Research Article  Open Access
José HurtadoAviles, Joaquín RocaGonzález, Konstantsin Sergeevich Kurochka, Jose Manuel SanzMengibar, Fernando SantonjaMedina, "Developing of a Mathematical Model to Perform Measurements of Axial Vertebral Rotation on ComputerAided and Automated Diagnosis Systems, Using Raimondi’s Method", Radiology Research and Practice, vol. 2021, Article ID 5523775, 9 pages, 2021. https://doi.org/10.1155/2021/5523775
Developing of a Mathematical Model to Perform Measurements of Axial Vertebral Rotation on ComputerAided and Automated Diagnosis Systems, Using Raimondi’s Method
Abstract
Introduction. Axial vertebral rotation (AVR) is a basic parameter in the study of idiopathic scoliosis and on physical twodimensional images. Raimondi’s tables are the most used method in the quantification of AVR. The development of computing technologies has enabled the creation of computeraided or automated diagnosis systems (CADx) with which measurement on medical images can be carried out more quickly, simply, and with less intra and interobserver variabilities than manual methods. Although there are several publications dealing with the measurement of AVR in CADx systems, none of them provides information on the equation or algorithm used for the measurement applying Raimondi’s method. Goal. The aim of this work is to perform a mathematical modelling of the data contained in Raimondi’s tables that enable the Raimondi method to be used in digital medical images more precisely and in a more exact manner. Methods. Data from Raimondi’s tables were tabulated on a first step. After this, each column of Raimondi’s tables containing values corresponding to vertebral body width (D) were adjusted to a curve determined by AVR = f (d). Third, representative values of each rotation divided by D were obtained through the equation of each column D. In a fourth step, a regression line was fitted to the data in each row, and from its equation, the mean value of the D/d distribution is calculated (value corresponding to the central column, D = 45). Finally, a curve was adjusted to the obtained data using the least squares method. Summary and Conclusion. Our mathematical equation allows the Raimondi method to be used in digital images of any format in a more accurate and simplified approach. This equation can be easily and freely implemented in any CADx system to quantify AVR, providing a more precise use of Raimondi’s method, as well as being used in traditional manual measurement as it is performed with Raimondi tables.
1. Introduction
Axial vertebral rotation (AVR) is a basic parameter in the study of idiopathic scoliosis in adolescents [1, 2]. Its measurement is necessary to evaluate the severity of scoliosis and quantify the risk of progression [1, 3–7] for the selection of treatment [3, 4, 8, 9] and for the analysis of orthopaedic surgery [1, 6–8, 10, 11].
Twodimensional medical images, especially anteriorposterior and lateral Xrays of the full spine when standing [12–14], continue to be the method of choice for diagnosing and monitoring scoliosis [15].
On physical twodimensional images, Raimondi’s tables [16–18] are the most used method in the quantification of AVR [19–21]. This method possesses a more than acceptable degree of reproducibility between observers, and it is easy to apply [19, 21, 22].
Figure 1 shows a schematic description of the anatomical references used to measure AVR in twodimensional images.
Based on the position of the two closest lateral faces vertebral body points and the two opposite sides of the shadow of the pedicle turned towards the centre of the vertebra in the anteroposterior projection, the software calculates the width of the vertebral body (D) and the distance from the centre of the pedicle to the side of the vertebral body (d). Adapted by permission from Nature/Springer, European Spine Journal, Vrtovec T, Pernus F, and Likar B, a review of methods was conducted for quantitative evaluation of axial vertebral rotation [7].
The development of computing technologies has enabled the creation of computeraided or automated diagnosis systems (CADx) with which manual measurement methods on medical images (for instance, Cobb angle, distances, and wedgings) can be applied more quickly, simply, and with less intra and interobserver variabilities [23–28].
The digital Xrays require a software for the clinicians to state a precise diagnosis and treatment, as well as quantify the followup changes. These kinds of software are usually not designed to quantify specific AIS parameters, or they are not easily accessible due to their complex use or higher cost.
Although there are several publications dealing with the measurement of AVR in CADx systems, none of them provides information on the equation or algorithm used for the measurement applying Raimondi’s method [3, 28–30].
The aim of this work is to perform a mathematical modelling based on the data contained in Raimondi’s tables that enables them to be applied on CADx systems, allowing the Raimondi method to be used in digital medical images more precisely and in a more exact manner.
Data from Raimondi’s tables were tabulated on a first step. After this, each column of Raimondi’s tables containing values corresponding to vertebral body width (D) were adjusted to a curve determined by AVR = f (d). In this mathematical equation, “d” represents each value of the column D. Third, representative values of each rotation divided by D were obtained through the equation of each column D. A representative value for each AVR was obtained this way. The fourth step was to adjust the data included in each row to a regression line, and this equation was used to calculate the average value of the distribution D/d (middle column value = 45). Finally, a curve was adjusted to the obtained data using the least squares method.
The novelties of our work are (a) to provide an adimentional equation to describe in a theoretically way the empirical data tabulated in Raimondi’s tables with more accuracy and less precision error; (b) our mathematical equation allows the Raimondi method to be used in CADx to quantify AVR in medical images without the need for a scale; and (c) this equation can be easily and freely implemented in any CADx system, providing a more precise use of Raimondi’s method than on the traditional printed Xray.
2. Materials and Methods
In Raimondi’s tables, the data pertaining to one same column correspond to a particular vertebral diameter, D, between 20 mm and 70 mm (Table 1). This value is given by the width of the vertebral body D projected on the Xray image.

Within each column, the data correspond to the distance, d, between the centre of the shadow of the pedicle furthest from the edge of the vertebra and said edge, in millimetres.
Each pair of values (D, d) corresponds to a value of AVR between 2° and 60°.
From the distribution of empirical data in Raimondi’s tables, a mathematical model that describes the information contained in them in an ideal manner is sought.
First, the data in each column are analysed separately. Using the least squares method, the equation of a curve,is pursued, such that every pair is as close as possible to this curve.
Using the “Statistix 10” software by Analytical Software, a rational function in the formwhere , , and are the constants, was chosen as the best option.
By adjusting the function model to each set of values in each column of Raimondi’s tables, the adjustment values “pseudo ” reflected in Table 2 are obtained.

The quality of the adjustment of each curve improves as the width of the vertebra, D, increases, reaching a maximum of = 0.9997. The explanation for this may lie in the difficulty of performing a good measurement when this refers to very small distances (Figure 2).
From the value = 48, decreases slightly until at = 68, we have = 0.9993.
It can also be discerned that for = 44, a value of outside of the collection of values described exists due to an atypical datum (Figure 3).
Residuals of the data of various columns from Raimondi’s tables regarding their respective regression curves can be observed in Figures 4(a) and 4(b).
(a)
(b)
When using the Raimondi’s tables on CADx systems, it is necessary for the vertebrae shown in the digital image to be of sufficient size () in order to be able to apply the equation [31]. In addition, the dimensions of variables D and d are given in millimetres.
All pairs of (D, d) values corresponding to a certain AVR are proportional, meaning they should produce ideally equal D/d values.
To avoid having to know the scale of the image to measure AVR, the independent variable will have the form D/d in the calculated equation, such that the value is adimensional.
To calculate the equationthe data in Raimondi’s tables is replaced by other data obtained in the following manner (Table 3):(i)For the set of data in each column, the nonlinear regression curve (equation (2)) is obtained(ii)From the equation of each regression curve, the d values corresponding to the rotations 2°, 4°, 6°, …, 60° are obtained to five decimal places(iii)The value D/d is entered to five decimal places in each cell in the tables

In Table 3, the values of each row are very similar. To find the representative value of each rotation, a regression line is adjusted to the data in each row, and from its equation, the average D/d value of distribution is calculated (value corresponding to the central column, = 45).
Figures 5–7 show that as the width of the vertebra increases, the linearity of the point cloud increases. It can also be observed how the general aspect of the data is quite regular, with the exception of those corresponding to = 60 and = 70, where the pattern followed loses its regularity. Consequently, each regression line is calculated by omitting some of the values as shown in the figures.
Table 4 shows the representative D/d values of each AVR calculated.

3. Results
The least squares method is used to adjust a curve to the data of Table 4. Using the “Statistix 10” software, a rational function in the formwhere , , , , and are the constants, is chosen. The adjustment obtained with this model is = 1.
The equation obtained is
Convergence criterion met after 16 iterations. Residual SS (SSE): 9.722E − 04 Residual MS (MSE): 3.889E − 05 Standard deviation: 6.236E − 03 Degrees of freedom: 25 AICc: −294.46 Pseudo R^{2}: 10000
4. Discussion
AVR is a necessary parameter in being able to correctly evaluate and provide a prognosis for monitor and treat scoliosis [1–7].
Raimondi’s tables are widely used on physical Xrays due to its notable reliability and ease of use in quantifying AVR [21, 32].
We are not aware that until now, a method to calculate ARV by CADx systems has been developed using the Raimondi method.
The data contained in the Raimondi’s tables were obtained experimentally from thousands of measurements on educational and anatomical spines [32]. In consequence, Raimondi recognises that the measurements obtained with his method involve systematic errors [19, 32]. These errors are small in comparison with the real value of the measures made, meaning their presence will not vary the decision on the evaluation or treatment of the patient. However, these errors may propagate along with random errors due to a multitude of variables present in the measuring process such as (A) the loss of information when showing a threedimensional structure on a flat image; (B) the characteristics of the medical image (e.g., existence of noise that hinders the legibility of the image); (C) limitations of the observer (e.g., exactitude in the identification of measurement references points, need for training, and experience); (D) the particular morphological characteristics of the vertebrae (e.g., deformations, the variability of the interpedicular distance between vertebrae); and (E) the position of the patient in the image.
On the other hand, medical images today reach the medical professional on a CD, and Raimondi’s tables cannot be applied to these digital images, when they do not contain information about the scale (that is, the image does not have the scale imprinted on it or the DICOM archive entry does not have this information).
The obtained equation allows the AVR measurement to be implemented in CADx systems with the Raimondi method, allowing more accurate and precise results to be obtained, due to the elimination of the systematic error contained in the Raimondi tables and the use of auxiliary tools that usually incorporate CADx systems, such as enlargement of regions of interest, application of filters, and automatic detection algorithms.
We have not found any previous work that offers an equation or an analogous algorithm to use Raimondi’s method in CADx systems that show it openly. The obtained equation is available to be used freely.
The form of the equation obtained also has the advantage that allows the Raimondi method to be used in digital medical images that do not contain information about its scale (for example, images in TIFF format, in DICOM files that do not contain this information or when it is not printed on the same image).
The limitations of our study are that the programming code to implement our equation in any software is not provided. Our equation was tested using the C++ programming language and OpenCV library, so we can recommend these tools to implement the obtained equation. Also, our equation does not innovate in Raimondi’s procedure but achieves a linear theoretical model based on its empirical data, improving its validity and reliability. Therefore, the highlight of our study is that this equation can be easily and freely implemented in any CADx system, allowing the use of the Raimondi method in digital medical images, whether or not they contain the scale, in a more exact and precise way.
5. Summary and Conclusion
A mathematical equation has been calculated that can be implemented in CADx systems from the mathematical modelling of the empirical data tabulated in Raimondi’s tables.
This allows the Raimondi method to be used in digital images of any format more accurately, simply, and quickly, by eliminating the small systematic error existing in the discrete values of the tables and being able to use in conjunction with auxiliary tools in the CADx system such as zooming over regions of interest, algorithms for enhancing medical image structures, adjusting brightness, and contrast.
The equation can be easily and freely implemented in any CADx system.
Nomenclature
a, b, c, d, e:  Constants 
x:  Variable 
D:  The value given by the width of vertebral body projected on the Xray image, in millimetres 
d:  The distance between the centre of the shadow of the pedicle furthest from the edge of the vertebra and said edge, in millimetres 
R^{2}:  The value of the adjustment obtained with an equation to a set of data. 
Data Availability
The data used to support the findings of this study are available from the corresponding author upon request.
Conflicts of Interest
The authors declare that they have no conflicts of interest.
References
 S. P. Mohanty, M. P. Kanhangad, and A. A Gullia, “Curve severity and apical vertebral rotation and their association with curve flexibility in adolescent idiopathic scoliosis,” Musculoskeletal Surgery, 2020. View at: Publisher Site  Google Scholar
 M. Mangone, P. Raimondi, M. Paoloni et al., “Vertebral rotation in adolescent idiopathic scoliosis calculated by radiograph and back surface analysisbased methods: correlation between the Raimondi method and rasterstereography,” European Spine Journal, vol. 22, no. 2, pp. 367–37110, 2013. View at: Publisher Site  Google Scholar
 S. M. Eijgenraam, T. F. M. Boselie, J. M. Sieben et al., “Development and assessment of a digital Xray software tool to determine vertebral rotation in adolescent idiopathic scoliosis,” The Spine Journal, vol. 17, no. 2, pp. 260–265, 2017. View at: Publisher Site  Google Scholar
 Q. N. Vo, E. H. Lou, and L. H. Le, “Measurement of axial vertebral rotation using threedimensional ultrasound images,” Scoliosis, vol. 10, no. 2, p. S7, 2015. View at: Publisher Site  Google Scholar
 M.L. Nault, J.M. MacThiong, M. RoyBeaudry et al., “Threedimensional spinal morphology can differentiate between progressive and nonprogressive patients with adolescent idiopathic scoliosis at the initial presentation,” Spine, vol. 39, no. 10, pp. E601–E606, 2014. View at: Publisher Site  Google Scholar
 B. B. Carlson, D. C. Burton, and M. A. Asher, “Comparison of trunk and spine deformity in adolescent idiopathic scoliosis,” Scoliosis, vol. 8, no. 1, p. 2, 2013. View at: Publisher Site  Google Scholar
 T. Vrtovec, F. Pernuš, and B. Likar, “A review of methods for quantitative evaluation of axial vertebral rotation,” European Spine Journal, vol. 18, no. 8, pp. 1079–1090, 2009. View at: Publisher Site  Google Scholar
 T. R. Kuklo, B. K. Potter, and L. G. Lenke, “Vertebral rotation and thoracic torsion in adolescent idiopathic scoliosis,” Journal of Spinal Disorders & Techniques, vol. 18, no. 2, pp. 139–147, 2005. View at: Publisher Site  Google Scholar
 Y. Tamura, N. Sugano, T. Sasama et al., “Surfacebased registration accuracy of CTbased imageguided spine surgery,” European Spine Journal, vol. 14, no. 3, pp. 291–297, 2005. View at: Publisher Site  Google Scholar
 J. Ochi, F. Zhu, L. Xu et al., “Comparison of the aorta impingement risks between thoracolumbar/lumbar curves with different convexities in adolescent idiopathic scoliosis: a computed tomography study,” European Spine Journal, vol. 21, no. 10, pp. 2043–2049, 2012. View at: Publisher Site  Google Scholar
 Y. Petit, C.É. Aubin, and H. Labelle, “Spinal shape changes resulting from scoliotic spine surgical instrumentation expressed as intervertebral rotations and centers of rotation,” Journal of Biomechanics, vol. 37, no. 2, pp. 173–180, 2004. View at: Publisher Site  Google Scholar
 N. D. Fletcher and R. W. Bruce, “Early onset scoliosis: current concepts and controversies,” Current Reviews in Musculoskeletal Medicine, vol. 5, no. 2, pp. 102–110, 2012. View at: Publisher Site  Google Scholar
 T. R. Kuklo, B. K. Potter, T. M. Schroeder, and M. F. O’Brien, “Comparison of manual and digital measurements in adolescent idiopathic scoliosis,” Spine, vol. 31, no. 11, pp. 1240–1246, 2006. View at: Publisher Site  Google Scholar
 M. Yazici, E. R. Acaroglu, A. Alanay, V. Deviren, A. Cila, and A. Surat, “Measurement of vertebral rotation in standing versus supine position in adolescent idiopathic scoliosis,” Journal of Pediatric Orthopaedics, vol. 21, no. 2, pp. 252–256, 2001. View at: Publisher Site  Google Scholar
 Sociedad Argentina de Pediatría, “Sociedad Argentina de ortopedia y traumatología infantil (saoti) y sociedad Argentina de patología de la columna vertebral (SAPCV),” Consenso de escoliosis idiopática del adolescente. Archivos Argentinos de Pediatría, vol. 114, no. 6, pp. 585–594, 2016. View at: Google Scholar
 P. Raimondi, “La misurazione della rotazione vertebrale sul piano orizzontale con il Metodo Raimondi,” Scienze Motorie, vol. 2, pp. 11–16, 2002. View at: Google Scholar
 P. Raimondi, G. Costanzo, A. P. Santaroni, and F. Bizzarri, “Ipotesi patogenetica sul ruolo di percezione e postura nello sviluppo delle scoliosi,” Résonances Européennes du Rachis, vol. 20, pp. 883–889, 1998. View at: Google Scholar
 P. Raimondi, G. Costanzo, and A. P. Santaroni, “Evaluation Chiffre de la rotation vertebrale par la mèthode de Paolo Raimondi,” Kinésithérapie Scientifique, vol. 358, pp. 19–22, 1996. View at: Google Scholar
 V. Prosperini, P. Raimondi, E. Martinelli, and F. Bizzarri, “Misurazione della rotazione vertebrale patologica su radiografie di piccolo e grande formato con il Metodo Raimondi,” Chinesiologia, vol. 1, pp. 23–27, 2010. View at: Google Scholar
 S. Kadoury, F. Cheriet, M. Beauséjour, I. A. Stokes, S. Parent, and H. Labelle, “A threedimensional retrospective analysis of the evolution of spinal instrumentation for the correction of adolescent idiopathic scoliosis,” European Spine Journal, vol. 18, no. 1, pp. 23–37, 2009. View at: Publisher Site  Google Scholar
 H.R. Weiss, “Measurement of vertebral rotation: perdriolle versus Raimondi,” European Spine Journal, vol. 4, no. 1, pp. 34–38, 1995. View at: Publisher Site  Google Scholar
 H. L. A. Defino and P. H. M. de Araújo, “Comparative study of the measurements of the vertebral rotation using Nash&Moe and Raimondi methods,” Acta Ortopédica Brasileira, vol. 3, no. 12, pp. 167–173, 2004. View at: Publisher Site  Google Scholar
 A. Safari, H. Parsaei, A. Zamani, and B. Pourabbas, “A semiautomatic algorithm for estimating cobb angle,” Journal of Biomedical Physics and Engineering, vol. 9, no. 3, pp. 317–326, 2019. View at: Publisher Site  Google Scholar
 J. Yang, K. Zhang, H. Fan et al., “Development and validation of deep learning algorithms for scoliosis screening using back images,” Communications Biology, vol. 2, no. 1, p. 390, 2019. View at: Publisher Site  Google Scholar
 Y. Pan, Q. Chen, T. Chen et al., “Evaluation of a computeraided method for measuring the Cobb angle on chest Xrays,” European Spine Journal, vol. 28, no. 12, pp. 3035–3043, 2019. View at: Publisher Site  Google Scholar
 L. Wang, Q. Xu, S. Leung, J. Chung, B. Chen, and S. Li, “Accurate automated Cobb angles estimation using multiview extrapolation net,” Medical Image Analysis, vol. 58, p. 101542, 2019. View at: Publisher Site  Google Scholar
 D. Forsberg, C. Lundström, M. Andersson, L. Vavruch, H. Tropp, and H. Knutsson, “Fully automatic measurements of axial vertebral rotation for assessment of spinal deformity in idiopathic scoliosis,” Physics in Medicine and Biology, vol. 58, no. 5, pp. 1775–1787, 2013. View at: Publisher Site  Google Scholar
 T. Vrtovec, R. Vengust, B. Likar, and F. Pernuš, “Analysis of four manual and a computerized method for measuring axial vertebral rotation in computed tomography images,” Spine, vol. 35, no. 12, pp. E535–E541, 2010. View at: Publisher Site  Google Scholar
 A. P. Pinheiro, M. C. Tanure, and A. S. Oliveira, “Validity and reliability of a computer method to estimate vertebral axial rotation from digital radiographs,” European Spine Journal, vol. 19, no. 3, pp. 415–420, 2010. View at: Publisher Site  Google Scholar
 J. Zhang, E. Lou, D. L. Hill et al., “Computeraided assessment of scoliosis on posteroanterior radiographs,” Medical & Biological Engineering & Computing, vol. 48, no. 2, pp. 185–195, 2010. View at: Publisher Site  Google Scholar
 M. D. Rigo and M. Villagrasa, “Technical error of vertebral rotation measurement directly on the computer screen, according to Raimondi method,” Scoliosis, vol. 5, no. S1, p. O14, 2010. View at: Publisher Site  Google Scholar
 P. Raimondi and V. Prosperini, “La misurazione della rotazione vertebrale nelle scoliosi,” Esperienze valutative e commento. chinesiologia n., vol. 3, pp. 22–26, 2007. View at: Google Scholar
Copyright
Copyright © 2021 José HurtadoAviles et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.