Case Report | Open Access
Shoko Merrit Yamada, Katsuya Gorai, Koichi Gonda, "Pinwheel-Shaped Titanium Plates Should Be Fixed to the Skull Using All Screw Holes to Protect the Plates from Being Bent", Case Reports in Surgery, vol. 2019, Article ID 5709285, 3 pages, 2019. https://doi.org/10.1155/2019/5709285
Pinwheel-Shaped Titanium Plates Should Be Fixed to the Skull Using All Screw Holes to Protect the Plates from Being Bent
Introduction. In cranioplasty, pinwheel-shaped titanium mini plates are frequently used to cover bone defects produced by burr holes, and it is common to insert screws through only a few of the holes in cranial flap fixation. Presentation of Case. A 69-year-old man who had undergone clipping surgery for subarachnoid hemorrhage 16 years previously visited our clinic because a titanium plate had penetrated his scalp one month after he was hit on the head by a wall cabinet. Imaging studies revealed that part of the titanium plate had bent outwards and penetrated the skin. The plate was surgically removed, a relief skin incision was made 6 cm posterior to the skin defect to suture the defected portion without causing tension, and a skin graft was applied to the relief skin incision portion. Two months after the maneuver, the skin graft had been successfully incorporated without infection. Discussion. Even after the subcutaneous and the cutaneous tissue have completely covered the pinwheel-shaped titanium mini plate, an edge without screw fixation can be easily bent by a hard blow to the overlying scalp. We recommend fixation of pinwheel-shaped titanium plates used in cranioplasty through all screw holes to protect against the plate being bent.
Titanium mini plates are indispensable in cranioplasty, contributing to a good cosmetic result [1–5]. Pinwheel-shaped titanium mini plates are frequently used to cover burr-hole defects. Such plates have several holes for screw fixation, but not all holes are used because four screws are enough to achieve stable bone flap fixation. We report here a case of plate protrusion at an unfixed edge of a pinwheel-shaped titanium mini plate after the patient was hit on the head.
2. Case Presentation
A 69-year-old Asian man noticed a titanium plate penetrating through his scalp in the right temporal region one week prior to visiting our clinic. The patient had a history of subarachnoid hemorrhage and had undergone clipping surgery of a ruptured middle cerebral artery aneurysm with pterional craniotomy 16 years previously. One month prior to visiting our clinic, he hit his right temporal region on a wall cabinet and noticed something hard bulging subcutaneously at the point of impact. Three weeks after the head trauma, he noticed the plate protruding through his scalp (Figure 1(a)). Three-dimensional computed tomography (3D-CT) scan revealed that part of the titanium plate had been bent and was penetrating the skin (Figure 1(b)). The plate was removed and the skin defect repaired. Figure 1(c) shows the removed plate and the bend in it.
Under general anesthesia, a skin incision was made along the plate to expose it (Figure 2(a)), and then the plate and screws were completely removed (Figure 2(b)). After debridement around the skin defect, a relief skin incision was made 6 cm posterior to it (Figure 2(c)) to enable suturing of the defect portion without causing tension (Figure 2(d), arrow). The relief skin incision portion was covered with a graft from the outer layer of the skin of the right thigh (Figure 2(e)). Two months later, the skin incision had healed without infection and the skin graft had been successfully incorporated (Figure 2(f)).
Exposure of a titanium mini plate several months to several years after cranioplasty or facial bone reconstruction is not rare [6–8]. In our case, exposure of the titanium plate was not caused by infection, which is the most frequent cause of plate protrusion . An edge of the patient’s pinwheel-shaped titanium mini plate was bent by relatively minor head trauma, and the upturned part of the plate compressed the scalp, resulting in focal skin ischemia and penetration of the skin. It might be considered that plates utilized for cranioplasty would rarely be bent after being covered by subcutaneous tissue in the absence of infection. However, this case demonstrates that a titanium mini plate can be easily bent by a blow to the scalp directly overlying the plate. Recently, thinner titanium plates (0.3 mm thick) have become more popular for cosmetic reasons . However, the thinner a titanium plate, the easier it would be to bend it. We therefore recommend that screws should be inserted into all holes of pinwheel-shaped titanium mini plates.
The corresponding author gave the patient informed consent for publication and obtained a written consent form with a signature of the patient.
Conflicts of Interest
All authors have no relevant financial or personal relationships with other individuals or organizations to disclose.
All authors express our gratitude to the Edanz Group for the editing of our manuscript.
- S. Ban, M. Ogata, T. Yamamoto et al., “Cranioplasty with titanium plate. Report of 114 cases,” Neurologia Medico-Chirurgica, vol. 27, no. 10, pp. 984–989, 1987.
- S. C. Smith and S. Pelofsky, “Adaptation of rigid fixation to cranial flap replacement,” Neurosurgery, vol. 29, no. 3, pp. 417-418, 1991.
- M. D. Cusimano and A. S. Suhardja, “Craniotomy revisited: techniques for improved access and reconstruction,” The Canadian Journal of Neurological Sciences, vol. 27, no. 1, pp. 44–48, 2000.
- K. R. Winston and M. C. Wang, “Cranial bone fixation: review of literature and description of a new procedure,” Journal of Neurosurgery, vol. 99, no. 3, pp. 484–488, 2003.
- S. Chibbaro, O. Makiese, D. Bresson et al., “Skull bone flap fixation - reliability and efficacy of a new grip-like titanium device (skull grip) versus traditional sutures: a clinical randomized trial,” Minimally Invasive Neurosurgery, vol. 54, no. 5/6, pp. 282–285, 2011.
- V. Bhatt, P. Chhabra, and M. S. Dover, “Removal of miniplates in maxillofacial surgery: a follow-up study,” Journal of Oral and Maxillofacial Surgery, vol. 63, no. 6, pp. 756–760, 2005.
- A. Haraji, M. H. Motamedi, and N. Moharamnejad, “Causes and incidence of miniplate removal following Le Fort I osteotomy,” Eplasty, vol. 9, article e45, 2009.
- K. Islamoglu, O. K. Coskunfirat, G. Tetik, and H. E. Ozgentas, “Complications and removal rates of miniplates and screws used for maxillofacial fractures,” Annals of Plastic Surgery, vol. 48, no. 3, pp. 265–268, 2002.
- S. S. Bukhari and M. Junaid, “Mini titanium plates and screws for cranial bone flap fixation; an experience from Pakistan,” Surgical Neurology International, vol. 6, no. 1, p. 75, 2015.
Copyright © 2019 Shoko Merrit Yamada et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.