference on the Applications of Magnetic Resonance in Food Science, vol. 16, pp. 77–85, 2016.View at: Google Scholar