Research Article | Open Access
Robert Murphy, Steven Wilcox, "A New Approach in the Design of High-Risk Infusion Technology", Journal of Healthcare Engineering, vol. 1, Article ID 834097, 14 pages, 2010. https://doi.org/10.1260/2040-22184.108.40.206
A New Approach in the Design of High-Risk Infusion Technology
The syringe infusion pump has been established as the instrument of choice for high-risk infusions, where potent drugs are often delivered at low rates of flow. However, numerous instances of unexpected flow error with consequent patient physiological impact have been reported. These include unwanted bolus delivery on release of line occlusion, dosage fluctuation due to pump height change and fluid reflux within the multiple pump installations now common in the Intensive Care Unit (ICU). This article examines the performance of a typical ICU syringe infusion pump and identifies mechanical compliance, inherent in commercial designs, as a source of flow error that should not be ignored by equipment designers. A prototype low compliance system is described and tested with performance compared to the conventional design, demonstrating advantages in terms of lower flow error.
- L. Dougherty and S. E. Lister, Eds., The Royal Marsden Hospital Manual of Clinical Nursing Procedures, Wiley-Blackwell, Hoboken, 6th edition, 2004, 264.
- J. D. Bronzino, Ed., Medical Devices and Systems, Taylor & Francis, London, 3rd edition, 2006, 68–2.
- MDA London, Infusion Systems - Device Bulletin, DB 2003(02) March 2003, ISBN 1 84182 6898.
- British Standards Institution, Medical electrical equipment - particular requirements for the safety of infusion pumps and controllers, BS EN 60601-2-24:1998, BS 5724-2.124:1998, IEC 60601-2-24: 1998.
- Bath Institute of Medical Engineering, http://www.bath.ac.uk/bime/evalcentre/pumps.html, accessed Sept. 2009.
- T. A. Neff, C. Fellmann, R. M. Fuechslin, A. C. Gerber, and M. Weiss, “The Panomat P-10 micro-volumetric infusion pump is suitable for continuous drug administration at minimal flow rates,” Can J Anaesth., vol. 49, pp. 1048–1052, 2002.
- M. Weiss, S. Gerber, R. M. Füchslin, and T. A. Neff, “Accurate continuous drug delivery at low infusion rate with a novel microvolumetric infusion pump (MIVP): pump design, evaluation and comparison to the current standard,” Anaesthesia, vol. 59, pp. 1133–1137, 2004.
- R. S. Murphy and S. J. Wilcox, “High risk infusions—accuracy compromised by changes in patient venous pressure,” Journal of Medical Engineering & Technology, vol. 33, no. 6, pp. 470–474, 2009.
- K. H. Krauskopf, J. Rauscher, and L. Brandt, “Disturbance of continuous, pump administration of cardiovascular drugs by hydrostatic pressure,” Der Anaesthesist, vol. 45, no. 5, pp. 449–52, 1996 May.
- H. Kern, A. Kuring, U. Redlich et al., “Downward movement of syringe pumps reduces syringe output,” Br J Anaesth, vol. 86, pp. 828–31, 2001.
- T. A. Neff, J. E. Fischer, G. Schulz, O. Baenziger, and M. Weiss, “Infusion pump performance with vertical displacement: effect of syringe pump and assembly type,” Intensive Care Med., vol. 27, no. 1, pp. 287–91, 2001 Jan.
- M. Weiss, M. I. Hug, T. Neff, and J. Fischer, “Syringe size and flow rate affect delivery from syringe pumps,” Can J Anaesth., vol. 47, no. 10, pp. 929–35, 2000 Oct.
- D. Murray, “Another central line complication: failure of inotrope delivery due to equipment incompatibility,” Anaesthesia, vol. 58, p. 715, 2003.
- T. A. Neff, J. Fischer, S. Fehr, O. Banziger, and M. Weiss, “Start-up delays of infusion syringe pumps,” Paediatr Anaesth, vol. 11, no. 5, pp. 561–565, 2001.
- M. Weiss, O. Baenziger, T. Neff, and S. Fanconi, “Influence of infusion line compliance on drug delivery rate during acute line loop formation,” Intensive Care Med., vol. 26, pp. 776–779, 2000.
- M. Weiss, T. Neff, A. C. Gerber, and J. Fischer, “Impact of infusion line compliance on syringe pump performance,” Paediatr Anaesth., vol. 10, pp. 595–599, 2000.
- R. M. Fuechslin, M. Weiss, R. Duenkel et al., United States Patent Nr 7,614,857 B2 Nov 2009 Assignee: Medinnovation AG Zurich (CH).
Copyright © 2010 Hindawi Publishing Corporation. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.