- About this Journal ·
- Abstracting and Indexing ·
- Aims and Scope ·
- Article Processing Charges ·
- Author Guidelines ·
- Bibliographic Information ·
- Citations to this Journal ·
- Contact Information ·
- Editorial Board ·
- Editorial Workflow ·
- Free eTOC Alerts ·
- Publication Ethics ·
- Recently Accepted Articles ·
- Reviewers Acknowledgment ·
- Submit a Manuscript ·
- Subscription Information ·
- Table of Contents

International Journal of Spectroscopy

Volume 2013 (2013), Article ID 726820, 8 pages

http://dx.doi.org/10.1155/2013/726820

## Resolution of Ternary Mixture of Aspirin, Atorvastatin, and Clopidogrel by Chemometric-Assisted UV Spectroscopic and Liquid Chromatography Methods

^{1}Pharmaceutical Analytical Chemistry, Department of Chemistry, Alaqsa University, P.O. Box 4051, Gaza 76888, Palestine^{2}Analytical Chemistry, Department of Chemistry, Alaqsa University, P.O. Box 4051, Gaza 76888, Palestine^{3}Inorganic Analytical Chemistry, Department of Chemistry, Alaqsa University, P.O. Box 4051 Gaza 76888, Palestine^{4}National Institute of Research for Electrochemistry and Condensed Matter, 060021 Bucharest, Romania

Received 26 February 2013; Accepted 20 August 2013

Academic Editor: L. J. Ming

Copyright © 2013 Mahmoud Mohamed Issa et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.

#### Abstract

Two chemometrics-assisted UV spectrophotometric methods were proposed for the resolution of ternary mixtures without any chemical pretreatment. The first method is based on modification of H-point standard addition method which permits simultaneous analysis of three species from a unique calibration set by making the simultaneous addition of the three analytes. Quotient between the spectra of aspirin, atorvastatin, and clopidogrel was obtained and the results showed that simultaneous determination of aspirin, atorvastatin, and clopidogrel can be obeyed in the linear range 2.5–20 *μ*g mL^{−1} of aspirin, 2.5–17.5 *μ*g mL^{−1} of atorvastatin, and 2.5–20 *μ*g mL^{−1} of clopidogrel in ternary mixture. The second method is based on the combination of the first derivative spectra and Cramer's matrix rule. In the matrix calculation, clopidogrel has zero crossing point at 316.8 and 212 nm, while for atorvastatin the zero crossing point at 250 nm where the matrix is greatly simplified and easily solved. The linear concentration ranges were 2.5–20 *μ*g mL^{−1} aspirin, 2.5–17.5 *μ*g mL^{−1} atorvastatin and 2.5–20 *μ*g mL^{−1} clopidogrel in ternary mixtures. The results proved that the simultaneous determination of aspirin, atorvastatin, and clopidogrel could be obeyed. Both methods were applied for capsules containing the three ingredients and results were in good concordance with alternative liquid chromatography.

#### 1. Introduction

Aspirin (ASP, acetylsalicylic acid) is used as an analgesic, antipyretic, anti-inflammatory medication and an antiplatelet agent. For clopidogrel bisulphate (CLOP), chemically it is methyl (+)-(S)-*α*-(2-chlorophenyl)-6,7-dihydrothieno[3, 2-c]pyridine-5(4H)acetate sulfate (1 : 1) and is an oral, thienopyridine class antiplatelet agent used to inhibit blood clots in coronary artery disease, peripheral vascular disease, and cerebrovascular disease. Atorvastatin calcium (ATOR), [R-]-2-(4-fluorophenyl)-*β*,*δ*,-dihydroxy-5-(1-methylethyl)-3-phenyl-4 [(phenylamino)carbonyl]-lH-pyrrole-1-heptanoic acid, calcium salt (2 : 1) trihydrate is an inhibitor of 3 hydroxy-3-methylglutaryl-coenzyme A (HMG-CoA) reductase. Chemical structures of ASP, CLOP, and ATOR are shown in Scheme 1.

A combination of ASP, CLOP, and ATOR drugs is available in the market as capsules. The ternary combination is used for atherosclerotic patients suffering from various heart diseases. Clinical trials showed that combination therapy when used in dyslipidaemic patient with coronary heart disease reduced cardiovascular events.

The official monographs describe the procedure for individual assay of ASP [1, 2], CLOP [1], and ATOR [2–4]. The literature survey revealed very few analytical methods such as spectrophotometry [5], HPTLC [6], HPLC [7, 8], and LC [9, 10] which have been reported for the determination of ASP, CLOP, and ATOR in the ternary combination.

Simultaneous analysis of a ternary mixture is difficult to perform by classical spectrophotometric method due to overlapping spectra. Chemometrics is one of the mathematical and statistical techniques used for the resolution of overlapping spectra of multicomponent mixtures. Three developed methods for resolution of a ternary mixture have been shown. Salinas et al.’s method [11] is based on the use of the derivative ratio spectra for a binary mixture. Nevado et al.’s method [12] is based on the measurements of the amplitude at the zero crossing points in the derivative spectrum of the ratio spectra. In Dinç and Onur's [13] method, the simultaneous analysis of a ternary mixture is based on the use of the derivative of the ratio spectrum obtained by dividing the absorption spectrum of the ternary mixture by a standard spectrum of a mixture of two of the three compounds in the mixture.

Recently, Hajian and Afshari [14] developed a new method for the simultaneous analysis of a ternary mixture by combination of double divisor-ratio spectra derivative and H-point standard addition method (HPSAM). The fundamentals of the HPSAM were outlined by Campins-Falco et al. [15, 16]. It is a modification of the standard addition method and has been frequently applied in spectrophotometry [17–23]. It was also developed for resolution of binary mixture with simultaneous addition of both analytes [24]. In 2006, Hasani et al. proposed a modification for resolution of ternary mixture by using a single calibration graph with simultaneous standard additions of three analytes [25].

In this work, the HPSAM with simultaneous standard addition of three analytes and combination of first derivatives spectra and Cramer's matrix rule has been applied for the simultaneous analysis of ASP, ATOR, and CLOP in their ternary mixture. The matrix calculation based on Cramer's rule for a system of equations with three unknowns can be used for simultaneous determination of these compounds in the same ternary mixture and pharmaceutical formulations [13, 26]. To the best of our knowledge, this is the first report on the simultaneous determination of ASP, ATOR, and CLOP using these methods. As an alternative method, HPLC method was used for simultaneous determination of these compounds in the same ternary mixture and pharmaceutical formulation [7].

#### 2. Theory

##### 2.1. HPSAM with Simultaneous Addition of Three Analytes

The absorbance of a ternary mixture () consisting of compounds ASP, ATOR, and CLOP at and is where and are selected according to Hence, where is the slop due to the addition of species of ASP, ATOR, and CLOP in the line obtained at or . The required data to apply the method are the absorbance of the mixture and the absorbance of the mixture spiked with known amount of ASP, ATOR, and CLOP at and ; therefore If (5) is multiplied by , the following equation can be obtained: At the intersection (H-point), ; consequently, the ASP concentration can calculate as where the superscript zero denotes the sample solution.

For analysis of ASP, the quotient between ATOR and CLOP spectra is applied and the wavelength pairs that show the same value are selected. Although this relation depends on the concentration of ATOR and CLOP, it will be equal at the two wavelengths. The concentration of ATOR and CLOP is determined by analogous procedure.

##### 2.2. Combination of First Derivatives and Cramer’sMatrix Calculation Method

Molar absorptivity (*ε*) values were calculated by using the absorbance measured at 212 nm, 250 nm, and 316.8 nm for first-order spectra for each compound in the ternary mixture. The wavelength values were selected because CLOP has zero-crossing point at 316.8 and 212 nm and zero-crossing point of ATOR at 250 nm, in which the matrix was greatly simplified and easily solved. By using (*ε*) values, a system of equations with three unknowns can be written for the compounds in the ternary mixture as follows:
where denotes the absorbance of the ternary mixture and (*ε*) represents the values of molar absorptivity for the calculated ASP, ATOR, and CLOP, respectively, at 212, 250 nm, and 316.8 nm. *C* is the molar concentration of ASP, ATOR, and CLOP.

The matrix simplifies and solves the system of equations with three unknowns as follows: This matrix can be solved and each compound was determined by solving the following operations: where

#### 3. Experimental

##### 3.1. Apparatus

A Shimadzu (Kyoto, Japan) UV-1650 PC, UV-Visible double-beam spectrophotometer with two matched 1 cm path-length quartz cells was used. The subsequent statistical manipulations were performed by transferring the spectral data to Microsoft Excel 2010 program and processing them with the standard curve fit package and matrix calculation.

A waters breeze 1515 chromatograph (Waters, USA), equipped with an isocratic pump, UV detector, and an rheodyne injector was used. The separation was achieved on a symmetry C_{18} column (3.5 *μ*m particle size, mm i.d.) with a mixture of acetonitrile: phosphate buffer pH 3.0; at a flow rate of 1.2 mL/min and the detection was monitored at 235 nm.

##### 3.2. Pharmaceutical Formulations

Commercial product ASPTORGREL capsules (produced by Middle East pharm, Palestine, Batch no. 39712 containing 75 mg of aspirin, 10 mg of atorvastatin, and 75 mg of clopidogrel per capsule) were analyzed.

Aspirin, atorvastatin, and clopidogrel were kindly donated by Middle East pharm. All chemical and reagents used (methanol, acetonitrile, and orthophosphoric acid) were of HPLC grade (Merck).

##### 3.3. Reagents

Stock solutions of 75 mg/100 mL ASP, 10 mg/100 mL ATOR, and 75 mg/100 mL CLOP were dissolved in methanol, separately. Standard solutions were prepared by serial dilution with methanol. Synthetic mixtures were prepared by mixing known amount of ASP, CLOP, and ATOR as shown in Table 1. Standard solutions were used in preparation of calibration graphs and for spectral measurements.

##### 3.4. Procedure for the Assay of the Pharmaceutical Formulation

Five capsules of ASPTORGREL were accurately weighed and the content mixed thoroughly together; an amount equivalent to one capsule was dissolved in methanol in a 100 mL calibrated flask. After shaking, the solution was filtered into a 100 mL calibrated flask. The residue was washed with methanol, then the volume was completed to the mark. All the methods were applied to the solutions, thus were prepared.

#### 4. Application of Methods

##### 4.1. HPSAM with Simultaneous Addition of Three Analytes

The absorption spectra of ASP, ATOR, and CLOP (5 *μ*g mL^{−1} each in methanol) and their ternary mixture were recorded in wavelength range 200–400 nm and saved as a text file. Synthetic samples containing different concentration ratios of ASP, ATOR, and CLOP were prepared and standard additions of them were made. Simultaneous determination of ASP, ATOR, and CLOP with HPSAM was performed by measuring the absorbance at 231.0, 317.0, 275.8, 329.2, 235.6, and 293.6 nm for each sample solution. The concentration ranges of ASP, ATOR, and CLOP for construction of HPSAM calibration curve were 2.5–20, 2.5–17.5, and 2.5–20 *μ*g mL^{−1}, respectively.

##### 4.2. Combination of First Derivative Spectra and Cramer’s Matrix Calculation

In this method, the first derivative spectra of ASP, ATOR, and CLOP and of different concentrations of ternary mixture were recorded from the data in the first method. The molar absorptivity (mol^{−1} L cm^{−1}) values were calculated by measuring the absorbance at 212, 250, and 316.8 nm for first-order spectra for each of the compounds in the ternary mixture. By using the molar absorptivity values, a system of equations with three unknowns can be obtained, which can be solved by means of Cramer's rule, and the concentration of ASP (2.5–20 *μ*g mL^{−1}), ATOR, (2.5–17.5 *μ*g mL^{−1}), and CLOP (2.5–20 *μ*g mL^{−1}) was determined.

##### 4.3. HPLC Method

The concentration of ASP, ATOR, and CIOP can be determined simultaneously by using Londhe et al.’s method [7]. Separation was achieved on an isocratic reversed-phase high-performance liquid chromatography with inertsil ODS analytical column ( mm; 3.5 *μ*m) with a mixture of acetonitrile phosphate buffer pH 3.0 as mobile phase and at a flow rate of 1.2 mL/min. UV detection was performed at 235 nm.

#### 5. Results and Discussion

##### 5.1. HPSAM with Simultaneous Addition of Three Analytes

The absorption spectra of ASP, ATOR, and CLOP and their ternary synthetic mixture are shown in Figure 1. As can be seen, the spectra overlapped in the region 200–400 nm. In order to resolve the mixture, the quotient between the spectra of ASP, ATOR, and CLOP must be obtained. Figure 2 shows these quotients; hence, the wavelength pairs that show the same ratio of absorbance value in order to calculate the analyte concentration can be obtained. The three wavelength pairs, 231.0–317.0, 275.8–329.2, and 235.6–293.6 nm, were chosen for determination of ASP, ATOR, and CLOP, respectively. Figure 3 shows the H-point standard addition plots for calculation of ASP, ATOR, and CLOP concentration from one calibration set.

Various mixture compositions of ASP, ATOR, and CLOP were prepared and tested between 2.5–20 *μ*g mL^{−1} ASP, 2.5–17.5 *μ*g mL^{−1} ATOR, and 2.5–20 *μ*g mL^{−1} CLOP in ternary mixture as shown in Table 1. Mean recoveries and relative standard deviation () of the method were obtained as 99.3 and 0.92% for ASP, 99.6 and 1.35% for ATOR, and also 99.9 and 1.43% for CLOP, respectively. Limits of detection were 0.28, 0.17, and 0.33 *μ*g mL^{−1}, respectively for ASP, ATOR, and CLOP, which were calculated as , where and SD_{H} are the mean and standard deviation for five replicate measurements of blank sample using HPSAM [27].

##### 5.2. Combination of First Derivative and Cramer’s Matrix Calculation Method

As seen in Figure 4, the spectra of the first derivative of the three compounds, ASP, ATOR, and CLOP, overlapped in the region 200–400 nm. The wavelength values of ASP, ATOR, and CLOP are chosen as 316.8, 250, and 212 nm, respectively. In application of matrix calculation, CLOP has zero crossing point at 316.8 and 212 nm, whilst zero crossing point of ATOR is at 250 nm. A system of equations with the three compounds (5.0 *μ*g mL^{−1} of each in methanol) can be written as
By using the matrix calculation, the concentration is 4.88, 4.90, and 5.16 *μ*g mL^{−1} for ASP, ATOR, and CLOP, respectively. Various mixture compositions of ASP, ATOR, and CLOP were prepared and tested between 2.5–20 *μ*g mL^{−1} ASP, 2.5–17.5 *μ*g mL^{−1} ATOR, and 2.5–20 *μ*g mL^{−1} CLOP in ternary mixtures as shown in Table 2. Mean recoveries and relative standard deviations () of the method were obtained as 100.0 and 1.44% for ASP, 99.8 and 1.35% for ATOR, and also 100.9 and 1.55% for CLOP, respectively.

Limits of detection were 0.23, 0.29, and 0.21 *μ*g mL^{−1}, respectively, for ASP, ATOR, and CLOP, which were calculated as , where is the slope and SD is the standard deviation of the regression line.

##### 5.3. HPLC Method

The chromatogram at 235 nm showed a complete resolution of all peaks (Figure 5). The retention times were 1.32, 4.040, and 8.855 min, for ASP, ATOR, and CLOP, respectively. The calibration graph was linear over the range of 5–30 *μ*g mL^{−1} ATOR and 30–105 *μ*g mL^{−1} for ASP and CIOP. Mean recoveries and relative standard deviation () for the HPLC method were found to be 99.59 and 0.56% for ASP, 99.73 and 0.69% for ATOR, and 100.17 and 0.25% for CLOP, respectively, in the synthetic mixture prepared from known amounts of ASP, ATOR, and CLOP.

##### 5.4. Application

To evaluate the applicability of the proposed methods, The two proposed methods were applied to simultaneous determination of ASP, ATOR, and CLOP in commercially available capsules. Table 3 shows a good agreement between the obtained results which indicates the successful applicability of the methods described in this work.

#### 6. Conclusion

Chemometric methods essentially reduce the solvents used and the duration of analysis by doing most of the works on the front desk using microcomputers with appropriate softwares on the primary data generated in the laboratory work. The proposed methods have been successfully applied to the simultaneous determination of ASP, ATOR, and CLOP in capsule dosage of synthetic samples. A comparative study of the use of HPLC and chemometrics-assisted methods for the resolution of ternary mixture of ASP, ATOR, and CLOP has been accomplished, showing that the proposed methods provide, with adequate software support, a good pattern of the high resolving power of this technique. HPSAM is a modification of the previously described HPSAM that permits the resolution of three species from a unique calibration set by making the simultaneous addition of the three analytes. The principal advantage of the method is using a single calibration curve with simultaneous standard additions of three analytes instead of individual standard addition for every analyte and a calibration curve for each analyte. Based on the results obtained, it has been shown that the combination of first derivative and Cramer’s matrix calculation method calculates the analyte concentrations, ASP, ATOR, or CLOP, in the presence of each other. The principal advantage of the method is using the zero crossing point in which the matrix is greatly simplified and easily solved by means of Cramer’s rule.

#### Acknowledgments

The authors are thankful to the Middle East pharm, (Gaza, Palestine) for providing samples of aspirin, atorvastatin, and clopidogrel pure formulations and commercial product ASPTORGREL capsules. Also, they would like to thank Alaqsa University for providing the necessary material and facilities for the research.

#### References

*US Pharmacopoeia*, US Pharmacopial Convention, Rockville, Md, USA, 23rd edition, 1955.*Indian Pharmacopoeia*, vol. 2, The Controller of Publications, Government of India, New Delhi, India, 1996.*British Pharmacopoeia*, vol. 2, HMSO Publication Centre, London, UK, 1993.*Martindale: The Complete Drug Reference*, Pharmaceutical Press, London, UK, 35th edition, 2007.- S. Singh, N. Dubey, and D. K. Jain, “Simultaneous estimation of atorvastatin, clopidogrel and aspirin in capsule dosage forms using UV-spectroscopy,”
*Asian Journal of Research in Chemistry*, vol. 3, no. 4, pp. 885–887, 2010. - S. Londhe, S. Mulgund, R. Deshmukh, and K. Jain, “Simultaneous HPTLC analysis of aspirin, atorvastatin calcium and clopidogrel bisulphate in the bulk drug and in capsules,”
*Acta Chromatographica*, vol. 22, no. 2, pp. 297–305, 2010. View at Publisher · View at Google Scholar · View at Scopus - S. V. Londhe, R. S. Deshmukh, S. V. Mulgund, and K. S. Jain, “Development and validation of a reversed-phase HPLC method for simultaneous determination of aspirin, atorvastatin calcium and clopidogrel bisulphate in capsules,”
*Indian Journal of Pharmaceutical Sciences*, vol. 73, no. 1, pp. 23–29, 2011. View at Scopus - H. O. Kaila, M. A. Ambasana, and A. K. Shah, “A Simple and rapid ultra-performance liquid chromatographic assay method for the simultaneous determination of aspirin, clopidogrel bisulphate and atorvastatin calcium in capsule dosage form,”
*International Journal of ChemTech Research*, vol. 3, no. 1, pp. 459–465, 2011. View at Scopus - H. J. Panchal, B. N. Suhagia, N. J. Patel, I. S. Rathod, and B. H. Patel, “Simultaneous estimation of atorvastatin calcium, ramipril and aspirin in capsule dosage form by RP-LC,”
*Chromatographia*, vol. 69, no. 1-2, pp. 91–95, 2009. View at Publisher · View at Google Scholar · View at Scopus - K. Alla, “A stability-indicating LC method for the simultaneous determination of metoprolol, atorvastatin and ramipril in combined pharmaceutical dosage form,”
*Journal of AOAC International*, vol. 90, pp. 1547–1553, 2007. - F. Salinas, J. J. B. Nevado, and A. E. Mansilla, “A new spectrophotometric method for quantitative multicomponent analysis resolution of mixtures of salicylic and salicyluric acids,”
*Talanta*, vol. 37, no. 3, pp. 347–351, 1990. View at Scopus - J. J. B. Nevado, C. G. Cabanillas, and F. Salinas, “Spectrophotometric resolution of ternary mixtures of salicylaldehyde, 3-hydroxybenzaldehyde and 4-hydroxybenzaldehyde by the derivative ratio spectrum-zero crossing method,”
*Talanta*, vol. 39, no. 5, pp. 547–553, 1992. View at Scopus - E. Dinç and F. Onur, “Application of a new spectrophotometric method for the analysis of a ternary mixture containing metamizol, paracetamol and caffeine in tablets,”
*Analytica Chimica Acta*, vol. 359, no. 1-2, pp. 93–106, 1998. View at Publisher · View at Google Scholar · View at Scopus - R. Hajian and N. Afshari, “The spectrophotometric multicomponent analysis of a ternary mixture of ibuprofen, caffeine and paracetamol by the combination of double divisor-ratio spectra derivative and H-point standard addition method,”
*E-Journal of Chemistry*, vol. 9, no. 3, pp. 1153–1164, 2012. View at Publisher · View at Google Scholar · View at Scopus - F. Bosch-Reig and P. Campins-Falco, “H-point standard additions method part 1 fundamentals and application to analytical spectroscopy,”
*The Analyst*, vol. 113, no. 7, pp. 1011–1016, 1988. View at Scopus - P. Campins-Falco, F. Bosch-Reig, and A. M. Benet, “Spectrophotometric analysis of mixtures of two components with extensively or completely overlapping spectra by the H-point standard additions method,”
*Fresenius' Journal of Analytical Chemistry*, vol. 338, no. 1, pp. 16–21, 1990. View at Publisher · View at Google Scholar · View at Scopus - P. Campins-Falco, F. Bosch-Reig, and J. Verdu-Andres, “Evaluation and elimination of the ‘blank bias error’ using the H-point standard addition method. Application to spectrophotometric determinations using absorbent blank,”
*Analytica Chimica Acta*, vol. 270, no. 1, pp. 253–265, 1992. View at Publisher · View at Google Scholar · View at Scopus - F. Bosch-Reig, J. Verdú-Andrés, P. Campíns-Falcó, and C. Molíns-Legua, “Study of the behaviour of the absorbent blanks in analytical procedures by using the H-Point standard additions method (HPSAM),”
*Talanta*, vol. 41, no. 1, pp. 39–52, 1994. View at Scopus - H. Abdollahi, “Simultaneous spectrophotometric determination of chromium(VI) and iron(III) with chromogenic mixed reagents by H-point standard addition method and partial least squares regression,”
*Analytica Chimica Acta*, vol. 442, no. 2, pp. 327–336, 2001. View at Publisher · View at Google Scholar · View at Scopus - R. Hajian, N. Shams, and A. Rad, “Application of H-point standard addition method for simultaneous spectrophotometric determination of hydrochlorothiazide and propranolol,”
*Journal of the Brazilian Chemical Society*, vol. 20, no. 5, pp. 860–865, 2009. View at Scopus - M. M. Issa, R. N. Nejem, A. A. Shanab, and N. T. Shaat, “Kinetic simultaneous spectrophotometric determination of paracetamol and ibuprofen using H-point standard addition method,”
*Journal of Chemical Pharmaceutical Research*, vol. 4, no. 7, pp. 3535–3540, 2012. - K. Lakshmi and S. Lakshmi, “Simultaneous spectrophotometric determination of valsartan and hydrochlorothiazide by H-point standard addition method and partial least squares regression,”
*Acta Pharmaceutica*, vol. 61, no. 1, pp. 37–50, 2011. View at Publisher · View at Google Scholar · View at Scopus - M. M. Issa, R. M. Nejem, A. A. Shanab, and N. T. Shaat, “Kinetic spectrophotometric H-point standard addition method for the simultaneous determination of diloxanide furoate and metronidazole in binary mixtures and biological fluids,”
*Spectrochimica Acta A*, vol. 114, pp. 592–598, 2013. - P. Campíns-Falcó, J. Verdú-Andrés, and F. Bosch-Reig, “H-point standard additions method for resolution of binary mixtures with simultaneous addition of both analytes,”
*Analytica Chimica Acta*, vol. 315, no. 3, pp. 267–278, 1995. View at Publisher · View at Google Scholar · View at Scopus - M. Hasani, L. Yaghoubi, and H. Abdollahi, “H-point standard addition method for simultaneous determination of Fe(II), Co(II) and Cu(II) in micellar media with simultaneous addition of three analytes,”
*Talanta*, vol. 68, no. 5, pp. 1528–1535, 2006. View at Publisher · View at Google Scholar · View at Scopus - H. G. Charlotte and T. L. Threlfall,
*UV-Vis Spectroscopy and Its Applications*, Springer, New York, NY, USA, 1992. - V. Thomsen, D. Schatzlein, and D. Mercuro, “Limits of detection in spectroscopy,”
*Spectroscopy*, vol. 18, no. 12, pp. 112–114, 2003. View at Scopus