Research Article | Open Access
Dongxia Ren, Chengjun Sun, Guanqun Ma, Danni Yang, Chen Zhou, Jiayu Xie, Yongxin Li, "Determination of Pyrethroids in Tea Brew by GC-MS Combined with SPME with Multiwalled Carbon Nanotube Coated Fiber", International Journal of Analytical Chemistry, vol. 2018, Article ID 8426598, 9 pages, 2018. https://doi.org/10.1155/2018/8426598
Determination of Pyrethroids in Tea Brew by GC-MS Combined with SPME with Multiwalled Carbon Nanotube Coated Fiber
A new method has been developed to simultaneously determine 7 pyrethroid residues in tea brew using gas chromatography-mass spectrometry (GC-MS) combined with solid phase microextraction (SPME) with multiwalled carbon nanotubes (MWCNTs) coated fiber. The MWCNTs coated fiber of SPME was homemade by using stainless steel wire as coating carrier and polyacrylonitrile (PAN) solution as adhesive glue. Under the optimized conditions, a good linearity was shown for bifenthrin, fenpropathrin, permethrin, and cyfluthrin in 1–50 ng mL−1 and for cypermethrin, fenvalerate, and deltamethrin in 5–50 ng mL−1. The correlation coefficients were in the range of 0.9948–0.9999. The average recoveries of 7 pyrethroids were 94.–107.3% and the relative standard deviations (RSDs) were less than 15%. The detection limit of the method ranged from 0.12 to 1.65 ng mL−1. The tea brew samples made from some commercial tea samples were analyzed. Among them, bifenthrin, fenpropathrin, and permethrin were found. The results show that the method is rapid and sensitive and requires low organic reagent consumption, which can be well used for the detection of the pyrethroids in tea brew.
Synthetic pyrethroids are a class of bionic synthetic ester pesticides whose structural or biological activity is similar to natural pyrethroids, with the features of high efficiency, broad spectrum, and less residue. Despite their low toxicity, long-term exposure and ingestion of these pesticides may result in endocrine disorders and affect nervous system function . China is the homeland of tea, and tea has several beneficial functions, such as refreshing and relieving restlessness. Nowadays, people not only require better quality of tea, but also pay particular attention to the harmful substances in it . Since tea is susceptible to suffer from pests and diseases, farmers usually spray pesticides to reduce the adverse effects, which results in pesticide residues [3–5]. The level of pesticide residues in tea brew is directly related to the intake amount of human, while the determination of these 7 pyrethroids in tea brew has rarely been reported [6, 7]. Therefore, it is necessary to develop a rapid and efficient method for the detection of these 7 pyrethroid pesticides in tea brew.
Recently, it is in tea that most researches are mainly focused on the detection of pyrethroid residues [2, 8]. Such methods as solid phase extraction, gel permeation chromatography, and matrix solid phase dispersion are commonly used as sample preparation techniques for analysis of pesticide residues in tea [9, 10]. However, organic solvents are employed during the extraction procedure, which poses a threat to the environment and human health. SPME is a new method of microextraction without solvent based on solid phase extraction, which has the advantages of simultaneous sampling, extraction and concentration, and low consumption of sample and reagents, as well as ease of operation and automation [8, 11]. Extraction phase is the core of SPME, whose performance directly determines the SPME efficiency, thus affecting the sensitivity of the method and the reliability of the results. However, the commercial fibers tend to swell off, and most of them use fused silica as the fiber carrier, which is expensive, is easy to break, and has short service life. In addition, the assortment of commercially available extraction phases is quite limited, which also limits their application. In order to improve the SPME performance, many researchers are committed to the research on new fibers.
In recent years, the emergence of many kinds of materials has brought new opportunities for the development of SPME fiber coating. Wang et al. used dispersive solid phase extraction with polyaniline-coated magnetic particles to determine 5 pyrethroids in tea drinks . Sun et al. determined the pollutants in the aquatic environment by SPME coupled with surface enhanced Raman spectroscopy, in which the SPME was coated with ZnO nanorods (ZnO NRs) decorated with Au@4-ATP@Ag core–shell nanoparticles (NPs) . Wang et al. synthesized a core−shell TiO2@C fiber for SPME to detect polycyclic aromatic hydrocarbons in river water samples . Wu et al. prepared a single-walled carbon nanotubes coated SPME fiber for extraction of several pyrethroids . Among these nanomaterials, MWCNTs, a new kind of carbon nanomaterials, have attracted much attention in many fields. As new adsorbents, MWCNTs present a promising future in the preconcentration of environmental contaminants, due to their unique hydrophobic structure and large specific surface area, as well as strong reaction with a variety of organic compounds [16, 17].
Up to now, the detection of these 7 pyrethroids in tea brew is rarely reported [6, 7]. In this study, a SPME with MWCNTs coated fiber was developed using stainless steel wire as the coating carrier, which in combination with GC-MS was applied to the detection of the target pyrethroids in tea brew with satisfactory results.
2.1. Chemicals and Materials
The standards of bifenthrin, fenpropathrin, permethrin, cyfluthrin, cypermethrin, fenvalerate, and deltamethrin were purchased from National Standard Material Center of China. n-Hexane, methanol, and acetone were of HPLC grade and provided by Tianjin Branch Miou Chemical Reagent Co., Ltd. MWCNTs (10–30 μm length, 10–20 nm outer diameter) were purchased from Chinese Academy of Sciences Chengdu Organic Chemistry Co., Ltd. N,N-Dimethylformamide (DMF) was obtained from Shanghai Guoyao Chemical Reagent Co., Ltd. Polyacrylonitrile (PAN) was purchased from Sigma (St. Louis, MO, USA). Water used throughout the experiments was obtained from a Millipore water purification system (18.25 M Ω cm, Millipore, USA).
Stainless steel wires (12 cm × 0.15 mm) were obtained from Shenzhen Santk Metal Material Co., Ltd. SPME needles (SPME-S-02) were purchased from Shanghai New Extension Analytical Instruments Technology Co., Ltd. SPME handle (57330-U) was obtained from Supelco.
2.2. Preparation of the Standard Solutions
The stock solutions of 100 μg mL−1 were prepared with pyrethroid standards containing bifenthrin, fenpropathrin, permethrin, cyfluthrin, cypermethrin, fenvalerate, and deltamethrin in n-hexane and stored at −20°C. Five levels of calibration scales were obtained by diluting the stock standard solutions with n-hexane, which were stored at 4°C until used.
2.3. Preparation of MWCTs Coated Fiber for SPME
The stainless steel wires were cleaned with methanol, acetone, and n-hexane in order to remove the surface contaminants and then dried in the air at room temperature. Meanwhile, 0.5 mg of PAN was dissolved in 7.5 mL of DMF, and the mixture was kept at 90°C for 1 h. Five mg of MWCNTs was taken into a centrifuge tube and mixed with PAN solution by ultrasound. Subsequently, the cleaned stainless steel wire was dipped into the coating suspension with a height of 1.5 cm for one minute [18–20] and then withdrew at a constant rate to ensure the formation of a smooth coating on the wire surface. After this, the fiber was dried at 180°C for 2 min. This above procedure from “dipped into the coating suspension” was repeated 3 times. Prior to use, the proposed SPME was conditioned at 280°C for 2 h.
2.4. Sample Preparation
The tea samples were randomly purchased from the local markets. After fully mixed, 3 g (accurate to 0.001 g) of each was weighed into a 250 mL beaker and 100 mL of boiled water was added and then kept for 3 h. 10 mL of the above-mentioned supernatant in a vial with cap was placed on a magnetic stirrer and extracted with the proposed SPME with MWCNTs coated fiber at 750 rpm for 20 min. The target compounds were desorbed at 280°C for 8 min and determined by GC-MS.
2.5. GC-MS Analysis
GC-MS analysis was performed by Agilent Technologies 7890A GC with a 5975C MS, equipped with a DB-5 MS fused silica capillary column (30 m × 0.25 mm × 0.25 μm) (Agilent Technologies, Santa Clara, CA, USA). Helium (purity 99.999%) served as carrier gas at a flow rate of 1 mL min−1. To ensure rapid and good separation of all the target compounds, a programmed temperature was employed. The initial column temperature was held at 60°C, and then to 150°C at 30°C min−1, and finally increased to 290°C at 10°C min−1, with hold time of 6 min. The injection port temperature was set at 280°C and the injection mode was in splitless mode.
In this study, the mass spectrometer was performed with an EI source in selected ion monitoring (SIM) mode. The electron energy was 70 eV with the temperature of the ion source and transfer line at 230°C and 280°C, respectively. The solvent delay was setting as 7 min. In the NIST search library, higher abundance of sub-ions was selected as qualitative ions and quantitative ions (Table 1).
3. Results and Discussion
3.1. Surface Structure of the Fiber
Carbon nanotubes have the advantages of large specific surface area, good chemical stability, superior mechanical strength, and high heat resistance and can set off π-π interactions [21, 22]. Therefore, they have a promising application potential in the extraction field, for example, as SPME fiber. In order to investigate the surface properties of MWCNTs coated SPME fibers, seven pyrethroid pesticides were extracted with the proposed fibers and the uncoated stainless steel wires under the same experimental conditions. The results showed that the MWCNTs coated SPME fibers have a strong adsorption ability. Additionally, scanning electron microscopy (SEM) was employed to further explore the surface characteristics of the proposed fiber, as shown in Figure 1. It could be seen from Figure 1(b) that the multiwalled carbon nanotubes were uniformly attached on the surface of the stainless steel wire and were semiexposed. Figure 1(c) showed that the typical porous structure of MWCNTs coating was formed on the fiber surface, which resulted in high surface area of the coating, thereby offering extraordinary adsorption capacity. And the thickness of the coating is about 70 μm.
3.2. Optimization of GC-MS Conditions
Tea has a challenging matrix, containing pigments, caffeine, and other impurities, which would interfere with the detection of the target compounds. In order to improve the accuracy and sensitivity of the method, it is necessary to select the appropriate mass spectrometry conditions. Under optimized GC-MS conditions, the chromatogram of the seven pyrethroids was shown in Figure 2.
3.3. Optimization of SPME Conditions
There are several main factors affecting SPME extraction and desorption efficiency, including stir rate, desorption temperature, and time, as well as extraction time. In this study, these factors were optimized as follows: the blank sample was spiked with the stock solution to obtain a final concentration of 20 ng mL−1. The analytes were extracted under different SPME conditions and analyzed by GC-MS. The optimal conditions were selected according to the peak area of the analytes.
3.3.1. Optimization of Extraction Time
According to the digital model of direct SPME and the dynamic adsorption model of nonbalance solid phase microextraction, the extraction efficiency is proportional to concentration of the target compound when other conditions are constant [23, 24]. In this study, analytes were extracted for 10, 15, 20, 25, and 30 min, respectively, with a stirring rate of 750 rpm and desorbing for 5 min at 280°C. The peak area of the analytes was used as the reference index. Each experiment was repeated for three times. As shown in Figure 3, the amount of the target pyrethroids increased with the increase of the extraction time. When the extraction time was 20 min, the extraction amount of the analytes reached the maximum and tended to be balanced. Therefore, taking into account the extraction efficiency and the analytical time, the extraction time of 20 min was selected.
3.3.2. Optimization of Stirring Rate
The essence of SPME is to extract the analytes from the sample solution to the stationary phase. During the extraction process, the analytes in the solution diffused from the solid-liquid layer to the surface of the solid phase coating. Thus, the appropriate agitation facilitates the diffusion of molecules in the solution to improve the extraction efficiency . The extraction efficiency of 7 pyrethroids was studied with stirring rates ranging from 0 to 1250 rpm when the other conditions were fixed, including extraction time of 20 min and desorption at 280°C for 5 min. Each experiment was repeated for three times. As shown in Figure 4, the peak area of the analytes increased as the stirring rate increased. Until 750 rpm, the peak area of the analytes reached the maximum. If the stirring rate continued to rise, it was prone to depression in the center of the liquid surface, which not only affected the normal operation of the experiments, but also led to a decrease of the extraction efficiency. Thus, 750 rpm was chosen as the stirring rate for subsequent experiments.
3.3.3. Optimization of Desorption Temperature
In SPME-GC-MS detection, if the desorption temperature is too low, the analytes cannot be effectively desorbed, and the carry-over would exhibit an adverse effect on both the results and the follow-up experiments, while if the temperature is too high, it might make the analytes decomposed and at the same time also affect the service life of SPME fibers [15, 26]. The influence of desorption temperature on the desorption efficiency for 7 pyrethroids was investigated in the range of 220–300°C. The other conditions were as follows: the stir rate was 750 rpm and the extraction time was 20 min with desorption for 5 min. Each experiment was repeated for three times. As Figure 5 shows, the elevated temperature facilitated the effective desorption of the analytes. The peak area of each analyte increased with the increase of the desorption temperature when it was in the range of 220–280°C. When the desorption temperature was increased to 300°C, the peak area of some of the analytes did not increase significantly and that of others decreased considerably, indicating that the analytes had been completely desorbed at 280°C. Therefore, 280°C was selected as the desorption temperature in subsequent experiments.
3.3.4. Optimization of Desorption Time
In order to investigate the effect of desorption time, the analytes were determined when the desorption time was 2, 5, 8, and 11 min, respectively, under the conditions of extraction for 20 min with a stir rate of 750 rpm and desorption at 280°C. Each experiment was repeated for three times. As shown in Figure 6, when the desorption time was 2–8 min, the peak area of the analytes increased with the desorption time. However, when the desorption time continued to increase to 11 min, the peak area of most analytes did not change significantly. Therefore, 8 min was selected as the desorption time.
3.4. Method Validation
3.4.1. Regression Equation, Linear Range, and Detection Limits
A standard series of 1, 5, 10, 30, and 50 ng mL−1 was prepared in the blank tea sample, and the assay was carried out in accordance with the procedure of 2.4. The standard curves were plotted with the peak area as the ordinate and the mass concentration of the standard solution as the abscissa (, ng mL−1). The limits of detection (LODs) and the limits of quantification (LOQs) are calculated on the basis of the 3-fold and 10-fold baseline noise, respectively. The results were presented in Table 2. There was a good linear relationship for bifenthrin, fenpropathrin, permethrin, and cyfluthrin in the range of 1 to 50 ng mL−1 and for cypermethrin, fenvalerate, and deltamethrin in the range of 5 to 50 ng mL−1, with the correlation coefficients in 0.9948–0.9999. The detection limits were 0.12–1.65 ng mL−1, and the limits of quantification were 0.36–4.95 ng mL−1.
3.4.2. Recovery and Precision of the Method
Under the optimized pretreatment and chromatographic conditions, the blank sample was spiked with the stock standard mixture to obtain a final concentration of 5, 10, and 50 ng mL−1, respectively, and then treated with MWCNTs coated SPME and analyzed by GC-MS. The recoveries were calculated according to the ratio of the determined concentration and the actual value. Each experiment was determined for 6 times. The results were shown in Table 3. The average recoveries ranged from 79.3% to 126% and the RSDs were less than 15%, indicating the favorable potential of the proposed method to be implemented in real application.
3.4.3. Determination of Commercial Samples
The proposed method was employed for the determination of 7 pyrethroid residues in tea brew samples made from different brands of tea randomly purchased from the local markets, and the results were shown in Figures 7–9. Bifenthrin, fenpropathrin, and permethrin were detected, although none reached the limit of quantification (the former two were found in jasmine tea and green tea, respectively, and the third one was only found in jasmine tea). Kang  determined 16 pesticide residues including fenpropathrin, cypermethrin, deltamethrin, fenvalerate, and lambda-cyhalothrin in tea brew by gas chromatography-electron capture detector (GC-ECD) after extracted by ethyl acetate for 3 times combining with LC-Florisil SPE column cleanup. The linear ranges and the recoveries of the method for the pyrethroids were in the range of 10 ng mL−1–1000 ng mL−1 and 63.1%–105.2%, respectively. When compared with this method, our method is more sensitive, is easier to operate, and consumes less organic reagents.
A multiwalled carbon nanotube coated SPME fiber was prepared by using stainless steel wire as the coating carrier and PAN as the adhesive. The SPME sampling combined with GC-MS analysis was employed to determine seven pyrethroids in tea brew samples. The proposed MWCNTs SPME not only has the advantages of fastness, efficiency, and convenience, but also can overcome the defects of the commercial fiber, such as ease of being broken and short service life as well as expensive price. The method is simple, rapid, and sensitive and with low consumption of organic reagents, which is suitable for the detection of pyrethroid pesticides in tea brew.
Conflicts of Interest
The authors declare that there are no conflicts of interest regarding the publication of this paper.
This work was supported by Project of Sichuan Health Bureau of China (no. 150111) and Project of Chengdu Science and Technology Bureau of China (2015-HM01-00182-SF).
- Y. B. Liu, L. H. Jia, R. F. Xue, and M. X. Zhang, “Rapid GC-MS Determination of Residual Pyrethroid Pesticides in Vegetables and Fruits,” Physical Testing And Chemical Analysis Part B Chemical Analysis, vol. 42, no. 8, pp. 637–643, 2006.
- H. P. Chen, X. Liu, Q. H. Wang, and Y. Jiang, “Simultaneous Determination of 72 Pesticide Residues in Tea by Gas Chromatography-Mass Spectrometry,” Food Science, vol. 32, no. 6, pp. 159–164, 2011.
- P. Schreinemachers and P. Tipraqsa, “Agricultural pesticides and land use intensification in high, middle and low income countries,” Food Policy, vol. 37, no. 6, pp. 616–626, 2012.
- J. P. Dos Anjos and J. B. De Andrade, “Determination of nineteen pesticides residues (organophosphates, organochlorine, pyrethroids, carbamate, thiocarbamate and strobilurin) in coconut water by SDME/GC-MS,” Microchemical Journal, vol. 112, pp. 119–126, 2014.
- M. Tankiewicz, C. Morrison, and M. Biziuk, “Multi-residue method for the determination of 16 recently used pesticides from various chemical groups in aqueous samples by using DI-SPME coupled with GC–MS,” Talanta, vol. 107, pp. 1–10, 2013.
- X. Li, Z. Zhang, P. Li, Q. Zhang, W. Zhang, and X. Ding, “Determination for major chemical contaminants in tea (Camellia sinensis) matrices: A review,” Food Research International, vol. 53, no. 2, pp. 649–658, 2013.
- M. Gupta and A. Shanker, “Persistence of acetamiprid in tea and its transfer from made tea to infusion,” Food Chemistry, vol. 111, no. 4, pp. 805–810, 2008.
- Q. Zhou, J. Xiao, and W. Wang, “Using multi-walled carbon nanotubes as solid phase extraction adsorbents to determine dichlorodiphenyltrichloroethane and its metabolites at trace level in water samples by high performance liquid chromatography with UV detection,” Journal of Chromatography A, vol. 1125, no. 2, pp. 152–158, 2006.
- G. Min, S. Wang, H. Zhu, G. Fang, and Y. Zhang, “Multi-walled carbon nanotubes as solid-phase extraction adsorbents for determination of atrazine and its principal metabolites in water and soil samples by gas chromatography-mass spectrometry,” Science of the Total Environment, vol. 396, no. 1, pp. 79–85, 2008.
- P. Liu, Y. Chen, C. Zhao, and L. Tian, “Determination of ten photoinitiators in fruit juices and tea beverages by solid-phase micro-extraction coupled with gas chromatography/mass spectrometry,” Chinese Journal of Chromatography, vol. 31, no. 12, pp. 1232–1239, 2013.
- F. Guo, “Progress of Sample Pretreatment Techniques in Determination of Pesticide Residue in Tea,” Studies of Trace Elements and Health, vol. 29, no. 1, pp. 61–64, 2012.
- Y. Wang, Y. Sun, Y. Gao et al., “Determination of five pyrethroids in tea drinks by dispersive solid phase extraction with polyaniline-coated magnetic particles,” Talanta, vol. 119, pp. 268–275, 2014.
- L. Sun, M. Zhang, V. Natarajan, X. Yu, X. Zhang, and J. Zhan, “Au@Ag core-shell nanoparticles with a hidden internal reference promoted quantitative solid phase microextraction-surface enhanced Raman spectroscopy detection,” RSC Advances, vol. 7, no. 38, pp. 23866–23874, 2017.
- F. Wang, J. Zheng, J. Qiu et al., “In Situ Hydrothermally Grown TiO2@ C CoreShell Nanowire Coating for Highly Sensitive Solid Phase Microextraction of Polycyclic Aromatic Hydrocarbons,” ACS applied materials interfaces, vol. 9, no. 2, pp. 1840–1846, 2017.
- F. Wu, W. Lu, J. Chen, W. Liu, and L. Zhang, “Single-walled carbon nanotubes coated fibers for solid-phase microextraction and gas chromatography-mass spectrometric determination of pesticides in Tea samples,” Talanta, vol. 82, no. 3, pp. 1038–1043, 2010.
- G. Wu, X. Bao, S. Zhao, J. Wu, A. Han, and Q. Ye, “Analysis of multi-pesticide residues in the foods of animal origin by GC-MS coupled with accelerated solvent extraction and gel permeation chromatography cleanup,” Food Chemistry, vol. 126, no. 2, pp. 646–654, 2011.
- Y. Li, “SPE-gas chromatography/mass spectrometry determination of pesticide residues in tea,” Chinese Journal of Health Laboratory Technology, vol. 20, no. 6, pp. 1271–1274, 2010.
- J. Lü, J. Liu, Y. Wei et al., “Preparation of single-walled carbon nanotube fiber coating for solid-phase microextraction of organochlorine pesticides in lake water and wastewater,” Journal of Separation Science, vol. 30, no. 13, pp. 2138–2143, 2007.
- F. Wu, J. Chen, W. Lu, Z. Lin, and L. Zhang, “Solid-phase microextraction using poly (1-hexadecene-co-trimethylolpropane trimethacrylate) coating for analysis of pesticides in tea samples by high-performance liquid chromatography,” Current Analytical Chemistry, vol. 7, no. 4, pp. 341–348, 2011.
- L. Chen, J. Zeng, C. Ma et al., “Preparation of multi-walled carbon nanotubes coated solid-phase microextraction fiber and its application on the analysis of polybromobiphenyls in seawater,” Scientia SinicaChimica, vol. 39, no. 12, pp. 1652–1657, 2009.
- T. Fukushima, A. Kosaka, Y. Ishimura et al., “Molecular ordering of organic molten salts triggered by single-walled carbon nanotubes,” Science, vol. 300, no. 5628, pp. 2072–2074, 2003.
- D.-Q. Yang, B. Hennequin, and E. Sacher, “XPS demonstration of π-π interaction between benzyl mercaptan and multiwalled carbon nanotubes and their use in the adhesion of Pt nanoparticles,” Chemistry of Materials, vol. 18, no. 21, pp. 5033–5038, 2006.
- D. Louch, S. Motlagh, and J. Pawliszyn, “Dynamics of organic compound extraction from water using liquid-coated fused silica fibers,” Analytical Chemistry, vol. 64, no. 10, pp. 1187–1199, 1992.
- J. Ai, “Solid Phase Microextraction for Quantitative Analysis in Nonequilibrium Situations,” Analytical Chemistry, vol. 69, no. 6, pp. 1230–1236, 1997.
- L. Chen, “Determination of pyrethrins in tea brew by using solid phase microextraction,” Journal ofMinjiang University, vol. 31, no. 5, pp. 103–107, 2010.
- L. Chen, W. Chen, C. Ma, D. Du, and X. Chen, “Electropolymerized multiwalled carbon nanotubes/polypyrrole fiber for solid-phase microextraction and its applications in the determination of pyrethroids,” Talanta, vol. 84, no. 1, pp. 104–108, 2011.
- W. B. Kang, “GC determination of organophosphorus, organochlorine and pyrethroid pesticide residues in tea drinker,” Fujian Analysis Testing, vol. 16, no. 4, pp. 47–49, 2007.
Copyright © 2018 Dongxia Ren et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.