- About this Journal ·
- Abstracting and Indexing ·
- Aims and Scope ·
- Annual Issues ·
- Article Processing Charges ·
- Articles in Press ·
- Author Guidelines ·
- Bibliographic Information ·
- Citations to this Journal ·
- Contact Information ·
- Editorial Board ·
- Editorial Workflow ·
- Free eTOC Alerts ·
- Publication Ethics ·
- Reviewers Acknowledgment ·
- Submit a Manuscript ·
- Subscription Information ·
- Table of Contents
BioMed Research International
Volume 2013 (2013), Article ID 289295, 8 pages
Microsphere Suspension Array Assays for Detection and Differentiation of Hendra and Nipah Viruses
Australian Animal Health Laboratory, CSIRO Animal, Food and Health Sciences, Geelong, VIC 3220, Australia
Received 19 October 2012; Accepted 25 December 2012
Academic Editor: Arun K. Bhunia
Copyright © 2013 Adam J. Foord et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Microsphere suspension array systems enable the simultaneous fluorescent identification of multiple separate nucleotide targets in a single reaction. We have utilized commercially available oligo-tagged microspheres (Luminex MagPlex-TAG) to construct and evaluate multiplexed assays for the detection and differentiation of Hendra virus (HeV) and Nipah virus (NiV). Both these agents are bat-borne zoonotic paramyxoviruses of increasing concern for veterinary and human health. Assays were developed targeting multiple sites within the nucleoprotein (N) and phosphoprotein (P) encoding genes. The relative specificities and sensitivities of the assays were determined using reference isolates of each virus type, samples from experimentally infected horses, and archival veterinary diagnostic submissions. Results were assessed in direct comparison with an established qPCR. The microsphere array assays achieved unequivocal differentiation of HeV and NiV and the sensitivity of HeV detection was comparable to qPCR, indicating high analytical and diagnostic specificity and sensitivity.
Bats harbour a wide range of viruses that have been implicated in spill over events into other mammalian hosts resulting in highly virulent and often fatal zoonoses. Henipa-, filo-, lyssa-, and coronaviruses are some of the most notable examples . The henipaviruses HeV and NiV are bat-borne paramyxoviruses which have been responsible for severe disease outbreaks in humans, horses, and pigs . HeV was first identified in Australia in 1994 as the cause of fatal infection in horses and humans . The closely related NiV was subsequently identified as the causative agent of infections in pigs and humans in Malaysia in 1998-99 . The fruit bat (Pteropus spp.) is the only known natural reservoir of these two viruses. NiV infections in humans have been identified in several countries including Malaysia, Singapore, Bangladesh, and India with mortality up to and exceeding 75% in some of these epidemics . Evidence for this virus or henipa-like viruses in bat populations in other South-East Asian locations have also been provided [5–7]. Henipa-like genomic sequences have also been detected in African bats  and Cedar paramyxovirus (CedPV), a novel henipa-like virus, was recently isolated in Australia . HeV is endemic in Australian Pteropus bats and can spread directly from bats to horses, causing severe disease. Human HeV infection has so far only resulted from close contact with the blood, body fluids, and tissues of infected horses. Although bats appear to be unaffected by HeV there is a high case-fatality rate in both humans and horses and spill-over events from bats to horses are occurring with increasing regularity [10, 11]. The wide range of viruses and their enormous genome sequence variation and evolution pose a challenge to the development of molecular diagnostic assays. Although next generation sequencing can identify viruses without any prior knowledge of their sequence , this approach is still not practical for screening larger numbers of samples in a diagnostic context. Various combinations of conventional PCR and sequencing, or qPCR, have been used for virus identification [12, 13]. Highly conserved genes and sequences within, or across, virus species in combination with degenerate PCR primer sequences have broadened the range of viruses detectable by a single PCR . The nature of these generic PCR assays necessitates the use of highly degenerate primers which can lead to a reduction in sensitivity and still requires confirmation of any resulting PCR products by DNA sequencing. Nevertheless, qPCR is highly specific, sensitive, and suitable for automation and screening of large sample numbers. However, the limited multiplexing capability of qPCR, typically no more than 2-3 combined assays, requires the setup of various and frequently, constituently different qPCR reactions when screening for multiple viruses. Microsphere suspension array assays offer advantages over qPCR in the level of readily achievable multiplexing. This allows for the simultaneous screening of many targets (up to 100 markers in the Luminex system) in a single reaction and has become a valuable tool for investigation of disease syndromes. Various assay panels for nucleic acid detection have been developed for medical or veterinary applications, including respiratory viral diseases , gastroenteritis pathogens , cystic fibrosis , biothreat agents [18, 19], and vesicular diseases of livestock . Polymerase chain reaction amplification of the regions of interest forms the first step of these assays. Proprietary polystyrene microspheres that contain dyes displaying distinct spectral characteristics form the substrate for these assays. Luminex MagPlex-TAG microspheres (Luminex Corporation) contain unique 24 nucleotide DNA “antiTAG” sequences covalently coupled to their surface. This facilitates hybridization of specifically amplified and labeled products containing complementary “TAG” sequences and allows identification by association with particular microsphere sets in a flow cytometry-based detection system. Microsphere suspension array assays can be designed as a modular system and combined into assay panels of increasing complexity to increase the range for detection of different viruses.
Here we report the development of microsphere array assays for detection and differentiation of RNA from HeV and NiV isolates and their analytical and diagnostic performance. Furthermore, we demonstrated the utility of these assays as modules for detection of HeV in Australian horses. Our intention is to include these assays in future development of multiplex assay panels for investigating infectious diseases, syndromes in livestock, and zoonoses.
2. Materials and Methods
2.1. Viruses and Diagnostic Samples
Viruses used in this study were HeV (hendra virus/Australia/horse/Hendra/1994, six HeV isolates from four of the 2011 outbreaks in Queensland and New South Wales, Australia); NiV (Nipah virus/Malaysia/1998, Nipah virus/Bangladesh/2004); Cedar paramyxovirus; nonrelated paramyxovirus (Tioman, Sendai, Menangle, Rinderpest, and J virus); other viruses (West Nile, Kunjin, Murray valley encephalitis, and Japanese encephalitis virus). All viruses were held as stocks at the Australian Animal Health Laboratory (AAHL). Diagnostic samples used in this study included blood, tissue, and swabs from horses submitted to AAHL during several outbreaks of HeV in the Australian States of Queensland and New South Wales in 2011. Further samples from time-course experimental studies on a limited number of HeV-infected horses were also used . To assist with an appropriate statistical analysis of the collective data, swab samples from submissions of quarantined European horses, presumed to be negative for HeV, were also utilized. Nucleic acid was isolated from each sample using the MagMAX 96 Viral RNA Extraction Kit (Life Technologies Cat. No. AM1836-5).
2.2. Microsphere Suspension Array Primer Design
The primer designs (Table 1) were based on the alignments of the available 17 HeV sequences (GenBank accession numbers: HM044317, HM044318, HM044319, HM044320, HM044321, JN255800, JN255801, JN255802, JN255803, JN255804, JN255805, JN255806, JN255812, JN255814, JN255817, JN255818, NC001906) and 20 NiV sequences (GenBank accession numbers AF376747, AJ564621, AJ564622, AJ564623, AJ627196, AY029767, AY029768, AY858110, AY988601, FJ513078, FN869553, JN808857, JN808858, JN808859, JN808860, JN808861, JN808862, JN808863, JN808864, NC002728) using Geneious Pro software . Primers were designed for two independent assays specific for either a 375 nucleotide region of the N-gene coding sequence (CDS) or 97 nucleotide region of the P-gene CDS of HeV and NiV (Figure 1). The N-gene assay accommodated three target-specific primer extension (TSPE) primers, one for generic detection of both HeV and NiV, and one specific for HeV or NiV only. The P-gene assay based on a qPCR assay  contains only a single TSPE primer for detection of both HeV and NiV. The incorporation of degenerate nucleotides was utilized to facilitate generic amplification and detection of all known HeV and NiV isolates. The design of two independent assays served as a contingency in the event of diagnostic failure should virus sequences change in an individual assay target region.
2.3. Microsphere Suspension Array Assay Procedure
2.3.1. Primary PCR
Single-step reverse transcription PCR (RT-PCR) was performed using Superscript III One-Step RT-PCR with Platinum Taq kit (Invitrogen) with the following conditions: 25 μL volume, 200 nM forward and reverse primers, and 2.0 mM MgSO4. Thermal cycling conditions were 30 min at 48°C (RT reaction), 2 min at 94°C (Taq activation), 45 cycles of 30 sec at 94°C, 40 sec at 50°C, and 40 sec at 68°C, followed by 68°C for 7 min. RT-PCR was performed separately for each target using the same conditions for both N and P-gene assays. The unincorporated dNTPs and primers from the initial RT-PCR were removed by treating with ExoSAP-IT (Affymetrix). Twenty-five microliters of RT-PCR were treated with 10 μL ExoSAP-IT and incubated at 37°C for 30 min, followed by 10 min at 80°C to inactivate the enzymes.
2.3.2. Target-Specific Primer Extension (TSPE)
Linear amplification was then performed in the presence of biotin-labeled cytosine with the required TSPE primers present in a given reaction mix. The treated RT-PCR products were then combined for TSPE reactions. The 5′ ends of the TSPE primers were designed to contain 24 base TAG sequences complementary to the particular microsphere sets, whereas the remainder of the primer sequence was designed to bind to targets within the RT-PCR product. Biotin-dCTP was incorporated in the reaction to allow detection by streptavidin-R-phycoerythrin (SA-PE). Each TSPE reaction contained 5 μL of Exo-SAP-treated RT-PCR product, 0.75 U Tsp DNA polymerase (Invitrogen), 25 nM TSPE primer 5 μM dATP/dTTP/dGTP and biotin-dCTP (Invitrogen), 1X Tsp DNA polymerase reaction buffer (Invitrogen), and 4.0 mM MgCl2. Thermocycling was performed at 95°C for 2 min, followed by 30 cycles of 94°C for 30 s, 50°C for 30 s, and 72°C for 40 s with a final extension at 72°C for 5 min.
2.3.3. Microsphere Hybridisation
Products from the TPSE reactions were multiplexed with relevant MagPlex-TAG microspheres (Luminex MagPlex-TAG). Five microliters of TSPE reaction were hybridized in 50 μL 1X hybridization buffer (0.2 M NaCl/0.1 M Tris/0.08% Triton X-100, pH 8.0) with 500 each (microspheres/microsphere set/well) of the appropriate MagPlex-TAG microspheres containing the antitag complimentary to the 5′ TAG on the TSPE primer. The hybridization mixture was incubated at 96°C for 90 sec and 37°C for 30 min. The microsphere mixture was transferred to a black-sided 96-well Bio-Plex flat bottom plate (Bio-Rad) and washing of magnetic microspheres was performed using an automated plate washer (Bio-Plex pro II wash station; Bio-Rad).
2.3.4. Microsphere Identification and Fluorescence Detection
Seventy-five microlitres of 1X hybridization buffer containing 2 mg/L streptavidin-R-phycoerythrin (Invitrogen) were added to each plate well and the mixture was incubated in the dark at 37°C for 15 min. Instrument procedure was as described by the manufacture; briefly, 50 μL of the microsphere/TSPE/streptavidin-R-phycoerythrin mixture was injected into a Bio-Plex 200 instrument (Bio-Rad), at a sample plate temperature of 37°C. Each assay plate was analysed in a Bio-Plex 200 fluorometer (Bio-Rad) running at high RP1 target setting with 100 of each microsphere set analysed per well. Fluorescence was measured as units of Median Fluorescence Intensity (MFI). A positive result was initially defined as a value greater than three times the MFI obtained from a known HeV negative control.
2.4. qPCR Assays
A HeV N-gene-specific qPCR assay  was used for comparative assessment of the microsphere assays. Assay conditions and oligonucleotides were as described in the paper. Cut-off values were cycle threshold (CT) ≤ 40 for positive and CT ≥ 45 for negative. Results with CT values between 40 and 45 were deemed indeterminate.
2.5. Statistical Analysis
Statistical analyses were performed using Microsoft Excel 2007 and MedCal Version 18.104.22.168. ROC curve analysis was performed using 77 positive and 61 negative samples by qPCR for the henipa N- and P-gene microsphere array assay. For this analysis infected and noninfected horses were given the statuses 1 and 0, respectively. The area under the ROC curve (AUC) (Figure 2) was 0.948 for the henipa N (at cutoff > 488MFI) and 0.893 for the P-gene microsphere array assay (at a cut-off of > 523MFI). In the ROC analysis, values of 0.9 < AUC < 1 are considered highly accurate. A perfect test with a Se and Sp of 100% would have an AUC of 1 and be in the upper left corner of the graph . An interactive dot diagram was plotted using results of the henipa N- and P-gene microsphere array assays in relation to the infected (1) and noninfected (0) category, to show false negative and false positive results at cut-offs with highest combined Se and Sp.
3. Results and Discussion
3.1. Analytical Specificity of Microsphere Suspension Array Assays
The analytical specificity of the N- and P-gene-based microsphere array assays was assessed using RNA extracted from different virus isolates and nonrelated laboratory reference virus strains (listed in Materials and Methods). HeV and NiV isolates were positive in the generic henipavirus assays and their corresponding virus type-specific assays. All other tested viruses including CedPV were negative in the N- and P-gene assays. Although the recently isolated CedPV from Australian bats has been suggested as a henipa-like virus, it is quite distinct from HeV and NiV. Importantly, CedPV contained multiple sequence changes in each of the N- and P-gene primer regions for HeV and NiV. HeV and NiV are phylogenetically closely related having nucleotide sequence identities of 68.2% for whole genome, 78.4% for N-gene, and 70.0% for P-gene CDS, whereas the more distantly related CedPV has identities of only 47.5–48.1% for whole genome, 60.5–60.2% for N-gene, and 42.5–42.4% for P-gene CDS.
3.2. Analytical Sensitivity of Microsphere Suspension Array Assays
The analytical sensitivity of the microsphere array assays was assessed in direct comparison to the HeV specific qPCR assay by determining the limit of detection using tenfold serially diluted RNA template derived from HeV (Hendra virus/horse/Hendra/1994) and NiV (Nipah virus/Malaysia/1998). The microsphere suspension array assays had a dynamic range for detection equal to qPCR and were at least as sensitive as qPCR (data not shown). The fitness of microsphere array assays for the sensitive detection of HeV infection was further confirmed using archival samples from horses experimentally infected with HeV . Results were compared with qPCR and loop-mediated amplification (LAMP) assays  (Table 2). HeV was detected by microsphere array assays as early as qPCR or LAMP assays, two days after infection. There was excellent correlation between the N- and P-gene microsphere assays. Day 2 results for horse 3 were the only occasion where an N-gene positive sample was negative in the corresponding P-gene assay. Overall, the utility of microsphere assays was confirmed by the high level of sensitivity as was apparent with positive results at days 2, 4, and 5 in horse 3 when corresponding qPCR indicated indeterminate or undetected results.
3.3. Diagnostic Evaluation of Microsphere Suspension Array Assays
Assays were evaluated in a retrospective analysis of archival diagnostic horse samples (). Expected negative values for each of the microsphere array assays were obtained from presumed negative horse samples (). The mean MFI (+3STD) of these samples was 121 (+120) for the henipa N-gene assay, 126 (+141) for the HeV N-gene assay, 145 (+147) for the NiV N-gene assay, and 257 (+261) for the henipa P-gene assay. Archival RNA from diagnostic submissions obtained during investigations of HeV in horses in Australia in 2011 was tested and results then correlated with the original diagnostic results obtained from qPCR assays (Table 3). Five HeV qPCR positive samples in the originally diagnostic assay were negative in both the henipa N- and P-gene assay of archival RNA (Table 3). When archival RNA was retested by qPCR all five samples were HeV negative, most likely due to degradation of the archival RNA from these samples. This indicates that the diagnostic performance of the microsphere array assays may have been underestimated in the retrospective analysis of microsphere array assays on archival RNA in comparison with original qPCR diagnostic results. Out of 7 HeV indeterminate samples (by qPCR), the henipa N-gene microsphere assay returned 4 positive and 3 negative results and the henipa P-gene 6 negative and 1 positive result. An advantage of these microsphere array assays was the clear differentiation between positive and negative results and an ability to resolve unambiguous results as exemplified by clear resolution of the indeterminate results observed at the limit of detection in qPCR.
For ROC curve analysis (Figure 2), all of the samples positive in the N-gene henipa-specific microsphere assay were also positive in the HeV specific N-gene assay and none of these were found to be positive in the NiV-specific N-gene assay. HeV-specific positive and negative values in all assays were generally well separated, so changes from optimal cutoff values did not significantly change results in a ROC curve analysis (results not shown). For the N-gene henipa-specific assay the optimal ROC curve calculated cutoff MFI = 488 yielded sensitivity (Se) and specificity (Sp) readings of 92.2 (95% CI 83.8–97.1) and 98.4 (95% CI 91.2–100.0), respectively (Figures 2(a) and 2(b)). For the P-gene henipa-specific assay the calculated optimal cutoff MFI 523 gave an Se of 84.4 (95% CI 74.4–91.7) and an Sp of 98.3 (95% CI 90.9–100.0) (Figures 2(c) and 2(d)). The analysis of Sp and Se data for the HeV-specific assay showed near identical results to the generic henipa N-gene assay (data not shown). The ROC curve determined that negative cut-offs for all four assays were also found to exceed the highest MFI values obtained in an analysis of 40 presumed HeV negative horse swab samples in each assay. Direct comparison of all assays by ROC curve analysis using qPCR assay as reference standard identified the Henipa and HeV N-gene assays as the best performers for HeV detection. Assay accuracy for the henipa N, HeV-N and henipa P-gene assays was 0.941, 0.940, and 0.874, respectively, for samples samples (qPCR confirmed), increasing to 0.948, 0.946, and 0.893 for samples including 40 presumed negative samples unconfirmed by qPCR.
The HeV N-gene test exhibited the higher Se producing less false negative results. The Sp was similar for both tests. High Se is highly desirable for a test that diagnoses zoonotic and potentially fatal disease. As the reference test (qPCR) may be imperfect, it cannot be assumed that the status of the samples is 100% accurate. Evaluation of further field samples from infected and noninfected horses would be required to obtain more robust estimates for Se and Sp.
The microsphere array assay modules were specific for HeV and NiV. Based on sequence conservation of target regions, these assays should detect all known HeV and NiV isolates. Furthermore, the N-gene assay reliably differentiated HeV and NiV. The analytical sensitivity of the microsphere array assays matched that of qPCR assays in the limit of detection study, the detection of virus in experimentally infected horses, and in the retrospective analysis of diagnostic field samples. The microsphere array assays are based on an open diagnostic platform allowing a high degree of customisation. This facilitates the expansion of individual assay components into larger and more complex arrays and the update of assays in response to new and emerging viruses. The microsphere array assays offer advantages over qPCR in the level of readily achievable multiplexing. Assays can be designed as a modular system and combined into assay panels of increasing complexity. This ensures that assay sensitivity and specificity are not adversely affected by difficulties often observed in multiplexed qPCR reactions. This study demonstrated the utility of the microsphere array assays for detection of HeV. Our aim is to incorporate these HeV and NiV microsphere array assays as modules in future higher multiplexed microsphere arrays. This will facilitate the development of syndrome-based assay panels for disease investigation and agent surveillance in horse, bat, pig, and human populations.
Conflict of Interests
The authors declare that there is no conflict of interests.
The authors wish to thank the AAHL PCR Group for release of retrospective test samples and associated qPCR data and Vicky Boyd for her earlier involvement in reagent optimisation and valuable ongoing advice and assistance. Dr. Glenn Marsh kindly provided the virus isolates needed for analysis of assay performance. Linda Wright is also thanked for retrieving required test sample IDs from the AAHL diagnostic archive as is Dr. Brian Meehan for his support and encouragement throughout this study.
- C. H. Calisher, J. E. Childs, H. E. Field, et al., “Bats: important reservoir hosts of emerging viruses,” Clinical Microbiology Reviews, vol. 19, no. 3, pp. 531–545, 2006.
- G. A. Marsh and L. F. Wang, “Hendra and Nipah viruses: why are they so deadly?” Current Opinion in Virology, vol. 2, no. 3, pp. 242–247, 2012.
- K. Murray, P. Selleck, P. Hooper et al., “A morbillivirus that caused fatal disease in horses and humans,” Science, vol. 268, no. 5207, pp. 94–97, 1995.
- K. B. Chua, K. J. Goh, K. T. Wong et al., “Fatal encephalitis due to Nipah virus among pig-farmers in Malaysia,” The Lancet, vol. 354, no. 9186, pp. 1257–1259, 1999.
- S. Wacharapluesadee, B. Lumlertdacha, K. Boongird et al., “Bat Nipah virus, Thailand,” Emerging Infectious Diseases, vol. 11, no. 12, pp. 1949–1951, 2005.
- J. M. Reynes, D. Counor, S. Ong et al., “Nipah virus in Lyle's flying foxes, Cambodia,” Emerging Infectious Diseases, vol. 11, no. 7, pp. 1042–1047, 2005.
- I. Sendow, H. E. Field, J. Curran et al., “Henipavirus in Pteropus vampyrus bats, Indonesia,” Emerging Infectious Diseases, vol. 12, no. 4, pp. 711–712, 2006.
- D. T. S. Hayman, R. Suu-Ire, A. C. Breed et al., “Evidence of henipavirus infection in West African fruit bats,” PLoS ONE, vol. 3, no. 7, Article ID e2739, 2008.
- G. A. Marsh, C. de Jong, J. A. Barr, et al., “Cedar virus: a novel henipavirus isolated from Australian bats,” PLOS Pathogens, vol. 8, no. 8, Article ID 100283, 2012.
- I. Smith, A. Broos, C. de Jong, et al., “Identifying Hendra virus diversity in pteropid bats,” PLoS ONE, vol. 6, no. 9, Article ID e25275, 2011.
- Queensland Horse Council, “Hendra Virus,” http://www.qldhorsecouncil.com/QHC%20Documents/Notifiable%20Diseases%20Information%20Sheets/Hendra%20Virus.pdf.
- J. F. Drexler, V. M. Corman, M. A. Muller, et al., “Bats host major mammalian paramyxoviruses,” Nature Communications, vol. 3, article 796, 2012.
- K. S. Baker, S. Todd, G. Marsh, et al., “Co-circulation of diverse paramyxoviruses in an urban African fruit bat population,” Journal of General Virology, vol. 93, part 4, pp. 850–856, 2012.
- S. Tong, S. W. W. Chern, Y. Li, M. A. Pallansch, and L. J. Anderson, “Sensitive and broadly reactive reverse transcription-PCR assays to detect novel paramyxoviruses,” Journal of Clinical Microbiology, vol. 46, no. 8, pp. 2652–2658, 2008.
- P. Jokela, H. Piiparinen, L. Mannonen, et al., “Performance of the Luminex xTAG respiratory viral panel fast in a clinical laboratory setting,” Journal of Virological Methods, vol. 182, no. 1-2, pp. 82–86, 2012.
- J. Liu, G. Kibiki, V. Maro et al., “Multiplex reverse transcription PCR Luminex assay for detection and quantitation of viral agents of gastroenteritis,” Journal of Clinical Virology, vol. 50, no. 4, pp. 308–313, 2011.
- M. A. Johnson, M. J. Yoshitomi, and C. S. Richards, “A comparative study of five technologically diverse CFTR testing platforms,” Journal of Molecular Diagnostics, vol. 9, no. 3, pp. 401–407, 2007.
- I. Janse, J. M. Bok, R. A. Hamidjaja, et al., “Development and comparison of two assay formats for parallel detection of four biothreat pathogens by using suspension microarrays,” PLoS ONE, vol. 7, no. 2, Article ID e31958, 2012.
- Y. Yang, J. Wang, H. Wen, et al., “Comparison of two suspension arrays for simultaneous detection of five biothreat bacterial in powder samples,” Journal of Biomedicine and Biotechnology, vol. 2012, Article ID 831052, 8 pages, 2012.
- B. J. Hindson, S. M. Reid, B. R. Baker et al., “Diagnostic evaluation of multiplexed reverse transcription-PCR microsphere array assay for detection of foot-and-mouth and look-alike disease viruses,” Journal of Clinical Microbiology, vol. 46, no. 3, pp. 1081–1089, 2008.
- G. A. Marsh, J. Haining, T. J. Hancock, et al., “Experimental infection of horses with Hendra virus/Australia/horse/2008/Redlands,” Emerging Infectious Diseases, vol. 17, no. 12, pp. 2232–2238, 2011.
- A. J. Foord, D. Middleton, and H. G. Heine, “Hendra virus detection using Loop-Mediated Isothermal Amplification,” Journal of Virological Methods, vol. 181, no. 1, pp. 93–96, 2012.
- A. B. Drummond AJ, S. Buxton, M. Cheung, et al., “Geneious v5.4,” 2012, http://www.geneious.com/.
- K. S. Feldman, A. Foord, H. G. Heine et al., “Design and evaluation of consensus PCR assays for henipaviruses,” Journal of Virological Methods, vol. 161, no. 1, pp. 52–57, 2009.
- M. Greiner, D. Pfeiffer, and R. D. Smith, “Principles and practical application of the receiver-operating characteristic analysis for diagnostic tests,” Preventive Veterinary Medicine, vol. 45, no. 1-2, pp. 23–41, 2000.