Research Article | Open Access
Elavarasan Subramani, Himanish Basu, Shyam Thangaraju, Sucheta Dandekar, Deepak Mathur, Koel Chaudhury, "Rotational Dynamics of Optically Trapped Human Spermatozoa", The Scientific World Journal, vol. 2014, Article ID 154367, 7 pages, 2014. https://doi.org/10.1155/2014/154367
Rotational Dynamics of Optically Trapped Human Spermatozoa
Introduction. Optical trapping is a laser-based method for probing the physiological and mechanical properties of cells in a noninvasive manner. As sperm motility is an important criterion for assessing the male fertility potential, this technique is used to study sperm cell motility behavior and rotational dynamics. Methods and Patients. An integrated optical system with near-infrared laser beam has been used to analyze rotational dynamics of live sperm cells from oligozoospermic and asthenozoospermic cases and compared with controls. Results. The linear, translational motion of the sperm is converted into rotational motion on being optically trapped, without causing any adverse effect on spermatozoa. The rotational speed of sperm cells from infertile men is observed to be significantly less as compared to controls. Conclusions. Distinguishing normal and abnormal sperm cells on the basis of beat frequency above 5.6 Hz may be an important step in modern reproductive biology to sort and select good quality spermatozoa. The application of laser-assisted technique in biology has the potential to be a valuable tool for assessment of sperm fertilization capacity for improving assisted reproductive technology.
An optical tweezer (optical trap) is a biophotonic device that uses the momentum generated by a sharp spatial gradient of laser intensity to trap micron-sized particles. This physical phenomenon was first described by Ashkin more than four decades ago , and the technique was first used for trapping live cells twenty-five years ago . The sharp spatial intensity gradient is accomplished by means of a microscope objective (usually 100X) with high numerical aperture, and the trapping strength that can be achieved depends upon the optical geometry of the trap and the object that is sought to be trapped . Optical tweezers are increasingly being used as a noninvasive method to study the mechanical properties, behavior, and physiology of cells and also to spatially manipulate single cells [4–6]. Characterization of the dynamics of the rotational motion of flagellar mutants in Chlamydomonas reinhardtii is one recent example of the utility of this technique  for studying live cells kept under physiological conditions.
Study of spermatozoa dynamics, specifically the generation of motility forces, has been a subject of contemporary interest [8–11]. Optical traps have been used to probe the force generated by motile spermatozoa and their variations in abnormal sperm cells [12–14]. More recently, laser tweezers have been used in combination with computer tracking software and robotic technology for high-throughput sorting and measurement of sperm characteristics [15–19]. Measurement of sperm motility and swimming force has generated considerable interest amongst clinical researchers, as these parameters are useful for determining fertilization rates. A correlation has been observed between sperm swimming force and swimming speed . Such relationships between swimming forces and swimming speeds of primate species, including chimpanzee, rhesus macaque, human, and gorilla, have also been used to deduce the evolutionary patterns of sperm in relation to mating patterns of different primates .
Classically optical trapping has been carried out using near-infrared (nIR) lasers, with spectral ranges of 700–1200 nm, as these wavelengths are only weakly absorbed by live cells [22, 23]. Although this spectral range is assumed to be safe and noninvasive to the cells, there are concerns that the laser light, per se, might have damaging effects on the certain types of living specimens, such as DNA damage, loss of membrane potential, cloning efficiency reduction, and loss of viability [24–27].
Although various properties of spermatozoa including swimming speed and swimming force before and after optical trapping have, to a certain extent, been studied in different mammalian species, no reports exist on the study of rotational dynamics of a single live sperm cell within a trap. In the present study, we report real-time changes in the rotational dynamics of optically trapped, abnormal human spermatozoa, including oligozoospermic and asthenozoospermic cells, as compared to normal sperm cells. Furthermore, their progressive motility before and after trapping is systematically investigated.
2. Materials and Methods
2.1. Subject Selection and Sperm Collection
Asthenozoospermic () and oligozoospermic () semen samples were randomly collected from subjects (28–40 years) reporting at the King Edward Memorial (KEM) Hospital, Mumbai, for male factor infertility. The control group consisted of proven fertile men () whose partners had delivered healthy babies during the last six months without assisted reproductive technology (ART). Semen ejaculates were collected after three to five days of sexual abstinence in sterile containers and were examined in the laboratory within 1 h as per World Health Organization (WHO) 2010 guidelines . Fifty sperm cells from each sample were optically trapped for analysis. Institutional approval was acquired from the Research Ethics Committee to carry out this study and written informed consent was obtained from all couples. Normal semen parameters as per WHO guidelines (2010) are sperm concentration >15 million/mL, motility >32%, and morphology >4%.
2.2. Optical Tweezers Set-Up
The optical trap set-up that was used in the present studies has been extensively described earlier  and is schematically represented in Figure 1. Briefly, a continuous wave, 1064 laser, beam from a diode-pumped Nd : YVO4 laser is coupled to a fluorescence-equipped inverted microscope (NIKON TE 2000U). To achieve trapping, the laser was focused through a 100X microscope objective with numerical aperture 1.3. To avoid possible damage to the trapped sperm cells, the power of the laser incident on the sperm samples was limited to be less than 20 mW. The dynamics of the live sperm cells under the trap were visualized and captured in real-time using a 25-frame/second CCD camera interfaced with a computer. Such videos were then analyzed frame by frame by Image J software.
2.3. Trapping Efficiency and Rotational Motion
The effect of trapping on the sperm motility is quantified hereafter as the trapping efficiency of our optical tweezers set-up for sperm cells. This was calculated using the following formula :
If the trap has no effect and the velocities before and after trapping remain unchanged, the trapping efficiency is 100%. However any difference between the pretrap and posttrap velocities will cause the efficiency to decrease.
As alluded to previously, upon optical trapping not only the translational motion of the sperm is halted, but also a rotational motion is established. Such motion has been studied in terms of the number of cells that are rotating, their rotational frequency and the beat frequency of the sperm tail through frame by frame analysis of the recorded real-time videos.
2.4. Statistical Analysis
Data for each set of cells were analyzed using paired and unpaired t-test. Analysis, discrimination, and classification of data were performed with Graphpad prism, Origin, MedCalc, and SPSS software. Statistical significance of the test was defined as and .
The optical tweezers set-up was used to analyze the rotational dynamics of various categories of sperm cells. As already noted, normal linear translational motion of all motile spermatozoa was halted upon trapping. Cells possessing a functional tail began to rotate under the trap in either a clockwise or anticlockwise direction (Figure 2(a)), with the rotational direction being stochastic. The rotational motion is due to the torque developed by the sperm tail force acting tangentially to the force generated within the optical trap which pins down the head of the sperm (see accompanying video in Supplementary Materials). The rotational speed of oligozoospermic and asthenozoospermic sperm cells was found to be significantly less when compared with controls (Figure 2(b)). Receiver operating characteristic (ROC) curve analysis indicated the beat frequency cut-off value to be 5.6 Hz for differentiating normal and abnormal sperm cells (both oligozoospermic and asthenozoospermic) (Figures 3(a) and 3(b)). Sensitivity and specificity values for differentiating abnormal spermatozoa from normal sperm cells at various beat frequencies are presented in Figures 3(c) and 3(d) for oligozoospermic and asthenozoospermic cases, respectively. The sensitivity and specificity for cut-off value of oligozoospermic and asthenozoospermic cases are 72% and 70% and 86% and 70%, respectively. The rotational speed or beat frequency of various categories of sperm cells also positively correlated () with swimming speed of the sperm (Figure 4(a)). Swimming speeds of oligozoospermic and asthenozoospermic sperm cells were observed to be significantly lower than those of controls (Figure 4(b)). In addition, no significant differences were found between the pretrap and posttrap swimming speeds of oligozoospermic, asthenozoospermic, and normal sperm cells (Figure 4(b)). The trapping efficiency values of the tweezers for oligozoospermic, asthenozoospermic, and normal sperm cells were determined to be 96.2%, 92.2%, and 95.9%, respectively by using formula (1).
Sperm motility is usually evaluated in research and infertility clinics by computer-assisted sperm analysis (CASA) systems which has been commercially available since the mid-1980s [31–33]. Several reports indicate that optical tweezers may be an acceptable alternative for the measurement of sperm swimming force, speed, energetics, and mitochondrial membrane potential [8–10]. Application of this technology not only leads to better understanding of sperm characteristics like motility and energetics, but also directs to inclusive assays for abnormal sperm behavior in infertility and fertilizing ability of sperms.
Upon trapping a sperm cell, we observed conversion of translational motion into rotational motion. We speculate that, following trapping, a reversible change in the molecular components of the mitochondrial apparatus is responsible for the transition from translational to rotational motion. This change gives rise to an enhanced polarizability that, in turn, interacts with the electric field associated with the laser light to generate a torque. The torque that is generated depends on the magnitude of the optical field and, hence, on the laser intensity. Earlier work in our laboratory with live red blood cells has established that rotational motion is dependent on incident laser light intensity (and, hence, on optical field) [34–37].
The sperm, on being optically trapped, rotates at a certain frequency called the beat frequency. This frequency is, in turn, dependent upon the force that is generated by the beating motion of the cell and it correlates well with the sperm swimming speed. The decreased rotational or beat frequency of trapped abnormal sperm cells indicates alteration in the flagellar apparatus or proteins. Further, beat frequency with a cut-off value of 5.6 Hz was found to differentiate oligozoospermic and asthenozoospermic sperm cells from those of controls. However, the area under the curve (AUC) analysis shows that this technique works better for the latter group (AUC = 0.8) than the former group (AUC = 0.714) (Figure 3(e)).
It is necessary to document swimming speed of sperm before and after optical trapping to monitor the sperm physiology before, during, and after optical trapping. Hence, the findings of our present study suggest that the optical tweezers set-up with nIR laser beam does not cause damage to the spermatozoa (Figure 4(b)). This is in good agreement with the findings of König et al.  who concluded that “optical micromanipulation of spermatozoa” with long wavelength nIR laser beam causes significantly less damage as compared to ultraviolet exposure. In contrast, long exposure to the laser trap is reported to have an adverse effect on sperm swimming speed and mitochondrial membrane potential due to laser-induced rise in temperature [24, 25]. Also, depolarization of mitochondria has been observed with prolonged exposure of domestic dog spermatozoa to optical tweezers .
The “WHO manual for the examination of human semen and sperm (semen-) cervical mucus interaction” (WHO 2010) is considered to be the gold standard for assessing semen quality. An interesting observation worth mentioning is that although, as per WHO guidelines, oligozoospermia refers to deficiency in the number of spermatozoa in semen with motility of the sperm cells remaining normal, beat frequency, an alternative parameter representing sperm motility, was found to be significantly less in oligozoospermia as compared to controls (Figure 2(b)). This is the first time that low motility is seen to be associated with oligozoospermic sperm samples (Figure 4(b)). This observation definitely warrants further investigation. Following validation in a larger sample size, this valuable information can be considered while defining the WHO sperm parameters which, in turn, describe sperm quality.
In conclusion, our work offers indications that optical tweezers are developing into an important tool for optically selecting and sorting sperm cells based on their rotational dynamics and motility. Although further work needs to be undertaken to gain deeper insights into the physics that determines the conversion of a trapped sperm’s translational motion into rotational motion, our study shows that a start may be made to utilize this technique to select good quality sperm cells that demonstrate a beat frequency above 5.6 Hz on being optically trapped. Moreover, by studying the pre and posttrap rotational dynamics we have shown that the nIR laser beam of optical tweezers does not damage the normal physiological characteristics of the trapped single live sperm cell. The system provides a multiparametric set of measurements on single live spermatozoa that gives an improved understanding of sperm swimming speed. We suggest that this technique holds considerable potential in selecting good quality sperm cells from infertile men for use in ART procedures such as in-vitro fertilization (IVF) and intra cytoplasmic sperm injection (ICSI), where the sperm quality is a decisive factor in deciding the quality of the resultant embryo.
Conflict of Interests
The authors do not have any conflict of interests and this paper is not submitted elsewhere.
Elavarasan Subramani is thankful to Dr. Muthu Rama Krishnan, Biomedical Engineering Centre, Ngee Ann Polytechnic, Singapore, for his support in data analysis.
Real-time video clip showing the rotational dynamics of a trapped sperm cell. The conversion of the translational motion into rotational motion upon trapping a sperm cell is seen.
- A. Ashkin, “Acceleration and trapping of particles by radiation pressure,” Physical Review Letters, vol. 24, no. 4, pp. 156–159, 1970.
- A. Ashkin, J. M. Dziedzic, and T. Yamane, “Optical trapping and manipulation of single cells using infrared laser beams,” Nature, vol. 330, no. 6150, pp. 769–771, 1987.
- A. Ashkin, “Forces of a single-beam gradient laser trap on a dielectric sphere in the ray optics regime,” Methods in Cell Biology, no. 55, pp. 1–27, 1998.
- K. Bambardekar, A. K. Dharmadhikari, J. A. Dharmadhikari, D. Mathur, and S. Sharma, “Measuring erythrocyte deformability with fluorescence, fluid forces, and optical trapping,” Journal of Biomedical Optics, vol. 13, no. 6, Article ID 064021, 2008.
- K. Bambardekar, J. A. Dharmadhikari, A. K. Dharmadhikari et al., “Shape anisotropy induces rotations in optically trapped red blood cells,” Journal of Biomedical Optics, vol. 15, no. 4, Article ID 041504, 2010.
- A. Ghosh, S. Sinha, J. A. Dharmadhikari et al., “Euler buckling-induced folding and rotation of red blood cells in an optical trap,” Physical Biology, vol. 3, no. 1, pp. 67–73, 2006.
- J. S. D'Souza, M. Gudipati, J. A. Dharmadhikari et al., “Flagella-generated forces reveal gear-type motor in single cells of the green alga, Chlamydomonas reinhardtii,” Biochemical and Biophysical Research Communications, vol. 380, no. 2, pp. 266–270, 2009.
- E. Araujo Jr., Y. Tadir, P. Patrizio et al., “Relative force of human epididymal sperm,” Fertility and Sterility, vol. 62, no. 3, pp. 585–590, 1994.
- L. Z. Shi, J. M. Nascimento, C. Chandsawangbhuwana, E. L. Botvinick, and M. W. Berns, “An automatic system to study sperm motility and energetics,” Biomedical Microdevices, vol. 10, no. 4, pp. 573–583, 2008.
- J. L. Nascimento, E. L. Botvinick, L. Z. Shi, B. Durrant, and M. W. Berns, “Analysis of sperm motility using optical tweezers,” Journal of Biomedical Optics, vol. 11, no. 4, Article ID 044001, 2006.
- N. Hyun, C. Chandsawangbhuwana, Q. Zhu, L. Z. Shi, C. Yang-Wong, and M. W. Berns, “Effects of viscosity on sperm motility studied with optical tweezers,” Journal of Biomedical Optics, vol. 17, no. 2, Article ID 025005, 2012.
- Z. N. Dantas, E. Araujo Jr., Y. Tadir, M. W. Berns, M. J. Schell, and S. C. Stone, “Effect of freezing on the relative escape force of sperm as measured by a laser optical trap,” Fertility and Sterility, vol. 63, no. 1, pp. 185–188, 1995.
- P. Patrizio, Y. Liu, G. J. Sonek, M. W. Berns, and Y. Tadir, “Effect of pentoxifylline on the intrinsic swimming forces of human sperm assessed by optical tweezers,” Journal of Andrology, vol. 21, no. 5, pp. 753–756, 2000.
- J. M. Nascimento, L. Z. Shi, C. Chandsawangbhuwana et al., “Use of laser tweezers to analyze sperm motility and mitochondrial membrane potential,” Journal of Biomedical Optics, vol. 13, no. 1, Article ID 014002, 2008.
- B. Shao, L. Z. Shi, J. M. Nascimento et al., “High-throughput sorting and analysis of human sperm with a ring-shaped laser trap,” Biomedical Microdevices, vol. 9, no. 3, pp. 361–369, 2007.
- M. M. Garcia, A. T. Ohta, T. J. Walsh et al., “A noninvasive, motility independent, sperm sorting method and technology to identify and retrieve individual viable nonmotile sperm for intracytoplasmic sperm injection,” Journal of Urology, vol. 184, no. 6, pp. 2466–2472, 2010.
- Y. Tadir, W. H. Wright, O. Vafa, T. Ord, R. H. Asch, and M. W. Berns, “Micromanipulation of sperm by a laser generated optical trap,” Fertility and Sterility, vol. 52, no. 5, pp. 870–873, 1989.
- L. Shi, B. Shao, T. Chen, and M. Berns, “Automatic annular laser trapping: a system for high-throughput sperm analysis and sorting,” Journal of Biophotonics, vol. 2, no. 3, pp. 167–177, 2009.
- L. Z. Shi, J. Nascimento, C. Chandsawangbhuwana, M. W. Berns, and E. L. Botvinick, “Real-time automated tracking and trapping system for sperm,” Microscopy Research and Technique, vol. 69, no. 11, pp. 894–902, 2006.
- T. Kasai, K. Ogawa, K. Mizuno et al., “Relationship between sperm mitochondrial membrane potential, sperm motility, and fertility potential,” Asian Journal of Andrology, vol. 4, no. 2, pp. 97–103, 2002.
- J. M. Nascimento, L. Z. Shi, S. Meyers et al., “The use of optical tweezers to study sperm competition and motility in primates,” Journal of the Royal Society Interface, vol. 5, no. 20, pp. 297–302, 2008.
- K. König, L. Svaasand, Y. Liu et al., “Determination of motility forces of human spermatozoa using an 800 nm optical trap,” Cellular and Molecular Biology (Noisy-le-grand), vol. 42, no. 4, pp. 501–509, 1996.
- K. König, Y. Tadir, P. Patrizio, M. W. Berns, and B. J. Tromberg, “Effects of ultraviolet exposure and near infrared laser tweezers on human spermatozoa,” Human Reproduction, vol. 11, no. 10, pp. 2162–2164, 1996.
- S. K. Mohanty, M. Sharma, and P. K. Gupta, “Generation of ROS in cells on exposure to CW and pulsed near-infrared laser tweezers,” Photochemical and Photobiological Sciences, vol. 5, no. 1, pp. 134–139, 2006.
- A. Mei, E. Botvinick, and M. Berns, “Monitoring sperm mitochondrial respiration response in a laser trap using ratiometric fluorescence,” in Optical Trapping and Optical Micromanipulation II, Proceedings of the SPIE 5930, pp. 1–11, San Diego, Calif, USA, August 2005.
- S. K. Mohanty, A. Rapp, S. Monajembashi, P. K. Gupta, and K. O. Greulich, “Comet assay measurements of DNA damage in cells by laser microbeams and trapping beams with wavelengths spanning a range of 308 nm to 1064 nm,” Radiation Research, vol. 157, no. 4, pp. 378–385, 2002.
- F. Martinez-Pastor, A. Johannisson, J. Gil et al., “Use of chromatin stability assay, mitochondrial stain JC-1, and fluorometric assessment of plasma membrane to evaluate frozen-thawed ram semen,” Animal Reproduction Science, vol. 84, no. 1-2, pp. 121–133, 2004.
- World Health Organization, WHO Laboratory Manual for the Examination and Processing of Human Semen, WHO Press, Geneva, Switzerland, 5th edition, 2010.
- H. Basu, A. K. Dharmadhikari, J. A. Dharmadhikari, S. Sharma, and D. Mathur, “Tank treading of optically trapped red blood cells in shear flow,” Biophysical Journal, vol. 101, no. 7, pp. 1604–1612, 2011.
- M. Muthu Rama Krishnan, A. Choudhary, C. Chakraborty, A. K. Ray, and R. R. Paul, “Texture based segmentation of epithelial layer from oral histological images,” Micron, vol. 42, no. 6, pp. 632–641, 2011.
- R. J. Aitken, F. S. Best, P. Warner, and A. Templeton, “A prospective study of the relationship between semen quality and fertility in cases of unexplained infertility,” Journal of Andrology, vol. 5, no. 4, pp. 297–303, 1984.
- W. Krause, “Computer-assisted semen analysis systems: comparison with routine evaluation and prognostic value in male fertility and assisted reproduction,” Human Reproduction, vol. 10, no. 1, pp. 60–66, 1995.
- L. Larsen, T. Scheike, T. K. Jensen et al., “Computer-assisted semen analysis parameters as predictors for fertility of men from the general population,” Human Reproduction, vol. 15, no. 7, pp. 1562–1567, 2000.
- J. A. Dharmadhikari, S. Roy, A. K. Dharmadhikari, S. Sharma, and D. Mathur, “Torque-generating malaria-infected red blood cells in an optical trap,” Optics Express, vol. 12, no. 6, pp. 1179–1184, 2004.
- S. Roy, J. Dharmadhikari, A. Dharmadhikari, D. Mathur, and S. Sharma, “Study of P. falciparum-infected erythrocytes and induced anisotropies under optical and fluid forces,” Journal of Vector Borne Diseases, vol. 44, no. 1, pp. 23–32, 2007.
- E. Zachariah, A. Bankapur, C. Santhosh, M. Valiathan, and D. Mathur, “Probing oxidative stress in single erythrocytes with Raman Tweezers,” Journal of Photochemistry and Photobiology B, vol. 100, no. 3, pp. 113–116, 2010.
- A. Bankapur, E. Zachariah, S. Chidangil, M. Valiathan, and D. Mathur, “Raman tweezers spectroscopy of live, single red and white blood cells,” PLoS ONE, vol. 5, no. 4, Article ID e10427, 2010.
Copyright © 2014 Elavarasan Subramani et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.