Research Article | Open Access
Hayam M. Lotfy, Sarah S. Saleh, "Investigating Advanced Approaches Based on Iso-Absorptivity Coefficient in Unresolved Spectral Signals of Binary Mixtures", Journal of Analytical Methods in Chemistry, vol. 2019, Article ID 7924821, 10 pages, 2019. https://doi.org/10.1155/2019/7924821
Investigating Advanced Approaches Based on Iso-Absorptivity Coefficient in Unresolved Spectral Signals of Binary Mixtures
Several spectrophotometric approaches utilize different functions of the iso-absorptivity coefficient in zero-order absorption signals and its manipulated spectra. This work introduced an investigation concerning the efficiency power of recent methods based on iso-absorptivity coefficient in different spectral signals. These methods were as follows: absorptivity centering method (a-Centering), absorbance subtraction method (AS), amplitude modulation method (AM,) and amplitude summation method (A-Sum). These methods were applied to determine the binary mixture of ofloxacin (OFX) and dexamethasone (DXM). Linearity of the proposed methods was investigated in the range of 1.0–10.0 μg/ml for both drugs. The proposed methods were validated as per ICH guidelines and were successfully applied for the simultaneous determination of OFX and DXM in their pharmaceutical preparation without interference from additives. Statistical analysis of the results obtained by the proposed spectrophotometric methods compared with a reported method revealed no significant difference between the proposed and reported methods, confirming accuracy and precision at 95% confidence limit.
Through this investigation, the proposed methods were applied to the binary mixture of ofloxacin (OFX) and dexamethasone (DXM) present in eye drop preparation. Ofloxacin (OFX) [9-fluoro-2, 3-dihydro-3-methyl-10-(4-methyl-1-piperazinyl)-7-oxo-7H-pyrido [1,2,3-de]-1, 4-benzoxazine-6-carboxylic acid] is a fluoroquinolone antibacterial agent which is active against wide range of bacteria, whereas dexamethasone (DXM) [9α-fluoro-11β, 17α, 21-trihydroxy-16α-methylpregna-1, 4-diene-3, 20-dione] is a corticosteroid with potent anti-inflammatory and immunosuppressive effects [1, 2]. By reviewing literature, several methods reported were reported for the determination of OFX, such as titrimetry , spectrophotometry [4, 5], and chromatography [6, 7]. For DXM, reported methods were found, such as spectrophotometry [8, 9], micellar liquid chromatography , HPLC [11, 12], and capillary electrophoresis . Simultaneous determination of ofloxacin and dexamethasone phosphate has been reported in the literature by spectrophotometric methods [14, 15] and chromatographic methods [16, 17].
The resolution and analysis of mixtures is an everyday challenge for analysts. Researchers in the analytical field have to come up with new ideas to solve different problems faced by analysts, such as the availability of apparatus, sensitivity, selectivity, cost, environmental aspects, and so on. Spectrophotometry has many advantages for analysts, as they are easy to operate, of low cost, available, and environment-friendly. Fortunately, the last era showed a remarkable development in the invention and application of smart spectrophotometric solutions. In this work, investigation of the efficiency power of recently introduced spectrophotometric methods, which are based on different functions of the absorptivity coefficient, was done.
The term “isosbestic point” describes the intersection point of spectra of prepared mixtures to obtain the sum of the concentrations of the absorbing components in the binary mixture. The iso-absorptive point is defined as the isosbestic point in zero-order absorption spectra where the two absorbing components, X and Y, have equal absorptivity coefficient [18, 19]. The iso-absorptive point of an absorption spectrum will be retained at the same point in the corresponding ratio spectrum, but it will be shifted to a different adjacent point in the corresponding derivative spectrum .
The function of iso-absorptive points has been utilized for different purposes, such as the estimation of a precursor fraction in a complex or conversion reaction [21–23], the direct calculation of concentration of single component using official methods , the study of dissolution rate of certain drugs with no inference of excipients , and determining the concentrations of several components in pharmaceutical mixtures [20, 25–27].
The aim of the work was to validate the four proposed methods via ICH guidelines, as well as to investigate the efficiency power of four spectrophotometric methods based on iso-absorptivity coefficient point and its manipulated spectra either ratio, using the interfering substance as a divisor or derivative. Absorptivity centering method (a-Centering) and absorbance subtraction method (AS) were applied to zero-order spectra, whereas amplitude modulation method (AM) was applied to ratio spectra and amplitude summation method (A-Sum) was applied to derivative spectra.
1.1. Theoretical Background of Absorptivity Centering Method (a-Centering)
This novel method  is applied to a mixture of two components (X and Y) having partially or completely overlapped spectra and showing an intersection at the iso-absorptive point (λiso). For the binary mixture of X and Y showing overlapped spectra (λiso) and X showing no contribution at (λ2), this method can be explained through the following five steps.
1.1.1. Calculating the Absorbance of Y at (λiso) via Absorptivity Factor
X and Y show overlapped spectra at (λiso), whereas Y exhibits absorbance at (λ2), with no contribution of X. So, the absorptivity factor is calculated by selecting these two wavelengths (λiso and λ2) at zero-order absorption spectra (D0) of different concentrations of pure Y; then, the average value is calculated as follows:
By multiplying the absorbance (), which is the absorbance of the mixture of (X + Y) at λ2, by the previously calculated absorptivity factor , the absorbance of Y in the mixture at λiso  is calculated as shown by the following equation:
1.1.2. Calculating the Absorptivity Inverse at Iso-Absorptive Point
The zero-order absorption spectra (D0) of different concentrations of pure Y at the selected wavelength (λiso) are recorded. Each concentration (C) is divided by its corresponding absorbance at the iso-absorptive point , and then, the average of these values is calculated. This can be summarized by the following equations:
At iso-absorptive point (λiso), ; thus,
1.1.3. Preparation of Normalized and Factorized Spectra of Y
The normalized spectrum (NS′) represents the absorptivity values of Y over the selected wavelength range. It is prepared by dividing each of the zero-order absorption spectra (D0) of Y by its corresponding concentration, and the average value is recorded:
The factorized spectrum (FS′) represents a spectrum with unity absorbance at the iso-absorptive point . It is prepared by dividing each of the zero-order absorption spectra (D0) of Y by its corresponding absorbance at the iso-absorptive point , and the average value is recorded:
1.1.4. Spectrum Recovery of Y via Normalized and Factorized Spectra
To recover the zero-order absorption spectrum (D0) of Y via the normalized spectrum (NS′), the absorbance of Y at λiso is multiplied by the calculated absorptivity inverse ; then, the result is multiplied by the normalized spectrum of Y (aY):
On the other hand, the absorbance of Y at λiso is directly multiplied by the factorized spectrum (FS′) to recover the zero-order absorption spectrum (D0) of Y:
1.1.5. Spectrum Subtraction to Recover X Spectrum
Samir et al.  previously introduced this complementary step. By subtracting the previously obtained D0 spectra of Y from the spectra of the mixtures (X + Y), the D0 spectra of X will be recovered. The concentration of X and Y is calculated from the corresponding regression equations obtained by plotting the absorbance values of the zero-order curves of X or Y at its λmax against the corresponding concentrations.
2.1. Instrument and Chemicals
Spectrophotometer UV-Visible double beam (Shimadzu - UV 1800, Nakagyo-ku, Kyoto, Japan) with matched 1 cm quartz cells operated by Shimadzu UV-Probe 2.32 system software. A solvent mixture was prepared using ethanol HPLC grade (S D Fine-Chem Limited, Mumbai) and distilled water in the ratio of 50 : 50 v/v.
2.2. Standard Solutions
Pure samples of ofloxacin (OFX), dexamethasone (DXM), and benzalkonium chloride (BNZ) were kindly supplied by Egyptian International Pharmaceutical Industries Co. (EIPICO), 10th Ramadan city, Cairo, Egypt. The purity was checked by the official methods , and it was found to be 100.33 ± 0.73 and 100.51 ± 0.83 for OFX and DXM, respectively. Dexaflox® eye drops (batch number: LB060) was manufactured by Jamjoom Pharma, Jeddah, Kingdom of Saudi Arabia (KSA). Each 1 ml is labeled to contain OFX (3 mg), DXM (1 mg), and BNZ (0.06 mg). Stock standard solutions of OFX, DXM, and BNZ were prepared of concentration 1.0 mg/ml, and further dilution was done using the previously prepared solvent mixture to reach a concentration of 20.0 μg/ml. of each component.
2.3. Spectral Manipulations
Solutions equivalent to 1.0–10.0 μg/mL of OFX and DXM were prepared separately in the solvent mixture, and the absorption spectra of the prepared solutions were measured at (200–400 nm) and stored in the computer. The calibration curves were constructed by plotting the absorbance of the zero-order absorption spectra (D0) of OFX and DXM at 298 and 239 nm, respectively, against the corresponding concentrations, and the linear regression equations were computed. The absorptivity factor of OFX was calculated at the wavelengths [a238/a298] by dividing the absorbance of different concentrations of pure OFX at 238 nm/298 nm. The normalized spectrum (NS′) is prepared by dividing difference (D0) of OFX by its corresponding concentration. The factorized spectrum (FS′) is prepared by dividing the difference (D0) of OFX by its corresponding absorbance at 238 nm. The absorptivity inverse at iso-absorptive point (1/a238nm) for OFX is calculated by dividing the zero-order absorption spectra (D0) of OFX by its normalized spectrum (NS′) to get the spectrum representing the concentration; then, the obtained spectrum is divided by its corresponding zero-order spectrum to get that spectrum absorptivity inverse (1/a) all over the wavelength range. Thus, the recorded value at the iso-absorptive point (238 nm) is equal to absorptivity inverse (1/a238nm). The average values for the constants were calculated. The stored absorption spectra of OFX and DXM were separately divided by the normalized OFX spectrum, and the obtained ratio spectra were recorded. Calibration curves were constructed by plotting both of their amplitudes at λiso 238.0 nm against their corresponding concentrations, and the regression equations were computed. Calibration curves were constructed at λmax for both drugs at 298.0 nm and 239.5 nm for OFX and DXM, respectively, against their corresponding concentrations, and the regression equations were computed. The zero-order absorption spectra were derivatized in first order using ∆λ8 and scaling factor 10 and the first derivative amplitudes D1 were recorded. The calibration curves were constructed by plotting the amplitudes of OFX and DXM at 246.5 nm against their corresponding concentrations, and the regression equations were computed. The amplitude factor of OFX was calculated at these selected wavelengths [D246.5/D307] by dividing the D1 amplitudes of different concentration of pure OFX at 246.5 nm/307 nm, and the average value of each was calculated.
2.4. Application to Synthetic Mixtures and Pharmaceutical Preparation
Into a series of 10 mL volumetric flasks, accurate aliquots of OFX and DXM were transferred from their working solutions (20.0 μg/ml) to prepare nine mixtures containing different ratios of the cited drugs. Aliquot equivalent to 1 μg/mL of BNZ is added to each prepared mixture. The volumes were completed with the solvent mixture. The spectra of the prepared solutions from 200 to 400 nm were recorded and stored. One millimeter of Dexaflox® eye drops was transferred into 50 mL volumetric flask, and the prepared solution was filtered through 0.45 μm Millipore syringe membrane filter. An appropriate dilution was made with the same solvent mixture to prepare the working solution claimed to contain 9.0 μg/ml OFX and 3.0 μg/ml DXM. Six replicates for each experiment were done. The concentrations of pure OFX and DXM were calculated using their corresponding regression equations.
3. Results and Discussion
These methods were applied to the binary mixture of ofloxacin (OFX) and dexamethasone (DXM). The zero-order spectra (D0) and chemical structures for the cited drugs are shown in Figure 1, where OFX and DXM show overlapped spectra at (λiso 238 nm), but OFX exhibits absorbance at 298 nm solely with no interference with DXM. The solvent mixture was prepared in that ratio in order to reduce the use of organic solvent and cost and have less impact on the environment.
3.1. Absorptivity Centering Method (a-Centering)
This method starts with the calculation of two constants for OFX: the absorptivity factor [a238/a298] and the absorptivity inverse at the iso-absorptive point (), which were found to be equal to 0.339 and 25.833, respectively. The absorbance of OFX in mixtures’ spectra at the iso-absorptive point λiso “A(238nm)” was calculated by multiplying the absorbance of OFX at 298.0 nm by the calculated absorptivity factor [a238/a298], as shown in the equation:
Two approaches could be adopted in order to recover the zero-order absorption spectra D0 of OFX from the D0 spectra of the mixtures. The first approach was done by multiplying the calculated absorbance of OFX (A238) by the calculated absorptivity inverse (), and then the resulting spectrum is multiplied by the normalized spectrum (NS′) of OFX, as shown in Figure 2. The second approach was done by multiplying the calculated absorbance of OFX (A238) by the prepared factorized spectrum (FS′) of OFX directly, as shown in Figure 2.
The zero-order absorption spectra D0 of DXM can be recovered by subtracting the recovered spectrum of OFX from the from the mixtures’ spectra. The concentration of OFX and DXM is calculated from the corresponding regression equations at their λmax 298.0 and 239.5 nm, respectively, as shown in Figure 1.
By putting this method in hand, it was found that the approach via factorized spectrum (FS′) is much simpler than that of the normalized spectrum (NS′). In spite of the several preparatory steps involved in this method (especially using the NS′ approach), its advantage not only lies in the calculation of each component’s concentration at its λmax, which ensures its accuracy and sensitivity, but also in the simplicity of recovering the whole spectrum of each component, which acts as a spectral profile for it, and can be further used in testing its purity .
3.2. Absorbance Subtraction Method (AS)
The binary mixture of OFX and DXM exhibits two iso-absorptive points (λiso) at 238 and 265 nm where the components exhibiting this point have equal absorptivity coefficient, as shown in Figure 3. The calculation was done at the first point (λiso) 238 nm because it has higher absorbance signal which increases accuracy and sensitivity of the results. The absorbance of OFX at λiso “A(238nm)” was calculated using the calculated absorptivity factor  as explained before in the absorptivity centering method; then, the absorbance of DXM ADXM(238) was obtained by subtracting the absorbance of OFX from the absorbance of the mixture Amix(238). The absorbance values of OFX and DXM at λiso 238 nm were used to calculate each of their concentration using the unified regression equation at the same wavelength.
This method exceeds in simplicity as it does not require any manipulating steps except for the factor calculation and both components can be determined using unified regression equation at λiso. The disadvantage of this method is measuring the absorbance at wavelength rather than λmax, which increases the risk of error when calculating low concentrations, in addition to the inability of testing purity by extracting the whole spectrum (as in absorptivity centering method).
3.3. Amplitude Modulation Method (AM)
This manipulating ratio-spectra technique depends on the fact that the iso-absorptive point in a zero-order absorption spectrum (D0) will be retained in the ratio spectrum at the same point. The zero-order absorption spectrum of OFX is extended over that of DXM. To eliminate the effect of the divisor choice, the normalized spectra (NS′) of OFX was used as a divisor. When the D0 spectrum of the binary mixture was divided by the (NS′) OFX divisor, a ratio spectrum with a constant amplitude value at the plateau region at (300–325 nm) was obtained, as shown in Figure 4. The recorded constant amplitude at this plateau region was modulated to be directly equal to the concentration of OFX (CRecorded of OFX):where Pmix is the P amplitude value of the binary mixture, POFX and PDXM are the ratio amplitudes of OFX and DXM, respectively, “a” is the absorptivity coefficient, and “C” is the concentration.
On the other hand, the amplitude at the λiso at 238 nm was equal to the sum of the concentrations of OFX and DXM. By subtracting the previously obtained constant from the recorded amplitude at λiso, the obtained amplitude was directly modulated to the concentration of DXM in the mixture (CRecorded of DXM):
One regression equation is constructed for both drugs at the iso-absorptive point (λiso 238 nm), where its slope is almost one and its intercept is almost zero. The equation is used to eliminate the signal to noise ratio for both OFX and DXM.
The amplitude modulation (AM) method excels over other manipulating ratio-spectra techniques by reducing the manipulation steps and elimination of the choice of the divisor. Only one divisor is needed to determine both components in the mixture and directly modulate the amplitudes into concentrations.
3.4. Amplitude Summation Method (A-Sum)
The spectrum of OFX and DXM mixture showed one isosbestic point (λiso) only at 246.5 nm, as shown in Figure 5, where OFX shows overlap with DXM, whereas at 307.0 nm, DXM shows no contribution. The D1 amplitude of OFX in the mixtures at (λiso) 246.5 nm was calculated by multiplying the D1 amplitude of the mixture at 307.0 nm by its amplitude factor [D246.5/D307]. Then, the amplitude of DXM at (λiso) 246.5 nm in the mixtures was obtained by subtracting the calculated amplitude of OFX at the same point. The D1 amplitude of OFX and DXM was used to calculate each of their concentration using the unified regression equation at 246.5 nm:where Dmix is the D1 amplitude value of the binary mixture, DOFX and DDXM are the D1 amplitudes of OFX and DXM, respectively, and  is the amplitude factor of OFX.
Few differences were observed when applying this method: The shift that occurred for the isosbestic point (from 238.0 to 246.5 nm) is due to derivatization manipulation Δλ = 8 and scaling factor 10. The main advantage in this method is that there is no need for a divisor and the sensitivity of the method is enhanced when compared with zero-order absorption spectra. In addition, the shift that occurred in the iso-point gives an opportunity for more complex mixtures to correct any interference of any component that may possesses a zero derivative value at the isosbestic point.
A brief comparison between the four methods, including the strength and weakness points, is summarized in Table 1. Overall, the absorptivity centering method (a-Centering) via normalized spectrum involves several manipulating steps, so it is more convenient when applied via factorized spectrum. Both approaches are recommended to be used in case of recovering the full spectrum of each component with maximum accuracy and precision, as measurements are done at λmax of each component. On the other hand, the three remaining methods determine both components via unified regression equation. The absorbance subtraction (AS) method is considered to be the simplest method with minimum manipulation but suffers from inaccuracy in case of low concentration of the extended drug which obstacles the calculation of the absorbance factor. The amplitude modulation (AM) method is considered to be the only method which can directly modulate amplitudes into concentrations without need for standard calibration curves. Finally, the amplitude summation (A-Sum) method becomes of use when shifting the isosbestic point is needed to eliminate interference from excipients or other components.
NS′: normalized spectrum; FS′: factorized spectrum.
3.5. Validation, System Suitability, and Statistical Analysis
The validation parameters for the proposed methods were applied as per the ICH guidelines , including linearity, range, accuracy, precision, and system suitability. Linear regression equations were constructed by plotting the concentration (within the selected range) as an independent variable (X) versus the corresponding absorbance as an independent variable (Y) to apply a best fit straight regression line. The precision study included the triplicate measurements of three concentrations of each drug which were done on the same day (repeatability) and on three successive days (interday precision). Satisfactory results were obtained for all validation parameters, as shown in Table 2. The accuracy of the proposed methods was assessed by the analysis of synthetic mixtures containing different ratios of the cited drugs, as shown in Table 3.
aAverage of three experiments. bMean ± standard deviation of 3 concentrations of each drug (4, 6, and 8 µg/mL). cStandard deviation/relative standard deviation of 3 concentrations of each drug (4, 6, and 8 µg/mL). dRobustness was checked by testing the effect of the solvent (29, 31, and 32% methanol). eOFX and DXM (6 µg/mL).
aAverage of three experiments. bRatio present in Dexaflox® eye drops.
System suitability  was measured for standard solutions containing 100% working concentration for six replicates, and the %RSD was calculated, as shown in Table 2. The proposed methods were applied for the determination of Dexaflox® eye drops and the validity of the proposed procedures is further assessed by comparing it with a reported method . The results obtained are shown in Table 4.
aOFX claimed to be 9 µg/mL, DXM: 3 µg/mL, average of six experiments. bTabulated value (5.050) at .
Statistical analysis testing (one-way ANOVA) was applied for the purpose of comparison between the recovery percent obtained by the analysis of the synthetic mixtures by the proposed methods; Table 5 shows that there was no significant difference between them.
aThere was no significant difference between the methods using one-way ANOVA at .
This work introduced an investigation concerning the efficiency of four methods utilizing different functions of absorptivity coefficient in different absorption spectra: absorptivity centering method (a-Centering), absorbance subtraction method (AS), amplitude modulation method (AM), and amplitude summation method (A-Sum). In conclusion, the absorptivity centering method was found to have the optimum efficiency power to resolve the unresolved spectra and recovering of their zero-order absorption spectra which acts as their own fingerprint and reflect the purity of the drugs. Thus, OFX and DXM can be directly determined by the suggested methods in pure and pharmaceutical preparations without introductory separation. Additionally, these methods do not need sophisticated techniques, instruments, or high-priced solvents or a harmful organic solvent. Consequently, they can be used for the routine analysis of cited drugs in laboratories lacking liquid chromatographic instruments.
The data used to support the findings of this study are included within the article.
Conflicts of Interest
The authors declare that there are no conflicts of interest.
- S. C. Sweetman, Martindale: The Complete Drug Reference, Pharmaceutical Press, London, UK, 2005.
- A. C. Moffat, D. M. Osselton, B. Widdop, and L. Y. Galichet, Clarke’s Analysis of Drugs and Poisons, Pharmaceutical Press, London, UK, 2005.
- The Stationery Office on behalf of the Medicines and Healthcare products Regulatory Agency (MHRA), vol. II, British Pharmacopoeia, London, UK, 2009.
- Y. Tu and L. Liu, “The determination of ofloxacin by fluorescence spectrophotometry,” Spectroscopy and Spectral Analysis, vol. 20, no. 6, pp. 880–882, 2000.
- S. Bindaiya, S. Bankey, and D. Jain, “Simultaneous determination of nitazoxanide and ofloxacin in tablet by ultraviolet spectrophotometry (dual wavelength method),” International Journal of ChemTech Research, vol. 2, pp. 11–15, 2010.
- M. S. Ali, M. Ghori, and A. Saeed, “Simultaneous determination of ofloxacin, tetrahydrozoline hydrochloride, and prednisolone acetate by high-performance liquid chromatography,” Journal of Chromatographic Science, vol. 40, no. 8, pp. 429–433, 2002.
- P. Katakam and K. R. Sireesha, “Simultaneous determination of ciprofloxacin hydrochloride and dexamethasone sodium phosphate in eye drops by HPLC,” E-Journal of Chemistry, vol. 9, no. 3, pp. 1077–1084, 2012.
- M. S. Collado, J. C. Robles, M. De Zan, M. S. Cámara, V. E. Mantovani, and H. C. Goicoechea, “Determination of dexamethasone and two excipients (creatinine and propylparaben) in injections by using UV-spectroscopy and multivariate calibrations,” International Journal of Pharmaceutics, vol. 229, no. 1-2, pp. 205–211, 2001.
- M. A. Hegazy, H. M. Lotfy, M. R. Rezk, and Y. R. Omran, “Novel spectrophotometric determination of chloramphenicol and dexamethasone in the presence of non labeled interfering substances using univariate methods and multivariate regression model updating,” Spectrochimica Acta Part A: Molecular and Biomolecular Spectroscopy, vol. 140, pp. 600–613, 2015.
- D. Pena-Garcıa-Brioles, R. Gonzalo-Lumbreras, R. Izquierdo-Hornillos, and A. Santos-Montes, “Method development for betamethasone and dexamethasone by micellar liquid chromatography using cetyl trimethyl ammonium bromide and validation in tablets: application to cocktails,” Journal of Pharmaceutical and Biomedical Analysis, vol. 36, no. 1, pp. 65–71, 2004.
- Y. Seid, A. Hymete, and A. A. Bekhit, “Application of a stability-indicating HPTLC method for simultaneous determination of chloramphenicol and dexamethasone sodium phosphate in eye drop,” Thai Journal of Pharmaceutical Sciences, vol. 36, pp. 94–107, 2012.
- L. Zivanovic, M. Zecevic, S. Markovic, S. Petrovic, and I. Ivanovic, “Validation of liquid chromatographic method for analysis of lidocaine hydrochloride, dexamethasone acetate, calcium dobesilate, buthylhydroxyanisol and degradation product hydroquinone in suppositories and ointment,” Journal of Chromatography A, vol. 1088, no. 1-2, pp. 182–186, 2005.
- V. Baeyens, E. Varesio, J. L. Veuthey, and R. Gurny, “Determination of dexamethasone in tears by capillary electrophoresis,” Journal of Chromatography B: Biomedical Sciences and Applications, vol. 692, no. 1, pp. 222–226, 1997.
- N. Y. Hassan, S. M. Elgizawy, H. M. Lotfy, and S. S. Saleh, “Comparative study of spectrophotometric methods versus chemometric methods; an application on a pharmaceutical binary mixture of ofloxacin and dexamethasone,” International Research Journal of Pure and Applied Chemistry, vol. 3, no. 2, pp. 90–110, 2013.
- M. Nebsen, G. M. Elsayed, M. AbdelKawy, and S. Z. Elkhateeb, “Determination of ofloxacin and dexamethasone in Dexaflox eye drops through different ratio spectra manipulating methods,” Bulletin of Faculty of Pharmacy, Cairo University, vol. 51, no. 2, pp. 175–184, 2013.
- H. A. Ahmed, B. A. Moussa, R. I. El-Bagary, and M. N. Darwish, “Three validated methods for simultaneous determination of ofloxacin and dexamethasone in binary mixture,” Journal of Planar Chromatography—Modern TLC, vol. 26, no. 1, pp. 56–61, 2013.
- K. R. Sireesha and K. Prakash, “HPLC-UV method for simultaneous determination of ofloxacin and dexamethasone sodium phosphate,” International Journal of Pharmacy and Pharmaceutical Sciences, vol. 4, pp. 415–418, 2012.
- S. V. Erram and H. P. Tipnis, “Simple spectrometric analysis of propranolol hydrochloride and hydrochlorothiazide from combined pharmaceutical dosages,” Indian Drugs, vol. 31, pp. 65–68, 1994.
- S. V. Erram and H. P. Tipnis, “Simple spectrometric analysis of frusemide and amiloride hydrochloride from combined pharmaceutical dosages,” Indian Drugs, vol. 30, no. 8, pp. 371–376, 1993.
- H. M. Lotfy, S. S. Saleh, N. Y. Hassan, and H. Salem, “A comparative study of novel spectrophotometric methods based on isosbestic points; application on a pharmaceutical ternary mixture,” Spectrochimica Acta Part A: Molecular and Biomolecular Spectroscopy, vol. 126, pp. 112–121, 2014.
- B. S. Berlett, R. L. Levine, and E. R. Stadtman, “Use of isosbestic point wavelength shifts to estimate the fraction of a precursor that is converted to a given product,” Analytical Biochemistry, vol. 287, no. 2, pp. 329–333, 2000.
- J. Zhou, H. Z. Yin, X. J. Qu, and Y. H. Shun, “Study of coordination equilibria in systems of coexisting mononuclear and polynuclear complexes by an isosbestic point -spectrophotometric method,” Talanta, vol. 43, no. 11, pp. 1863–1867, 1996.
- N. N. Kazantseva, A. Ernepesova, A. Khodjamamedov, O. A. Geldyev, and B. S. Krumgalz, “Spectrophotometric analysis of iodide oxidation by chlorine in highly mineralized solutions,” Analytica Chimica Acta, vol. 456, no. 1, pp. 105–119, 2002.
- A. M. Wahbi, E. M. Hassan, and A. A. Gazy, “Use of isosbestic points of derivative absorption curves: the dissolution rate determination of Aspirin,” Journal of Pharmaceutical and Biomedical Analysis, vol. 12, no. 7, pp. 951–954, 1994.
- M. G. El-Bardicy, H. M. Lotfy, M. A. El-Sayed, and M. F. El-Tarras, “Smart stability-indicating spectrophotometric methods for determination of binary mixtures without prior separation,” Journal of AOAC International, vol. 91, no. 2, pp. 299–310, 2008.
- H. M. Lotfy, “Absorbance subtraction and amplitude modulation as novel spectrophotometric methods for the analysis of binary mixtures,” International Journal of Pharmacy and Pharmaceutical Sciences, vol. 6, pp. 735–741, 2014.
- H. M. Lotfy, M. A. Hegazy, M. R. Rezk, and Y. R. Omran, “Comparative study of novel versus conventional two-wavelength spectrophotometric methods for analysis of spectrally overlapping binary mixture,” Spectrochimica Acta Part A: Molecular and Biomolecular Spectroscopy, vol. 148, pp. 328–337, 2015.
- H. M. Lotfy and Y. R. Omran, “Novel absorptivity centering method utilizing normalized and factorized spectra for analysis of mixtures with overlapping spectra in different matrices using built-in spectrophotometer software,” Spectrochimica Acta Part A: Molecular and Biomolecular Spectroscopy, vol. 200, pp. 167–178, 2018.
- A. Samir, H. M. Lotfy, H. Salem, and M. Abdelkawy, “Development and validation of simultaneous spectrophotometric and TLC-spectrodensitometric methods for determination of beclomethasone dipropionate and salbutamol in combined dosage form,” Spectrochimica Acta Part A: Molecular and Biomolecular Spectroscopy, vol. 128, pp. 127–136, 2014.
- H. M. Lotfy and S. S. Saleh, “Testing the purity of spectral profiles: finger-print resolution of complex matrices and extraction of absorbance signals,” Spectrochimica Acta Part A: Molecular and Biomolecular Spectroscopy, vol. 205, pp. 160–169, 2018.
- Q2b Icohiqa, “Validation of analytical procedures: Methodology,” Federal Register, vol. 62, no. 96, pp. 27463–27467, 1997.
- R. S. Nijhu, D. T. Akhter, and Y. M. Jhanker, “Development and validation of UV spectrophotometric method for quantitative estimation of nitroglycerin in pharmaceutical dosage form,” International Current Pharmaceutical Journal, vol. 1, no. 1, pp. 1–5, 2011.
Copyright © 2019 Hayam M. Lotfy and Sarah S. Saleh. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.