Review Article  Open Access
Christian Morawe, Markus Osterhoff, "Hard XRay Focusing with Curved Reflective Multilayers", XRay Optics and Instrumentation, vol. 2010, Article ID 479631, 8 pages, 2010. https://doi.org/10.1155/2010/479631
Hard XRay Focusing with Curved Reflective Multilayers
Abstract
This paper provides a comprehensive overview on the utilization of curved graded multilayer coatings as focusing elements for hard Xrays. It concentrates on the KirkpatrickBaez (KB) focusing setup that has been developed at 3rd generation synchrotron sources worldwide. The optical performance of these devices is evaluated applying analytical and numerical approaches. The essential role of the multilayer coating and its meridional dspacing gradient are discussed as well as important technological issues. Experimental data and examples of operational KB focusing devices and applications complement the work.
1. Introduction
Over the last decade, hard Xray optics has evolved rapidly, accelerated by the intense use of extremely brilliant 3rd generation synchrotron radiation sources. The use of smaller samples, the study of local areas, or the need to scan an object with high spatial resolution pushed the development of focusing devises. At present, Xray beams can be focused down to spots below 50 nm using various optical elements such as Fresnel Zone Plates (FZPs) [1], Compound Refractive Lenses (CRLs) [2], Wave Guides (WGs) [3], capillary optics [4], curved Total Reflection Mirrors (TRMs) [5, 6], transmission Multilayer Laue Lenses (MLLs) [7, 8], and Reflective Multilayers (RMLs) [9, 10]. Several theoretical studies, mainly numerical wave optical calculations, were published treating the cases of FZPs [11], CRLs [12], TRMs [13, 14], and MLLs [15, 16]. Advanced numerical ray tracing calculations have been applied to RMLs [17, 18]. Alternatively, a purely analytical approach was proposed [19, 20]. The topic of this publication is the application of curved graded RMLs to KB focusing devices [21]. The KB geometry consists of two perpendicularly oriented curved mirrors and is particularly useful on synchrotron beamlines since it can transform the elongated synchrotron source into a nearly circular spot (Figure 1). Being an offaxis system the device produces a clean spot with low background and does not require beam stops or protecting apertures. A single focusing mirror per dimension operating under grazing incidence, as outlined in chapter 2, is not suitable for imaging applications since the Abbé sine condition [22] is not fulfilled. However, two successive mirrors (elliptic/hyperbolic), the so called Wolter geometry [23], can overcome this limitation.
2. Basic Optics Considerations
2.1. Resolution Limits
As in visible light optics the diffraction limited resolution (Full Width at Half Maximum, FWHM) of an Xray optical element with a flat rectangular aperture is given by where is the numerical aperture with the opening angle 2 and the optical index and the wavelength of the radiation. The minimum achievable spot size, however, is also limited by the demagnified image of the source of size where is the source distance and the image distance with respect to the optics. The phenomenon can also be treated in terms of the coherence properties of the beam. With a sourcesize limited transversal coherence length [24] and assuming that the optical aperture selects only the coherent section of the incoming beam, the diffraction limited spot size becomes This means that, apart from a geometrical factor, the diffraction limited spot size of a coherently illuminated optical element is of the order of the size of the source image. Additional performance limiting effects of real optics are figure errors, scattering from defects or nonuniform areas, and further experimental obstacles like alignment, vibration, or drift.
Whilst the short wavelength of hard Xrays alone would principally allow for a spot size far below the nm scale, the main drawback in Xray optics remains the low achievable numerical aperture. The latter is caused by the weak interaction of highenergy photons with matter, as given by the complex optical index where describes the dispersion and the absorption in the material. Most development efforts in focusing optics for hard Xrays are therefore dedicated to increasing the numerical aperture.
In the case of reflecting optical elements and neglecting phase shifts upon reflection, the ideal geometry for a pointtopoint focusing setup would be an ellipsoid. Since this work deals with KB setups operated at very small grazing angles, the scheme can be very well approximated by perpendicular projections on two elliptic cylinders. Thus, the treatment can be reduced to two individual ellipses.
2.2. Total Reflection Mirrors
In the case of a single elliptic TRM the range of grazing angles is limited to the upper side by the critical angle of total external reflection (Figure 2) given by and to the lower side only by the physical mirror length when approaching the minimum angle of incidence at the intersection point with the semiminor axis of the ellipse. Since is a small quantity for hard Xrays, the achievable opening is very limited. A quantitative estimate can be made assuming a full opening of up to half of the critical angle [9]. leading to a diffraction limit of where is the classical electron radius and the electron density of the material. This means that the focusing power of a TRM is entirely limited by material properties. For a Pt mirror one obtains a minimum spot size of about 25 nm FWHM. It should be noted that, in contrast to other techniques, TRM focusing has the advantage of being nondispersive.
2.3. Reflective Multilayers
Considerable improvement can be obtained by the use of RMLs where a stack of periodic bilayers reflects the incoming wave field. Provided the ML dspacing matches the modified Bragg equation at any point along the mirror, constructive interference enhances the reflectivity up to values that can reach 90% or more. The Bragg angle of a RML can be several times bigger than the critical angle of a TRM. Figure 2 shows a comparison of the simulated reflectivity between a Pt TRM and a 200period W/B_{4}C RML at a photon energy of 50 keV.
The typical number of bilayers in RMLs is of the order of 100, leading to an energy resolution in the percent range. In conjunction with high peak reflectivity, this means that RMLs can provide an integrated reflectivity exceeding those of perfect crystals by two orders of magnitude. Flat RMLs are well understood and have been used as Xray optical elements for many years [25, 26]. Thanks to the flexible dspacing and the choice of suited material pairs the optical performance of RMLs can be optimized according to photon energy, incidence angle, and further experimental parameters. This choice is also affected by properties like interface diffusion, stress, and chemical stability.
Instead of periodic RMLs, nonperiodic layered structures can be deposited to obtain broader reflectivity profiles [27]. In combination with meridional thickness gradients they can also be applied to KB setups [28].
To focus Xrays an RML has to be curved in accordance with the angular dependence of the ellipse where is its semiminor axis.
As for TRMs, the achievable resolution of curved RMLs can be estimated analytically [9]. Applying the Bragg law without refraction correction which is a good approximation for small dspacings and low wavelengths, respectively, the diffraction limit can be calculated as Here, Λ_{1,2} indicate the ML dspacings at the respective edges of the mirror. The resolution is therefore limited only by the lateral dspacing gradient of the RML. For a short period RML with strong gradient () focal spots of about 5 nm FWHM appear realistic. Equation (12) can be compared with the corresponding formula for a linear Fresnel Zone Plate (FZP) with outermost zone width . Considering a RML as an equivalent to half of a linear zone plate (1/2 FZP) with outermost repetition period and setting the largest ML dspacing Λ_{1} to infinity (half of central FZP zone) one obtains Equation (14) shows that, within the given approximation, both optical elements are different representations of the same focusing configuration (Figure 3).
For applications it is interesting to note that none of the above equations (8) or (12) contains any explicit energy dependence.
3. Models and Simulations
3.1. Ray Tracing Calculations
So far the RML was treated as a single reflecting surface. In order to investigate the impact of the beam penetration into the ML structure, both numerical and analytical ray tracing simulations were developed [18–20]. An analytical approach is shown in Figure 4 for the case of a parabolic RML. It can be shown that the elliptic case is virtually identical for the high ellipse eccentricities that occur in synchrotron applications. The RML stack is approximated by a single thick layer of average optical index . Multiple reflections and refractions within the stack are neglected, and refraction at the surface is treated in linear approximation. The aim is to calculate the principal intersection points of rays that penetrate into the structure and that are reflected by the second interface . An ensemble of parallel incoming beams forms an envelope curve of enhanced intensity (caustic) near the ideal focus .
The analytical derivation is carried out to first order in , which is a small quantity for hard Xrays. The resulting intersection points of the outgoing beams with the optical axis X on the one hand and with the focal plane Y on the other can be written as with and being as indicated in Figure 4. Rays impinging at lower grazing angles produce stronger aberrations in both x and y directions. Both the layer thickness and the refracting power of the medium amplify the effect. The situation is illustrated in Figure 5 for a source distance , an image distance , and a photon energy .
Figure 6 shows the intersection points Y with the focal plane given in Figure 5 as a function of the grazing angle . The purple line represents a calculation using the approximation (15) while the red line was obtained by exact numerical ray tracing on the same structure. Both curves are in good agreement for higher angles. Closer to the critical angle , where the linear approximation of Snell’s law fails, the curves diverge. The solid circles are independent calculations made at Osaka University using a different ray tracing code. There is a reasonably good agreement between the two methods [29]. The examples in Figures 5 and 6 were calculated for a parameter set corresponding to realistic focusing setups at a 3rd generation synchrotron beamline. The expected aberrations in the focal plane would be lower than 10 nm for a typical grazing angle of .
The analytical approach has the benefit that equations (15) allow for an estimate of chromatic aberrations and for a comparison with other types of focusing elements. Chromaticity enters through the energy dependent refractive decrement It can be shown that [20], for a given optical element, the tolerated bandwidth for a given focal spot size (FWHM) takes the form summarized in Table 1. It has to be underlined that, in contrast to FZPs and CRLs, where the macroscopic focal length f appears in the denominator, only the small aberration enters the formula for the RML, which makes this element nearly achromatic. The physical background of the different impact on the chromaticity is clearly visible. For FZPs and CRLs the phase shift induced by the optical elements is at the origin of their focusing properties, and chromaticity is therefore unavoidable. In the case of RMLs, however, it is rather a parasitic effect that should be suppressed as much as possible.

It has to be emphasized that the above analysis is based on ideal elliptically or parabolically shaped RMLs. It is known, however, that the optimum shapes of focusing RMLs differ from these perfect mathematical curves. It is a general practice to assume that RMLs have to fulfill Bragg’s law (9) locally [30, 31] in order to provide constructive interference at the same photon energy and at any point along the optical surface. The ML dspacings and consequently the interface positions in space change as the optical indices differ from unity. There is strong evidence that this refraction correction implicitly leads to the optimum RML shape [18]. Full wave optical computations will be required to derive structures that minimize or suppress aberrations. They will also have to include effects caused by figure errors and by the finite source size.
3.2. Wave Optical Calculations
A first step to minimize the aberration of RMLs is wave optical calculations for a single reflecting surface and the related wave propagation in free space from the source to the mirror and further to the focal plane. An isotropic and monochromatic point source emits a number of rays (typically ten thousands), which are traced onto the mirror’s surface S, where they are reflected. The amplitude is multiplied by the corresponding Fresnel coefficient, so the outgoing field A(s) is known in amplitude and phase. In a region around the nominal focus point the intensity distribution is calculated by a numerical solution of Kirchhoff’s integral of diffraction, treated here in two dimensions corresponding to the plane of incidence :
Figure 7 shows the field intensity emitted by an isotropic point source and focused by an ideal elliptic Pd mirror at a photon energy of 8 keV. Source and image distances are and , respectively. The angle of incidence at the mirror center was set to 4 mrad = 0.2.
The geometrical cone of the converging beam, the diffraction waist, as well as interference features from the finite aperture are clearly visible. In the present case, a diffraction limited spot size of 71 nm FWHM was obtained. The model was refined to treat extended sources as well as figure errors on the mirror surface. Figure 8 shows the performance of the same Pd mirror as in Figure 7, this time assuming a Gaussian source of 85 m FWHM and a figure error of about 5 nm peaktovalley (PV). The simulation was made for an ensemble of partially coherent point sources representing a spatially extended source. The focal spot size has increased to 190 nm. The blurring and the disappearance of the interference fringes are mainly due to the finite and only partially coherent source.
It is of interest to compare wave optical results with those that can be obtained by purely geometrical considerations as outlined in Section 2. Figure 9 shows the diffraction limited focal spot size versus the illuminated footprint along a mirror, calculated from (12) (blue curve), from the exact NA (purple curve), and using a wave optical code (red curve). The parameters used in the calculations correspond to those given in Figure 5. The incoming beam aperture is increased while remaining centered at .
It can be seen that the geometrical approach assuming a flat numerical aperture NA (thin lens approximation) agrees well with the wave optical calculations for short mirror sections. As the length of the footprint becomes comparable with the image distance, the geometrical approach underestimates the spot size. In the latter case, the strongly asymmetric aperture can no longer be approximated by a thin lens. Equation (12) returns systematically lower values than the other methods, mainly due to the neglected refraction correction. Despite the observed differences, all three descriptions return values that are consistent within the given degree of simplification.
4. Technological Options
4.1. Overview
At present, three alternative approaches have been developed to produce curved RMLs for applications at 3rd generation synchrotron sources. APS (USA) investigated profile coatings to provide optical surfaces with the required fixed focus elliptic curvature [6], essentially without ML coatings. Work at the ESRF (France) concentrated on ML coated flat mirrors integrated into dynamical benders to approach the optimum curvature [32]. SPring8 and the University of Osaka (Japan) developed deterministic polishing techniques [33] and differential deposition to obtain ultra precise surfaces and the required ML coatings [34]. KB focusing systems for Xrays, though mainly based on TRMs, are commercially available from various suppliers. A nonexhaustive list is given in [35–41].
4.2. ML Deposition
The principal challenge in the fabrication of an RML is the accurate deposition of the dspacing gradient that fulfills (9). In order to guarantee both uniform reflectivity and phase the tolerated thickness errors of the individual layers must be far below the width of the ML Bragg peak. In practice, the accuracy must be of the order of 0.5% or better. Today, the most common deposition method is magnetron sputtering. It combines high deposition rates (0.1–1.0 nm/s) with reasonably stable operation over a full coating cycle. The high kinetic energy of sputtered particles favors the formation of smooth and dense thin layers. Relative motion between sputter source and substrate, sometimes complemented by the insertion of masks, is the method of choice to produce the correct thickness gradient. This technique is also known as differential deposition.
4.3. Alternative Solutions
An interesting alternative focusing scheme was proposed by Montel [42] where the two perpendicular reflecting surfaces are positioned side by side. This approach offers a more compact arrangement and enables equal demagnification in both directions. The critical issue is the corner line where both mirrors are in contact. Single reflection optics with ellipsoidal shape is attractive thanks to its higher transmission and potentially larger collection angle. The fabrication is particularly challenging in terms of both figuring and polishing. Both types of optics are commercially available [43–46] and are mainly used in combination with laboratory Xray sources.
5. Experimental Progress
With Pt coated mirrors diffraction limited spot sizes of about 25 nm were measured [47]. More recently, this value could be significantly improved by the use of Pt/C RMLs. During experiments at Spring8, a line focus of 7 nm could be obtained with a monochromatic beam at 20 keV [10]. This remarkable performance was achieved by inserting an online wave front correction mirror to account for residual figure errors on the focusing element itself [48]. An example of a line profile is given in Figure 10, containing both experimental data and a numerical simulation (courtesy H. Mimura). It is important to note that these experimental findings are in line with the theoretical estimates made in Sections 2 and 3.
The main drawback of fixed focus KBs is their lack of flexibility required for operation at variable energies. Dynamical setups based on bent RMLs, that offer the possibility to switch the photon energy, have achieved best results of about 45 nm [49]. So far, figure errors and mechanical vibrations were identified as principal obstacles to reach smaller spots.
6. Comparison with Other Focusing Elements
It is not within the scope of this paper to provide a full comparison of the various optical elements presently used to focus hard Xrays. Nevertheless, Table 2 gives a brief overview on the typical experimental performance along some key parameters such as the smallest achieved spot size , typical energy range , intrinsic bandwidth , and practical aspects concerning geometry and preferred mode of operation. It is not exhaustive and subject to ongoing evolution. The corresponding references and abbreviations were introduced in Section 1. A similar summary was published in [50]. A comprehensive overview on the different types of Xray optics can be found in [51].

7. Applications
Over the last 10 years, at ESRF, more than 20 RML based KB focusing devices were delivered to different beamlines. They provide focal spots from 100 m down to less than 100 nm to the user community. In contrast to crystal optics the high reflectivity and the broad bandwidth of the ML elements allow for high intensity, making them the system of choice for flux limited experiments. The most brilliant ESRF KB nanofocusing setups provide flux densities of about ph/s/mA/nm^{2}. Novel techniques like projection tomography [52] and fluorescence imaging [53] on the nanometer scale have evolved thanks to the excellent performance of this optical system. They provide new opportunities for various scientific and engineering disciplines.
8. Summary and Outlook
KB focusing devices based on graded RML coatings play a major role in present day Xray optics, in particular on 3rd generation synchrotron beamlines. Thanks to their ML coating their numerical aperture can be increased by a factor of about 5 compared with TRMs. They are widely used and have generated diffraction limited focal spots down to 7 nm FWHM. While both design and fabrication of curved RMLs have developed rapidly over the last decade, a full wave optical description is still lacking.
Several approaches to further reduce the spot size might be considered. The use of higherorder Bragg peaks would increase the numerical aperture, on the expense of reflectivity and flux. A real technological breakthrough would be the use of two coherently illuminated curved RMLs facing each other. A rough estimate shows [9] that such a setup would reduce the diffraction limit by a factor 4 compared to a single element. If one adds a second RML on each side, leading to two successive reflections (Wolter geometry), the gain in aperture would be about 8.
From a fundamental point of view the 1 nm limit seems to be within reach. The task to align several optical elements to within atomic distances, however, represents a major technological challenge.
Acknowledgments
The authors would like to thank the members of the Department of Precision Science and Technology, Graduate School of Engineering, Osaka University (Japan) for their kind support. The research leading to these results has received funding from the European Community’s Seventh Framework Programme (FP7/2007–2013) under grant agreement no. 226716.
References
 W. Chao, B. D. Harteneck, J. A. Liddle, E. H. Anderson, and D. T. Attwood, “Soft Xray microscopy at a spatial resolution better than 15 nm,” Nature, vol. 435, no. 7046, pp. 1210–1213, 2005. View at: Publisher Site  Google Scholar
 C. G. Schroer, O. Kurapova, J. Patommel et al., “Hard xray nanoprobe based on refractive xray lenses,” Applied Physics Letters, vol. 87, no. 12, Article ID 124103, pp. 1–3, 2005. View at: Publisher Site  Google Scholar
 A. Jarre, C. Fuhse, C. Ollinger, J. Seeger, R. Tucoulou, and T. Salditt, “Twodimensional hard xray beam compression by combined focusing and waveguide optics,” Physical Review Letters, vol. 94, no. 7, Article ID 074801, 2005. View at: Publisher Site  Google Scholar
 D. H. Bilderback, S. A. Hoffman, and D. J. Thiel, “Nanometer spatial resolution achieved in hard xray imaging and Laue diffraction experiments,” Science, vol. 263, no. 5144, pp. 201–203, 1994. View at: Google Scholar
 H. Mimura, S. Matsuyama, H. Yumoto et al., “Hard Xray diffractionlimited nanofocusing with KirkpatrickBaez mirrors,” Japanese Journal of Applied Physics, vol. 44, no. 16–19, pp. L539–L542, 2005. View at: Publisher Site  Google Scholar
 W. Liu, G. E. Ice, J. Z. Tischler et al., “Short focal length KirkpatrickBaez mirrors for a hard xray nanoprobe,” Review of Scientific Instruments, vol. 76, no. 11, Article ID 113701, pp. 1–6, 2005. View at: Publisher Site  Google Scholar
 H. C. Kang, J. Maser, G. B. Stephenson et al., “Nanometer linear focusing of hard X rays by a multilayer Laue lens,” Physical Review Letters, vol. 96, no. 12, Article ID 127401, pp. 1–4, 2006. View at: Publisher Site  Google Scholar
 H. C. Kang, H. Yan, R. P. Winarski et al., “Focusing of hard Xrays to 16 nanometers with a multilayer Laue lens,” Applied Physics Letters, vol. 92, no. 22, Article ID 221114, 2008. View at: Publisher Site  Google Scholar
 Ch. Morawe, O. Hignette, P. Cloetens et al., “Graded multilayers for focusing hard X rays below 50nm,” in Advances in XRay/EUV Optics, Components, and Applications, vol. 6317 of Proceedings of SPIE, 63170F. View at: Publisher Site  Google Scholar
 H. Mimura, S. Handa, T. Kimura et al., “Breaking the 10 nm barrier in hardXray focusing,” Nature Physics, vol. 6, pp. 122–125, 2009. View at: Publisher Site  Google Scholar
 F. Pfeiffer, C. David, J. F. van der Veen, and C. Bergemann, “Nanometer focusing properties of Fresnel zone plates described by dynamical diffraction theory,” Physical Review B, vol. 73, no. 24, Article ID 245331, 2006. View at: Publisher Site  Google Scholar
 C. G. Schroer and B. Lengeler, “Focusing hard X rays to nanometer dimensions by adiabatically focusing lenses,” Physical Review Letters, vol. 94, no. 5, Article ID 054802, 2005. View at: Publisher Site  Google Scholar
 K. Yamauchi, K. Yamamura, H. Mimura et al., “Nearly diffractionlimited line focusing of a hardXray beam with an elliptically figured mirror,” Journal of Synchrotron Radiation, vol. 9, no. 5, pp. 313–316, 2002. View at: Publisher Site  Google Scholar
 C. M. Kewish, L. Assoufid, A. T. Macrander, and J. Qian, “Waveoptical simulation of hardxray nanofocusing by precisely figured elliptical mirrors,” Applied Optics, vol. 46, no. 11, pp. 2010–2021, 2007. View at: Publisher Site  Google Scholar
 C. G. Schroer, “Focusing hard X rays to nanometer dimensions using Fresnel zone plates,” Physical Review B, vol. 74, no. 3, Article ID 033405, 2006. View at: Publisher Site  Google Scholar
 H. Yan, J. Maser, A. Macrander et al., “TakagiTaupin description of Xray dynamical diffraction from diffractive optics with large numerical aperture,” Physical Review B, vol. 76, no. 11, Article ID 115438, 2007. View at: Publisher Site  Google Scholar
 H. Mimura, S. Matsuyama, H. Yumoto et al., “Reflective optics for sub10 nm hard Xray focusing,” in Advances in XRay/EUV Optics and Components II, vol. 6705 of Proceedings of SPIE, 67050L. View at: Publisher Site  Google Scholar
 H. Mimura, S. Handa, Ch. Morawe et al., “Raytracing analysis in aberration of a laterally graded multilayer mirror,” Nuclear Instruments and Methods in Physics Research A, vol. 616, no. 23, pp. 251–254, 2010. View at: Publisher Site  Google Scholar
 J.P. Guigay, Ch. Morawe, V. Mocella, and C. Ferrero, “An analytical approach to estimating aberrations in curved multilayer optics for hard xrays: 1. Derivation of caustic shapes,” Optics Express, vol. 16, no. 16, pp. 12050–12059, 2008. View at: Publisher Site  Google Scholar
 Ch. Morawe, J.P. Guigay, V. Mocella, and C. Ferrero, “An analytical approach to estimating aberrations in curved multilayer optics for hard xrays: 2. Interpretation and application to focusing experiments,” Optics Express, vol. 16, no. 20, pp. 16138–16150, 2008. View at: Publisher Site  Google Scholar
 P. Kirkpatrick and A.V. Baez, “Formation of optical images by Xrays,” Journal of the Optical Society of America, vol. 38, p. 766, 1984. View at: Google Scholar
 E. Abbe, “über die Bedingungen des Aplanatismus der Linsensysteme,” Jenaisch. Ges. Med. Naturw., vol. 129, 1879. View at: Google Scholar
 H. Wolter,, “Spiegelsysteme streifenden Einfalls als abbildende Optiken für Röntgenstrahlen,” Annals of Physics, vol. 445, p. 94, 1952. View at: Google Scholar
 J.A. Nielsen and D. McMorrow, Elements of Modern XRay Physics, John Wiley & Sons, London, UK, 2001.
 J. Underwood and T. W. Barbee Jr., “Layered synthetic microstructures as Bragg diffractors for X rays and extreme ultraviolet: theory and predicted performance,” Applied Optics, vol. 20, no. 17, pp. 3027–3034, 1981. View at: Google Scholar
 Ch. Morawe, “Graded multilayers for synchrotron optics,” in Proceedings of the 9th International Conference on Synchrotron Radiation Instrumentation, vol. 879, pp. 764–769, January 2007. View at: Publisher Site  Google Scholar
 Ch. Morawe, E. Ziegler, J.C. Peffen, and I. V. Kozhevnikov, “Design and fabrication of depthgraded Xray multilayers,” Nuclear Instruments and Methods in Physics Research A, vol. 493, no. 3, pp. 189–198, 2002. View at: Publisher Site  Google Scholar
 Ch. Morawe, C. Borel, E. Ziegler, and J.C. Peffen, “Application of double gradient multilayers for focusing,” in XRay Sources and Optics, vol. 5537 of Proceedings of SPIE, pp. 115–126, November 2004. View at: Publisher Site  Google Scholar
 Ch. Morawe, J.P. Guigay, V. Mocella et al., “Aberrations in curved xray multilayers,” in Advances in XRay/EUV Optics and Components III, vol. 7077 of Proceedings of SPIE, September 2008, 70770T. View at: Publisher Site  Google Scholar
 M. Schuster and H. Göbel, “Parallel beam coupling into channelcut monochromators using curved graded multilayers,” Journal of Physics D: Applied Physics, vol. 28, p. A270, 1995. View at: Google Scholar
 Ch. Morawe, P. Pecci, J.C. Peffen, and E. Ziegler, “Design and performance of graded multilayers as focusing elements for Xray optics,” Review of Scientific Instruments, vol. 70, no. 8, pp. 3227–3232, 1999. View at: Google Scholar
 O. Hignette, P. Cloetens, G. Rostaing, P. Bernard, and Ch. Morawe, “Efficient sub 100 nm focusing of hard X rays,” Review of Scientific Instruments, vol. 76, no. 6, pp. 1–5, 2005. View at: Publisher Site  Google Scholar
 Y. Mori, K. Yamauchi, K. Yamamura et al., “Development of plasma chemical vaporization machining and elastic emission machining systems for coherent Xray optics,” in XRay Mirrors, Crystals, and Multilayers, vol. 4501 of Proceedings of SPIE, pp. 30–42, July 2001. View at: Publisher Site  Google Scholar
 H. Mimura, S. Matsuyama, H. Yumoto et al., “Reflective optics for sub10 nm hard Xray focusing,” in Advances in XRay/EUV Optics and Components II, vol. 6705 of Proceedings of SPIE, August 2007, 67050L. View at: Publisher Site  Google Scholar
 http://www.crystalscientific.com/?KBfocus=.
 http://www.idtnet.co.uk/.
 http://www.irelec.fr/en/produit.asp?idproduit=39.
 http://www.jtec.co.jp/english/focusing/index.html.
 http://www.seso.com/uk/index.htm.
 http://www.zeiss.de/en.
 http://www.winlightx.com/.
 M. Montel, “Xray microscopy with catamegonic roof mirrors,” in XRay Microscopy and Microradiography, pp. 177–185, Academic Pres, New York, NY, USA, 1957. View at: Google Scholar
 http://www.axodresden.de/index1.html.
 http://www.incoatec.de/.
 http://www.rigaku.com/optics.
 http://www.xenocs.com/accueil.htm.
 H. Mimura, H. Yumoto, S. Matsuyama et al., “Efficient focusing of hard x rays to 25 nm by a total reflection mirror,” Applied Physics Letters, vol. 90, no. 5, Article ID 051903, 2007. View at: Publisher Site  Google Scholar
 T. Kimura, S. Handa, H. Mimura et al., “Wavefront control system for phase compensation in hard Xray optics,” Japanese Journal of Applied Physics, vol. 48, no. 7, Article ID 072503, 2009. View at: Publisher Site  Google Scholar
 O. Hignette, P. Cloetens, Ch. Morawe et al., “Nanofocusing at ESRF using graded multilayer mirrors,” in Proceedings of the 9th International Conference on Synchrotron Radiation Instrumentation (SRI '07), vol. 879, pp. 792–795, January 2007. View at: Publisher Site  Google Scholar
 A. Snigirev and I. Snigireva, “Hard Xray microoptics,” in Modern Developments in XRay and Neutron Optics, A. Erko, M. Idir, T. Krist, and A. G. Michette, Eds., pp. 255–285, Springer, Berlin, Germany, 2008. View at: Google Scholar
 http://www.xrayoptics.com/.
 R. Mokso, P. Cloetens, E. Maire, W. Ludwig, and J.Y. Buffìre, “Nanoscale zoom tomography with hard x rays using KirkpatrickBaez optics,” Applied Physics Letters, vol. 90, no. 14, Article ID 144104, 2007. View at: Publisher Site  Google Scholar
 P. Bleuet, A. Simionovici, L. Lemelle et al., “Hard Xrays nanoscale fluorescence imaging of Earth and Planetary science samples,” Applied Physics Letters, vol. 92, no. 21, Article ID 213111, 2008. View at: Publisher Site  Google Scholar
Copyright
Copyright © 2010 Christian Morawe and Markus Osterhoff. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.