Research Article  Open Access
Xue Dong, Michael Petrongolo, Tianye Niu, Lei Zhu, "LowDose and ScatterFree ConeBeam CT Imaging Using a Stationary Beam Blocker in a Single Scan: Phantom Studies", Computational and Mathematical Methods in Medicine, vol. 2013, Article ID 637614, 8 pages, 2013. https://doi.org/10.1155/2013/637614
LowDose and ScatterFree ConeBeam CT Imaging Using a Stationary Beam Blocker in a Single Scan: Phantom Studies
Abstract
Excessive imaging dose from repeated scans and poor image quality mainly due to scatter contamination are the two bottlenecks of conebeam CT (CBCT) imaging. Compressed sensing (CS) reconstruction algorithms show promises in recovering faithful signals from lowdose projection data but do not serve well the needs of accurate CBCT imaging if effective scatter correction is not in place. Scatter can be accurately measured and removed using measurementbased methods. However, these approaches are considered unpractical in the conventional FDK reconstruction, due to the inevitable primary loss for scatter measurement. We combine measurementbased scatter correction and CSbased iterative reconstruction to generate scatterfree images from lowdose projections. We distribute blocked areas on the detector where primary signals are considered redundant in a full scan. Scatter distribution is estimated by interpolating/extrapolating measured scatter samples inside blocked areas. CSbased iterative reconstruction is finally carried out on the undersampled data to obtain scatterfree and lowdose CBCT images. With only 25% of conventional fullscan dose, our method reduces the average CT number error from 250 HU to 24 HU and increases the contrast by a factor of 2.1 on Catphan 600 phantom. On an anthropomorphic head phantom, the average CT number error is reduced from 224 HU to 10 HU in the central uniform area.
1. Introduction
Onboard conebeam CT (CBCT) is being increasingly implemented on radiation therapy machines for accurate patient positioning and tumor targeting in imageguided radiation therapy (IGRT). The use of CBCT increases patient setup accuracy and also opens possibilities of CBCTbased accurate tumor delineation and therapeutic dose calculation. Nevertheless, the wide application of CBCT in IGRT is limited by excessive imaging dose and poor image quality.
The repeated CBCT scans during the treatment procedure produce high dose to healthy organs. It has been reported that the dose delivered from a CBCT system could be as high as 5~10 cGy per scan and 100~300 cGy per treatment course [1–6]. Although radiotherapy patients are being exposed to higher radiation doses for cancer treatment, the additional CBCT dose leads to skin burns cataracts, and increased risks of radiationinduced cancer or genetic defects [1]. Moreover, the CBCT dose is particularly risky for radiationsensitive groups [5]. For example, CBCTguided radiation therapy is essentially prohibitive for pediatric patients, resulting in suboptimal treatment outcomes. Patient dose can be lowered by optimizing both hardware and software designs of the CT systems. Existing approaches include optimization of data acquisition protocols (e.g., automatic exposure control), improvement of detector quantum efficiency, regionofinterest (ROI) reconstruction [7] from reduced projections, and noise suppression with degraded spatial resolution. However, after continuous development of CT systems for decades, further dose reduction from these techniques is limited or costly. Decreasing the total number of incident photons of each projection ray (i.e., mAs) and reducing the number of Xray projections also lower the patient dose but with degraded image qualities in the conventional filtered back projection (FBP) reconstruction [8]. The recent advances in compressed sensing (CS) enable accurate CT image recovery from undersampled data [9]. Compared to the analytical algorithms, total variation (TV) based CS methods [10, 11] have demonstrated significant improvements in both fanbeam and conebeam CT reconstruction especially when projection data are undersampled with sparse views [11–13] or with missing data in a single view [11]. These reconstruction algorithms minimize the TV of the CT image constrained by the data fidelity and image nonnegativity, which show promises in reducing CT dose without significantly degrading image qualities.
Besides excessive patient dose, CBCT images are also subject to severe contamination from scatter radiation. Scatter signals induce large image artifacts and CT number nonlinearity, which limit the applications of CBCT. For a middlesize human torso, the average scattertoprimary ratio (SPR) is around 2~3, which leads to CT number errors up to 350 HU [14–17]. Extensive studies have been conducted on scatter correction techniques. These published techniques can be divided into two major categories, based on whether scatter signals are directly measured or not. Nonmeasurementbased methods either prevent scattered radiation from reaching the detector (e.g., using an antiscatter grid [18, 19], limiting the field of view (FOV), and increasing the air gap between the object and the detector [20]) or predict the scatter distribution (using, e.g., analytical modeling [21], modulation methods [22–25], and Monte Carlo (MC) simulation [26, 27]). These methods improve the image quality to a certain extent, but their performances are limited in clinical applications [28]. An antiscatter grid results in primary signal loss, thus, increasing image noise and degrades image qualities [18, 19]. The airgap between the object and the detector is limited by the size of operation room [20]. Monte Carlo simulation generates accurate scatter signals but is computationally intense [26, 27]. On the other hand, methods of direct scatter measurement conveniently obtain accurate scatter estimates with negligible computational cost [17]. In the measurementbased method, a beam blocker is typically inserted between the Xray source and the object, and scatter signals are estimated inside the detector shadows of the beam blocker [15, 29–31], where primary signals are fully attenuated. The scatter distribution of the whole field is then obtained via interpolation/extrapolation on the scatter samples inside the shadows, since scatter distributions have dominant lowfrequency components [15, 32, 33]. The method achieves accurate scatter estimation without prior knowledge of Xray source, object, imaging geometry, and is easy to implement. Nonetheless, primary signal loss is inevitable due to the insertion of the beam blocker. As a result, severe image artifacts appear in the conventional [34] reconstruction if the missing primary signals are not compensated for [17]. Two projections per view, one with the blocker and the other without [14], or moving blockers during the scan [35], are designed to compensate for the primary loss. These hardware modifications complicate the data acquisition and increase scan time and patient dose. Recently, we developed a “crossingfinger”shape beam blocker, which makes use of the data redundancy condition in a 360deg fullfan CT scan. This method achieves accurate scatter estimation and reconstruction within one single scan and thus is considered clinically more attractive. Though demonstrated promise, the “crossingfinger”shape blocker is of complex structure, and the insertion of beam blocker complicates the FDK reconstruction algorithm.
For years research has been developed independently on dose reduction and scatter correction. Nevertheless, little effort has been devoted to exploit the full potential of image improvement from a combination of the above two schemes. Scatter measurement accurately corrects for scatter but leads to primary loss, which makes most of the measurementbased correction methods unpractical. CSbased iterative algorithm lowers imaging doses and obtains accurate reconstruction even on the insufficient data from sparse views or a reduced number of detector pixels. Considering the complimentary capabilities of these two approaches, in this work, we propose to use an improved stationary beam blocker in the CBCT system for simultaneous dose reduction and scatter measurement and an iterative algorithm for accurate reconstruction on the projections with missing data in a single scan. The new method explores the strengths of measurementbased scatter correction and iterative reconstruction while eliminates their shortcomings and obtains lowdose and scatterfree CBCT images.
In the new method, the lead strips of the blocker are placed in the longitudinal direction and located asymmetrically with respect to the central longitudinal line of the detector. If one ray is blocked by the strip, its conjugate is still measured after around 180deg rotation even if it is in the offplane. We insert the beam blocker between the Xray source and the object, where scatter distribution is obtained by interpolation/extrapolation on the scatter samples inside the strip shadow. The insertion of blocker also reduces patient dose since Xray primary signals are attenuated [36]. We further reduce the patient dose by decreasing the projection number. Our recently developed CSbased iterative reconstruction, accelerated barrier optimization for compressed sensing (ABOCS) [8], is carried out on the blocked data to obtain scatterfree and lowdose CBCT images. We evaluate the performance of the method on the Catphan 600 phantom and an anthropomorphic head phantom.
2. Method
2.1. Blocker Design
In a circular conebeam CT scan, one projection ray can be specified by , where and are the angles of the ray in the transverse and axial directions, respectively, and is the projection angle of the source. It can be easily verified that no projection rays are redundant in such geometry except those in the midplane (i.e., ). Nevertheless, if we employ a commonly used approximation of small cone angle (i.e., for the whole projection), the redundant rays have the following relationships: with a full rotation and full object coverage; half of the CBCT projection data are considered to be redundant. Under the smallconeangle approximation, each projection ray in a CBCT full scan has a corresponding redundant ray measured from the opposing direction. The two lines are referred to as a conjugate ray pair and this condition is referred to as the data redundancy. We can therefore block some of these redundant rays for other purposes (e.g., scatter measurement) while still maintaining an accurate reconstruction [37]. No hardware compensation for the missing primary data is necessary and the data acquisition is complete with one singlescan.
Guided by this principle, we place lead strips in the longitudinal direction, which is perpendicular to the rotation plane, to block only redundant rays for scatter measurement. One ray blocked by the strip is measured through its conjugate after around 180deg rotation. The beam blocker is designed to block less than 50% of full illuminated field and are placed asymmetrically with respect to the central longitudinal line of the detector, such that at least one ray from its conjugate ray pair can be measured on the detector. Note that, the central longitudinal line of the detector is always left unblocked to avoid the missing rays passing through the object center.
Figure 1 shows the geometry of the proposed method and our experimental setup. The designed blocker is placed between the Xray source and the object. The lead strips are placed along the longitudinal direction and uniformly distributed in the lateral direction. The strips have a thickness of around 3 mm and attenuate more than 99.99% of incident Xray photons. Only scatter samples are measured inside the shadows on the detector. Besides the strip placement, two more parameters are needed in the blocker design: sampling period () and strip width (). To guarantee the measurement of at least one ray from its conjugate pair, we choose a relatively large sampling period ( mm on the detector) based on the observed maximum spatial frequency of scatter signals in our previous studies [15, 17, 38] as well as in the literature [39]. The strip width cannot be too small since the penumbra effects on the strips limit scatter measurement accuracy [38]. Moreover, wider blocker contributes more to the dose reduction. Based on our previous study [16], is chosen as about 17 mm on the detector, which blocks 33% of the illuminated area.
2.2. Scatter Estimation and Correction
The tabletop system geometry is shown in Figure 1. As shown in our previous studies [16, 17, 24] and the literature [39], the insertion of the beam blocker does not greatly perturb the spatial frequency spectrum of scatter in conebeam projections and scatter is still predominantly lowfrequency. The whole field scatter distribution is therefore accurately estimated using interpolation/extrapolation on the measured samples. To avoid the penumbra effect of the strips, only the measured data inside the central twothird of the strip shadows are used in the scatter estimation. Since the lead strips cover the whole blocker in the longitudinal direction, a onedimensional (1D) cubic interpolation is carried out on each lateral line to estimate the scatter distribution over the whole detector area. The estimated scatter is then subtracted from the raw projection to generate the scattercorrected CBCT projection.
2.3. Reconstruction on the Incomplete Data
These corrected CBCT projections are incomplete due to the insertion of the blocker and the angular undersampling. Severe artifacts therefore appear in the conventional FDK reconstruction. To improve the image quality, we first compensate for the missing primary in the blocked area using their conjugate rays. As described by (1), the two detector points corresponding to the conjugate ray pair are symmetric with respect to the detector central longitudinal line, and their projection angle has a difference of . Due to the discretization of the data acquisition in both spatial and angular directions, the missing primary is compensated for using its conjugate point by interpolating on the scattercorrected sinogram.
An inhouse CSbased iterative reconstruction is applied to further improve the image quality. The algorithm is referred to as the accelerated barrier optimization for compressed sensing (ABOCS) reconstruction algorithm, which minimizes the image TV term with data fidelity and nonnegativity constraints [8]. ABOCS formulates the TV minimization constrained by the data fidelity into a form similar to that of the conventional TV regularization but with an automatically adjusted penalty weight. The automatic penalty weight is controlled by the data fidelity tolerance, which is estimated from the raw projections according to the Poisson statistics and the data error in the current iteration. Consistent reconstruction performances are achieved using the same algorithm parameters on scans with different noise levels and/or on different objects. The problem is then solved efficiently by gradient projection with an adaptive BarzilaiBorwein stepsize selection scheme. Readers are referred to [8] for more details. Note that, image noise increases significantly after scatter correction [14]. An additional penalized weighted leastsquares (PWLS) algorithm [14] is performed to reduce the noise in the reconstructed images.
2.4. Evaluation
The performance of the proposed method is evaluated on the Catphan 600 phantom with a diameter of 200 mm (The Phantom Laboratory, Salem, NY) and an anthropomorphic head phantom on our CBCT tabletop system. The geometry of this system exactly matches that of a Varian Onboard Imager (OBI) CBCT system on the TrueBeam radiation therapy machine. A detailed system configuration is described in [16]. The lead sheet of the designed blocker is first shaped using a waterjet cutting system. To improve the mechanical strength of the blocker, the lead is then sandwiched between two layers of thin steel (~0.2 mm) using JB WELD epoxy adhesive (http://www.grainger.com/).
CBCT images are compared with and without the proposed method. A total of 655 projections are acquired for the conventional FDK reconstruction. Fewview projection data are generated from the 655 projections with an evenly distributed angular spacing. The estimated dose reduction ratio is calculated based on the number of measured projection lines [16]. In both phantom studies, we use 219 projections and block 33% of illuminated area in each projection, resulting in the dose reduction ratio of around 75%. The proposed method is also compared with lowdose CBCT without scatter correction, where 163 projections are used, resulting in 75% dose reduction. Note that, the scatter estimation is also performed on the sparse projections.
For a quantitative error analysis, an additional set of projections is acquired with a fanbeam geometry, which narrows the collimator open width to around 10 mm on the detector for inherent scatter suppression. The resultant images are used as references. Image quality metrics are used to quantitatively evaluate the performance of the proposed method. For the selected ROIs, the CT number error is calculated as the square root of the mean square error (RMSE) and defined as where represents the index of ROI and is the mean reconstructed value inside the ROI and the value in the reference image and is the total number of ROIs. The image contrast is calculated as where is the mean reconstructed value inside the ROI and is the mean reconstructed value in the surrounding area.
3. Results
3.1. Catphan 600 Phantom Results
Figure 2 shows the 1D horizontal profiles of scatter signals, raw projections, and line integrals of one projection on the Catphan 600 phantom. The reference scatter signals are obtained as the difference of the conebeam and fanbeam projections. As seen in Figure 2(a), the estimated scatter profile using our method matches well with that of the measurement in the central region pixels (250–800) with an estimation error of less than 6.5%. Relatively large deviations are found around and outside the phantom boundary. However, the intensity of primary signals in these areas is high, which leads to a negligible estimation error of line integral. Figure 2(b) shows the line integrals with and without the proposed method. The blue circles indicate the scatter corrected primary signals measured in the illuminated area. The missing primary signals due to the insertion of the beam blocker are compensated with their conjugate rays analytically and shown as the green triangles in Figure 2(b). Our method significantly enhances the intensities of the line integrals, which are close to those of the ground truth, that is, fanbeam CT.
(a)
(b)
Figure 3 shows the reconstructed image with and without the proposed lowdose and scatterfree CBCT imaging scheme. Without scatter correction, the ABOCS reconstruction reduces the dose by 75%, however, severe shading artifacts are still observed (see Figures 3(b) and 4). Our proposed method significantly suppresses the shading artifacts (see Figure 3(c)). After improvement, the image quality is comparable to that of the reference (i.e., fanbeam CT in Figure 3(d)). For the quantitative evaluation of the performance using our method, the average CT numbers and contrasts are calculated for the contrast rods in one of the phantom slice. The results are summarized in Table 1 using those from the fanbeam CT as the references. In the selected ROIs, the proposed method reduces the mean CT number error from over 250 HU to around 24 HU and increases the contrast by a factor of 2.1 on the average. Figure 4 shows the comparison of 1D profiles passing through two high contrast rods inside the phantom, as indicated by the red line in Figure 3(d).

(a)
(b)
(c)
(d)
3.2. Head Phantom Results
Figure 5 shows the axial views of the reconstructed head phantom images using the conventional FDK reconstruction, the lowdose ABOCS reconstruction, and the proposed method. The fullscan fanbeam CT image is generated as the ground truth. Similar to the Catphan 600 phantom results, the shading artifacts in Figures 5(a) and 5(b) are significantly suppressed with the proposed method. The mean CT number error is reduced from over 220 HU (Figures 5(a) and 5(b)) to 10 HU (Figure 5(c)) in the central uniform area as indicated by the white circle. The overall image uniformity of our result (Figure 5(c)) is close to that in the fanbeam result (Figure 5(d)), with only 25% radiation dose of a routine CBCT scan. The comparison of 1D profiles passing through the central horizontal line, as indicated by the red line in Figure 5(d), is shown in Figure 6.
(a)
(b)
(c)
(d)
4. Discussion
In this paper, we propose a lowdose and scatterfree CBCT imaging method in a single scan using a stationary beam blocker for scatter measurement and ABOCS reconstruction on the incomplete primary signals. Superior performance of the proposed method is demonstrated on the Capthan 600 phantom and an anthropomorphic head phantom.
For demonstration purposes, only 2D images are reconstructed in this work. In the future, we will extend our method to 3D reconstruction. Due to the huge size of system matrix, it is not practical to store the whole matrix in the computer memory for iterative CT reconstruction. Instead, we will formulate the multiplication of system matrix as a forward projection operation and speed up the calculation using hardware acceleration technique, for example, on a graphics processing unit (GPU) [12, 40]. The other issue with the 3D extension is that the cone angle can be as large as 6° the offplanes in the OBI system [41], which makes the small coneangle approximation less accurate. Nevertheless, we would like to emphasize that the artifacts stemming from a large cone angle are generic issues in circular CBCT [42, 43]. For example, the small cone angle approximation is also used in 3D FDK reconstruction [34], the current standard algorithm implemented on commercial systems. For our applications of scatter correction with reduced projection measurement, we have shown in our previous study that the enlarged cone angle leads to negligible image quality degradation on clinical CBCT systems [17]. We would expect a similar performance of offplane imaging for the proposed method in this paper.
5. Conclusion
In this paper, we propose a practical CBCT imaging method for dose reduction and scatter correction using a stationary blocker in a single scan. In the tabletop phantom studies, our method reduces the overall CT number error form 250 HU to less than 24 HU and increases the image contrast by a factor of 2.1 in the selected ROIs with only 25% dose of a conventional CBCT scan. Our method has the promise to become a practical solution for scatter correction and lowdose imaging on different CBCT systems.
Acknowledgments
This work is supported by the Varian Management Research Agreement Grant and partially by the Natural Science Foundation of China (NSFC Grant no. 81201091).
References
 D. J. Brenner and E. J. Hall, “Computed tomography—an increasing source of radiation exposure,” New England Journal of Medicine, vol. 357, no. 22, pp. 2277–2284, 2007. View at: Publisher Site  Google Scholar
 J. R. Sykes, A. Amer, J. Czajka, and C. J. Moore, “A feasibility study for image guided radiotherapy using low dose, high speed, cone beam Xray volumetric imaging,” Radiotherapy and Oncology, vol. 77, no. 1, pp. 45–52, 2005. View at: Publisher Site  Google Scholar
 S. P. Sorensen, P. E. Chow, S. Kriminski, P. M. Medin, and T. D. Solberg, “Imageguided radiotherapy using a mobile kilovoltage Xray device,” Medical Dosimetry, vol. 31, no. 1, pp. 40–50, 2006. View at: Publisher Site  Google Scholar
 M. J. Murphy, J. Balter, S. Balter et al., “The management of imaging dose during imageguided radiotherapy: report of the AAPM Task Group 75,” Medical Physics, vol. 34, no. 10, pp. 4041–4063, 2007. View at: Publisher Site  Google Scholar
 M. W. K. Kan, L. H. T. Leung, W. Wong, and N. Lam, “Radiation dose from cone beam computed tomography for imageguided radiation therapy,” International Journal of Radiation Oncology Biology Physics, vol. 70, no. 1, pp. 272–279, 2008. View at: Publisher Site  Google Scholar
 G. X. Ding and C. W. Coffey, “Radiation dose from kilovoltage cone beam computed tomography in an imageguided radiotherapy procedure,” International Journal of Radiation Oncology Biology Physics, vol. 73, no. 2, pp. 610–617, 2009. View at: Publisher Site  Google Scholar
 Y. Zou, X. Pan, and E. Y. Sidky, “Image reconstruction in regionsofinterest from truncated projections in a reduced fanbeam scan,” Physics in Medicine and Biology, vol. 50, no. 1, pp. 13–27, 2005. View at: Publisher Site  Google Scholar
 T. Niu and L. Zhu, “Accelerated barrier optimization compressed sensing (ABOCS) reconstruction for conebeam CT: phantom studies,” Medical Physics, vol. 39, no. 7, pp. 4588–4598, 2012. View at: Publisher Site  Google Scholar
 Y. Tsaig and D. L. Donoho, “Extensions of compressed sensing,” Signal Processing, vol. 86, no. 3, pp. 549–571, 2006. View at: Publisher Site  Google Scholar
 E. Y. Sidky and X. Pan, “Image reconstruction in circular conebeam computed tomography by constrained, totalvariation minimization,” Physics in Medicine and Biology, vol. 53, no. 17, pp. 4777–4807, 2008. View at: Publisher Site  Google Scholar
 E. Y. Sidky, C. M. Kao, and X. Pan, “Accurate image reconstruction from fewviews and limitedangle data in divergentbeam CT,” Journal of XRay Science and Technology, vol. 14, no. 2, pp. 119–139, 2006. View at: Google Scholar
 X. Jia, Y. Lou, R. Li, W. Y. Song, and S. B. Jiang, “GPUbased fast cone beam CT reconstruction from undersampled and noisy projection data via total variation,” Medical Physics, vol. 37, no. 4, pp. 1757–1760, 2010. View at: Publisher Site  Google Scholar
 Z. Tian, X. Jia, K. Yuan, T. Pan, and S. B. Jiang, “Lowdose CT reconstruction via edgepreserving total variation regularization,” Physics in Medicine and Biology, vol. 56, no. 18, pp. 5949–5967, 2011. View at: Publisher Site  Google Scholar
 L. Zhu, J. Wang, and L. Xing, “Noise suppression in scatter correction for conebeam CT,” Medical Physics, vol. 36, no. 3, pp. 741–752, 2009. View at: Publisher Site  Google Scholar
 L. Zhu, Y. Xie, J. Wang, and L. Xing, “Scatter correction for conebeam CT in radiation therapy,” Medical Physics, vol. 36, no. 6, pp. 2258–2268, 2009. View at: Publisher Site  Google Scholar
 X. Dong, X. Jia, T. Niu, and L. Zhu, “Lowdose and scatterfree conebeam CT imaging: a preliminary study,” in Medical Imaging 2012: Physics of Medical Imaging, vol. 8313 of Proceedings of the SPIE, February 2012. View at: Publisher Site  Google Scholar
 T. Niu and L. Zhu, “Scatter correction for fullfan volumetric CT using a stationary beam blocker in a single full scan,” Medical Physics, vol. 38, no. 11, pp. 6027–6038, 2011. View at: Publisher Site  Google Scholar
 M. Endo, T. Tsunoo, N. Nakamori, and K. Yoshida, “Effect of scattered radiation on image noise in cone beam CT,” Medical Physics, vol. 28, no. 4, pp. 469–474, 2001. View at: Publisher Site  Google Scholar
 J. H. Siewerdsen, D. J. Moseley, B. Bakhtiar, S. Richard, and D. A. Jaffray, “The influence of antiscatter grids on softtissue detectability in conebeam computed tomography with flatpanel detectors,” Medical Physics, vol. 31, no. 12, pp. 3506–3520, 2004. View at: Publisher Site  Google Scholar
 J. H. Siewerdsen and D. A. Jaffray, “Optimization of Xray imaging geometry (with specific application to flatpanel conebeam computed tomography),” Medical Physics, vol. 27, no. 8, pp. 1903–1914, 2000. View at: Publisher Site  Google Scholar
 J. M. Boone and J. A. Seibert, “An analytical model of the scattered radiation distribution in diagnostic radiology,” Medical Physics, vol. 15, no. 5, pp. 721–725, 1988. View at: Publisher Site  Google Scholar
 H. Gao, R. Fahrig, N. R. Bennett, M. Sun, J. StarLack, and L. Zhu, “Scatter correction method for Xray CT using primary modulation: phantom studies,” Medical Physics, vol. 37, no. 2, pp. 934–946, 2010. View at: Publisher Site  Google Scholar
 H. Gao, L. Zhu, and R. Fahrig, “Modulator design for Xray scatter correction using primary modulation: material selection,” Medical Physics, vol. 37, no. 8, pp. 4029–4037, 2010. View at: Publisher Site  Google Scholar
 L. Zhu, N. R. Bennett, and R. Fahrig, “Scatter correction method for Xray CT using primary modulation: theory and preliminary results,” IEEE Transactions on Medical Imaging, vol. 25, no. 12, pp. 1573–1587, 2006. View at: Publisher Site  Google Scholar
 J. S. Maltz, W. E. Blanz, D. Hristov, and A. BaniHashemi, “Cone beam Xray scatter removal via image frequency modulation and filtering,” in Proceedings of the 27th Annual International Conference of the Engineering in Medicine and Biology Society (IEEEEMBS '05), pp. 1854–1857, September 2005. View at: Google Scholar
 A. P. Colijn and F. J. Beekman, “Accelerated simulation of cone beam XRay scatter projections,” IEEE Transactions on Medical Imaging, vol. 23, no. 5, pp. 584–590, 2004. View at: Publisher Site  Google Scholar
 Y. Kyriakou, T. Riedel, and W. A. Kalender, “Combining deterministic and Monte Carlo calculations for fast estimation of scatter intensities in CT,” Physics in Medicine and Biology, vol. 51, no. 18, article 008, pp. 4567–4586, 2006. View at: Publisher Site  Google Scholar
 T. Niu and L. Zhu, “Overview of Xray scatter in Conebeam computed tomography and its correction methods,” Current Medical Imaging Reviews, vol. 6, no. 2, pp. 82–89, 2010. View at: Publisher Site  Google Scholar
 J. S. Maltz, B. Gangadharan, M. Vidal et al., “Focused beamstop array for the measurement of scatter in megavoltage portal and cone beam CT imaging,” Medical Physics, vol. 35, no. 6, pp. 2452–2462, 2008. View at: Publisher Site  Google Scholar
 R. Ning, X. Tang, and D. Conover, “Xray scatter correction algorithm for cone beam CT imaging,” Medical Physics, vol. 31, no. 5, pp. 1195–1202, 2004. View at: Publisher Site  Google Scholar
 F. C. Wagner, A. Macovski, and D. G. Nishimura, “Dualenergy Xray projection imaging: two sampling schemes for the correction of scattered radiation,” Medical Physics, vol. 15, no. 5, pp. 732–748, 1988. View at: Publisher Site  Google Scholar
 Y. Kyriakou, M. Meyer, and W. A. Kalender, “Technical note: comparing coherent and incoherent scatter effects for conebeam CT,” Physics in Medicine and Biology, vol. 53, no. 10, pp. N175–N185, 2008. View at: Publisher Site  Google Scholar
 T. Niu, M. Sun, J. StarLack, H. Gao, Q. Fan, and L. Zhu, “Shading correction for onboard conebeam CT in radiation therapy using planning MDCT images,” Medical Physics, vol. 37, no. 10, pp. 5395–5406, 2010. View at: Publisher Site  Google Scholar
 I. A. Feldkamp, L. C. Davis, and J. W. Kress, “Practical conebeam algorithm,” Journal of the Optical Society of America A, vol. 1, no. 6, pp. 612–619, 1984. View at: Google Scholar
 L. Zhu, N. Strobel, and R. Fahrig, “Xray scatter correction for conebeam CT using moving blocker array,” in Medical Imaging 2005: Physics of Medical Imaging, vol. 5745 of Proceedings of the SPIE, pp. 251–258, February 2005. View at: Publisher Site  Google Scholar
 S. Cho, T. Lee, J. Min, and H. Chung, “Feasibility study on manyview undersampling technique for lowdose computed tomography,” Optical Engineering, vol. 51, no. 8, Article ID 080501, 2012. View at: Publisher Site  Google Scholar
 R. Pua, J. Min, B. Yoo, K. W. Kim, G. Cho, and S. Cho, “Backprojectionfiltration image reconstruction from partial conebeam data for scatter correction,” in Medical Imaging 2011: Physics of Medical Imaging, vol. 7961 of Proceedings of the SPIE, February 2011. View at: Publisher Site  Google Scholar
 X. Dong, T. Niu, X. Jia, and L. Zhu, “Relationship between Xray illumination field size and flat field intensity and its impacts on Xray imaging,” Medical Physics, vol. 39, no. 10, pp. 5901–5909, 2012. View at: Publisher Site  Google Scholar
 J. Wang, W. Mao, and T. Solberg, “Scatter correction for conebeam computed tomography using moving blocker strips: a preliminary study,” Medical Physics, vol. 37, no. 11, pp. 5792–5800, 2010. View at: Publisher Site  Google Scholar
 X. Jia, H. Yan, L. Cervino, M. Folkerts, and S. B. Jiang, “A GPU tool for efficient, accurate, and realistic simulation of cone beam CT projections,” Medical Physics, vol. 39, no. 12, pp. 7368–7378, 2012. View at: Publisher Site  Google Scholar
 J. H. Siewerdsen and D. A. Jaffray, “Conebeam computed tomography with a flatpanel imager: magnitude and effects of Xray scatter,” Medical Physics, vol. 28, no. 2, pp. 220–231, 2001. View at: Publisher Site  Google Scholar
 L. Zhu, S. Yoon, and R. Fahrig, “A shortscan reconstruction for conebeam CT using shiftinvariant FBP and equal weighting,” Medical Physics, vol. 34, no. 11, pp. 4422–4438, 2007. View at: Publisher Site  Google Scholar
 L. Zhu, J. Starman, and R. Fahrig, “An efficient estimation method for reducing the axial intensity drop in circular conebeam CT,” International Journal of Biomedical Imaging, vol. 2008, Article ID 242841, 11 pages, 2008. View at: Publisher Site  Google Scholar
Copyright
Copyright © 2013 Xue Dong et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.