Symmetries, Differential Equations, and Applications: Galois Bicentenary
View this Special IssueResearch Article  Open Access
M. B. Akgül, M. Pakdemirli, "Magnetohydrodynamic and Slip Effects on the Flow and Mass Transfer over a MicrocantileverBased Sensor", Journal of Applied Mathematics, vol. 2012, Article ID 289459, 11 pages, 2012. https://doi.org/10.1155/2012/289459
Magnetohydrodynamic and Slip Effects on the Flow and Mass Transfer over a MicrocantileverBased Sensor
Abstract
Hydromagnetic flow and mass transfer of a viscous incompressible fluid over a microcantilever sensor surface are studied in the presence of slip flow. In addition, chemical reaction at the sensor surface is taken into account. The governing equations for the flow are reduced to a local nonsimilarity form. Resulting equations are solved numerically for various values of flow parameters. Effects of physical quantities on the velocity and concentration profiles are discussed in detail.
1. Introduction
Diagnosis of small molecules is extremely important in biomedical and environmental applications. Recent advances in micro and nanofabrication techniques have boosted developments of novel sensors for this purpose. In particular, microcantileverbased sensors (Mcs) are distinguished from the others because of their high sensitivity, lower cost, simple procedure, quick response, and realtime sensing characteristics [1, 2]. Microcantileverbased sensors are simple mechanical devices. They are anchored from one end to a fixed support and have various transducer mechanisms such as atomic force microscopy (AFM), mass, temperature, electromagnetic field, surface stress, and medium viscoelasticity [3–10].
Usually, only one surface of the Mcs is coated with a specific receptor, which reacts with the analytes in the solution. As a result of the reaction, adhesion of analytes to surface leads to deflection at the free tip of the Mcs, which can be detected with optical or electrical methods. This motion gives useful information for the diagnosis of biological and chemical agents, drug discovery, and monitoring of complex diseases [2, 11–13].
Microcantileverbased sensors are often mounted inside a fluidic cell to provide faster detection of target analytes and to prevent possible contamination of samples during the analysis [14–16]. However, sensing and detection performance of the Mcs are significantly affected by the surrounding flow conditions. Khaled and Vafai [17] considered hydromagneticsqueezed flow and heat transfer over a sensor surface and found that both local wall shear stress and local Nusselt number increase with increasing magnetic and squeezing free stream velocity. Mahmood et al. [18] analyzed flow and heat transfer over a porous sensor surface and found that when the channel is squeezed, the skin friction reduces but the heat transfer coefficient increases on the sensor surface. Khaled et al. [19] investigated microcantilever deflections under oscillating flow conditions in the presence of chemical reaction at the receptor surface. Khanafer and Vafai [20] studied geometrical and flow configurations for enhanced microcantilever detection within a fluidic cell. Results of that investigation showed that as the height of the fluidic cell decreases, mass transfer enhances due to an increase in the axial velocity along the microcantilever. Islam et al. [21] tried enhancing microcantilever capability with integrated AC electroosmotic trapping. Khanafer et al. [22] considered fluidsolid interaction of flow and heat transfer characteristics around a flexible microcantilever in a fluidic cell.
The abovementioned studies are conducted with the noslip assumption at the solidliquid interface. Nonetheless, slip flow may exist due to small dimensions of microcantilever and may play an important role in the formation of momentum and mass transfer boundary layers [23]. The velocity slip of liquid on the sensor surface is related to the slip length and the gradient of velocity [24]. In the present paper, we consider the hydromagnetic flow and mass transfer of a viscous incompressible fluid over the Mcs in the presence of slip flow with adhesion of analytes taken into account at the surface of Mcs. The governing equations are converted into the dimensionless system of equations using nonsimilarity variables. The local nonsimilarity solutions are obtained for the velocity and concentration distributions. The results are shown in figures and effects of flow parameters are discussed.
2. Governing Equations
Consider the steady twodimensional, incompressible, laminar flow of an electrically conducting viscous fluid over a Mcs, which is inserted into hydrodynamic entrance region of a fluidic cell as shown in Figure 1. The height of the fluidic cell is assumed much greater than the boundary layer thickness. Hence, using the boundary layer approximation, the governing equations for the flow are where is the coordinate along the Mcs surface and is the coordinate vertical to , , and are the velocity components in the and coordinates, respectively. is the fluid density; is the concentration of the fluid. is the kinematic viscosity, is the coefficient of mass diffusivity, is the fluid pressure, and is the electric conductivity. is the externally applied magnetic field and defined as , where is the constant component of . Also the external electric field is zero and induced electric field is assumed negligible. is the free stream velocity.
It is worth mentioning that the adhesion of analytes to the Mcs surface can be formulated by equating mass diffusion at the surface with the adhesive flux. The adhesive flux can be modeled as a firstorder heterogeneous reaction [25]. Taking account of the fact, the boundary conditions are where is the slip length, is the bulk concentration of analytes, and is the reaction rate constant.
In order to obtain nonsimilar solutions, one may introduce where is the transformed streamwise coordinate and is a pseudosimilarity variable. , and are arbitrary constants and is the stream function. Substitution of these variables and functions into the original equations yields The associated boundary conditions are where primes denote differentiation with respect to the pseudosimilarity variable and the flow parameters in the above equations are
Here is the dimensionless magnetic parameter, is the slip parameter, is the dimensionless reaction rate constant, and Sc is the Schmidt number.
Approximate solutions of (2.4) will be obtained by employing the local nonsimilarity method. Details of the application of the local nonsimilarity method to solutions of nonsimilar boundary layer equations can be found in [26, 27]. We introduce the new dependent variables Substituting (2.7) into the nonsimilar boundary layer equation (2.4), we obtain Differentiating (2.8) and boundary conditions in (2.5) with respect to , we obtain auxiliary equations and boundary conditions as follows: The derivatives of , , and with respect to are neglected. Hence, the boundary layer equations and auxiliary equations with their boundary conditions can be written as In contrast to the local similarity method, the nonsimilar terms on the righthand side of the boundary layer equations are retained. As a consequence of this, it is expected that results obtained from the local nonsimilarity method were more accurate than those of the local similarity method. The set of coupled equations (2.10)–(2.14) can be treated as a system of ordinary differential equations with being treated as a constant parameter in any streamwise location.
3. Results and Discussions
The nonlinear ordinary differential equation system (2.10)–(2.13) with boundary conditions in (2.14) are solved numerically by applying a finite difference code which implements the threestage Lobatto IIIA formula that provides continuous solutions of fourthorder accuracy in the problem domain. Description of the numerical method is given in [28, 29]. Results are obtained only for , and their derivatives, which are physically relevant and displayed graphically for various values of problem parameters to see developments in the velocity and concentration fields.
The variation of dimensionless horizontal velocity profiles is shown for various values of the velocity powerlaw index in Figure 2 for (in the absence of external magnetic field) and . It is clear that the velocity boundary layer thickness over the wedge type sensor () is thinner than that of a flat sensor type (). The effects of the streamwise location on the velocity profiles are also shown. Horizontal velocity decreases with increase of , which is expected.
Figure 3 indicates that an increase in the slip parameter causes a reduction in the boundary layer thickness. Figure 4 is depicted to show the variations in the velocity field for flat and wedge sensor types in the presence of magnetic field. The horizontal velocity increases and hence the boundary layer thickness decreases for an applied magnetic field.
Figure 5 displays concentration profiles of the analytes for different sensor types. It is obvious that surface concentration values increase with increase of m and boundary layer thickness increases along the streamwise direction.
Figures 6 and 7 are plotted to examine the effects of the slip parameter and magnetic parameter on the concentration fields. It can be seen that analyte concentration increases with increase of either or in the boundary layer region.
The variation of concentration profiles for different values of dimensionless reaction rate constant is presented in Figure 8. As reaction rate constant increases, the analyte concentration at the sensor surface decreases. For large values of , reaction becomes faster and consumption (adhesion) of the analytes at the sensor surface increases. Thus, the boundary layer thickness increases with increase of .
Tables 1 and 2 show the dimensionless velocity, concentration, and their gradient values at the sensor surface for various streamwise locations. It is understood that surface velocity increases notably with increase of , , and for any streamwise location. In addition to this, the velocity gradient increases with increase of and , which is expected. On the other hand, values of decreases with increase of slip parameter because of the increasing velocity at the sensor surface. Moreover, the surface concentration of analytes decreases along the streamwise direction, while concentration gradient first increases to a certain streamwise location and after this location, it decreases for various values of , , and as shown in Tables 1 and 2.


4. Concluding Remarks
This paper deals with the steady twodimensional, incompressible, laminar flow, and mass transfer over a sensor surface, taking into account the effects of the slip condition and applied magnetic field. The governing equations are derived using the boundary layer approximation and reduced to local nonsimilar ones. Resulting equations are solved numerically. Dimensionless velocity and concentration profiles are shown graphically for various flow parameters.
It is found that velocity and concentration increase with increase of velocity powerlaw index and dimensionless magnetic parameter , whereas dimensionless analyte concentration decreases with an increase of reaction rate. Moreover, mass transfer boundary layer thickness decreases with increase of slip parameter . The surface concentration of analytes decreases along the streamwise direction. Finally, concentration gradient first increases until a certain streamwise location and then after this location, it decreases.
References
 E. A. Wachter and T. Thundat, “Micromechanical sensors for chemical and physical measurements,” Review of Scientific Instruments, vol. 66, pp. 3662–3667, 1995. View at: Google Scholar
 R. Raiteri, M. Grattarola, H.J. Butt, and P. Skladal, “Micromechanical cantileverbased biosensors,” Sensors and Actuators, vol. 79, pp. 115–126, 2001. View at: Google Scholar
 J. K. Gimzewski, C. Gerber, E. Meyer, and R. R. Schlittler, “Observation of a chemical reaction using a micromechanical sensor,” Chemical Physics Letters, vol. 217, no. 56, pp. 589–594, 1994. View at: Google Scholar
 J. Lai, T. Perazzo, Z. Shi, and A. Majumdar, “Infrared photodetection in the picowatt range using micromechanical sensors,” in Proceedings of ASME Conference on Microelectromechanical Systems (MEMS '96), pp. 55–60, 1996. View at: Google Scholar
 P. I. Oden, P. G. Datskos, T. Thundat, and R. J. Warmack, “Uncooled thermal imaging using a piezoresistive microcantilever,” Applied Physics Letters, vol. 69, pp. 3277–3279, 1996. View at: Google Scholar
 P. G. Datskos, P. I. Oden, T. Thundat, E. A. Wachter, R. J. Warmack, and S. R. Hunter, “Remote infrared radiation detection using piezoresistive microcantilevers,” Applied Physics Letters, vol. 69, no. 20, pp. 2986–2988, 1996. View at: Google Scholar
 E. A. Wachter, T. Thundat, P. I. Oden, R. J. Warmack, P. G. Datskos, and S. L. Sharp, “Remote optical detection using microcantilevers,” Review of Scientific Instruments, vol. 67, no. 10, pp. 3434–3439, 1996. View at: Google Scholar
 R. Berger, C. Gerber, J. K. Gimzewski, E. Meyer, and H.J. Güntherodt, “Thermal analysis using a micromechanical calorimeter,” Applied Physics Letters, vol. 69, pp. 40–42, 1996. View at: Google Scholar
 T. Thundat, R. J. Warmack, G. Y. Chen, and D. P. Allison, “Thermal and ambientinduced deflections of scanning force microscope cantilevers,” Applied Physics Letters, vol. 64, no. 21, pp. 2894–2896, 1994. View at: Publisher Site  Google Scholar
 T. Thundat, G. Y. Chen, R. J. Warmack, D. P. Allison, and E. A. Wachter, “Vapor detection using resonating microcantilevers,” Analytical Chemistry, vol. 67, no. 3, pp. 519–521, 1995. View at: Google Scholar
 M. E. Wright, D. K. Han, and R. Aebersold, “Mass spectrometrybased expression profiling of clinical prostate cancer,” Molecular and Cellular Proteomics, vol. 4, no. 4, pp. 545–554, 2005. View at: Publisher Site  Google Scholar
 L. A. Liotta, V. Espina, A. I. Mehta et al., “Protein microarrays: meeting analytical challenges for clinical applications,” Cancer Cell, vol. 3, no. 4, pp. 317–325, 2003. View at: Publisher Site  Google Scholar
 H. F. Ji, X. Yon, J. Zhang, and T. Thundat, “Molecular recognition of biowarfare agents using micromechanical sensors,” Expert Review of Molecular Diagnostics, vol. 4, no. 6, pp. 859–866, 2004. View at: Publisher Site  Google Scholar
 W. Shu, E. D. Laue, and A. A. Seshia, “Investigation of biotinstreptavidin binding interactions using microcantilever sensors,” Biosensors and Bioelectronics, vol. 22, no. 910, pp. 2003–2009, 2007. View at: Publisher Site  Google Scholar
 L. A. Pinnaduwage, T. Thundat, J. E. Hawk et al., “Detection of 2,4dinitrotoluene using microcantilever sensors,” Sensors and Actuators, B, vol. 99, no. 23, pp. 223–229, 2004. View at: Publisher Site  Google Scholar
 H. P. Lang, M. K. Baller, R. Berger et al., “An artificial nose based on a micromechanical cantilever array,” Analytica Chimica Acta, vol. 393, pp. 59–65, 1999. View at: Google Scholar
 A. R. A. Khaled and K. Vafai, “Hydromagnetic squeezed flow and heat transfer over a sensor surface,” International Journal of Engineering Science, vol. 42, no. 56, pp. 509–519, 2004. View at: Publisher Site  Google Scholar
 M. Mahmood, S. Asghar, and M. A. Hossain, “Squeezed flow and heat transfer over a porous surface for viscous fluid,” Heat and Mass Transfer, vol. 44, no. 2, pp. 165–173, 2007. View at: Publisher Site  Google Scholar
 A. R. A. Khaled, K. Vafai, M. Yang, X. Zhang, and C. S. Ozkan, “Analysis, control and augmentation of microcantilever deflections in biosensing systems,” Sensors and Actuators, B, vol. 94, no. 1, pp. 103–115, 2003. View at: Publisher Site  Google Scholar
 K. Khanafer and K. Vafai, “Geometrical and flow configurations for enhanced microcantilever detection within a fluidic cell,” International Journal of Heat and Mass Transfer, vol. 48, no. 14, pp. 2886–2895, 2005. View at: Publisher Site  Google Scholar
 N. Islam, M. Lian, and J. Wu, “Enhancing microcantilever capability with integrated AC electroosmotic trapping,” Microfluidics and Nanofluidics, vol. 3, no. 3, pp. 369–375, 2007. View at: Publisher Site  Google Scholar
 K. Khanafer, A. Alamiri, and I. Pop, “Fluidstructure interaction analysis of flow and heat transfer characteristics around a flexible microcantilever in a fluidic cell,” International Journal of Heat and Mass Transfer, vol. 53, no. 910, pp. 1646–1653, 2010. View at: Publisher Site  Google Scholar
 S. Kiwan and M. A. AlNimr, “Investigation into the similarity solution for boundary layer flows in microsystems,” Journal of Heat Transfer, vol. 132, no. 4, Article ID 041011, 9 pages, 2010. View at: Publisher Site  Google Scholar
 F. M. White, Viscous Fluid Flow, McGrawHill, New York, NY, USA, 1991.
 T. David, S. Thomas, and P. G. Walker, “Platelet deposition in stagnation point flow: an analytical and computational simulation,” Medical Engineering and Physics, vol. 23, pp. 299–312, 2001. View at: Google Scholar
 E. M. Sparrow, H. Quack, and C. J. Boerner, “Local nonsimilarity boundary layer solutions,” AIAA Journal, vol. 8, no. 11, pp. 1936–1942, 1970. View at: Google Scholar
 E. M. Sparrow and H. S. Yu, “Local nonsimilarity thermal boundarylayer solutions,” Journal of Heat Transfer. ASME, pp. 328–334, 1971. View at: Google Scholar
 L. F. Shampine, I. Gladwell, and S. Thompson, Solving ODEs with MATLAB, chapter 3, Cambridge University Press, Cambridge, UK, 2003. View at: Publisher Site  Zentralblatt MATH  MathSciNet
 J. Kierzenka and L. F. Shampine, “A BVP solver based on residual control and the MATLAB PSE,” ACM Transactions on Mathematical Software, vol. 27, no. 3, pp. 299–316, 2001. View at: Publisher Site  Google Scholar  MathSciNet
Copyright
Copyright © 2012 M. B. Akgül and M. Pakdemirli. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.