- About this Journal ·
- Abstracting and Indexing ·
- Aims and Scope ·
- Annual Issues ·
- Article Processing Charges ·
- Articles in Press ·
- Author Guidelines ·
- Bibliographic Information ·
- Citations to this Journal ·
- Contact Information ·
- Editorial Board ·
- Editorial Workflow ·
- Free eTOC Alerts ·
- Publication Ethics ·
- Reviewers Acknowledgment ·
- Submit a Manuscript ·
- Subscription Information ·
- Table of Contents
International Journal of Dentistry
Volume 2009 (2009), Article ID 634567, 20 pages
Use of Cone Beam Computed Tomography in Endodontics
1Division of Radiology and Imaging Science, Department of Surgical and Hospital Dentistry, The University of Louisville School of Dentistry, 501 South Preston Street, Louisville, KY 40292, USA
2Department of Endodontics, School of Dental Medicine, University of Pennsylvania, Philadelphia, PA 19103, USA
3Kodak Dental Imaging/Practiceworks, 1765 The Exchange, Atlanta, GA 30339, USA
Received 16 September 2009; Accepted 15 December 2009
Academic Editor: Eiichiro I. Ariji
Copyright © 2009 William C. Scarfe et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Cone Beam Computed Tomography (CBCT) is a diagnostic imaging modality that provides high-quality, accurate three-dimensional (3D) representations of the osseous elements of the maxillofacial skeleton. CBCT systems are available that provide small field of view images at low dose with sufficient spatial resolution for applications in endodontic diagnosis, treatment guidance, and posttreatment evaluation. This article provides a literature review and pictorial demonstration of CBCT as an imaging adjunct for endodontics.
Since Kells first reported the usefulness of visualizing a lead wire in a root canal on a “radiogram” in establishing the length of a root canal in 1899 [1, 2], radiography has been a pivotal tool in the practice of endodontics. Almost a century later, building on the pioneering efforts of those using conventional computed tomography (CT) and micro-CT, the introduction of maxillofacial CBCT in 1996 provided the first clinically practical technology demonstrating application of 3D imaging for endodontic considerations .
2. Role of Imaging in Endodontics
Radiography is essential to successful diagnosis of odontogenic and nonodontogenic pathoses, treatment of the pulp chamber and canals of the root of a compromised tooth via intracoronal access, biomechanical instrumentation, final canal obturation, and assessment of healing. Imaging serves at all stages in endodontics .(1)Preoperative Assessment. Imaging achieves visualization of dental and alveolar hard tissue morphology and pathologic alterations to assist correct diagnosis. It provides information on the morphology of the tooth including location and number of canals, pulp chamber size and degree of calcification, root structure, direction and curvature, fractures, iatrogenic defects, and the extent of dental caries. The effects of periradicular and periapical disease can be determined, including the degree of root resorption and characteristics of periapical osteolysis. Larger lesions, only determined by imaging, may necessitate adjunctive surgical procedures in addition to conventional intracanal therapy. Diagnostic radiographs help predict the potential for complications, permit root fracture detection, and demonstrate periapical lesions. (2)Intraoperative. During therapy two intraoral periapical images may be performed. The first is a “working” radiograph achieved by placement of a metallic file(s) into the root canal(s) to a length that approximates that of the root as radiological and anatomic root apexes are almost never coincident. This ensures that mechanical debridement of the intracanal contents extends to the apical terminus of the canal and that obturation is dense, homogeneous, and contained within the root canal system. In addition, prior to final obturation, a “final” or pre-condensation radiograph is made to assure proper fitting of the master cone.(3)Postoperative. A “postoperative” radiograph immediately after root canal obturation is made to assess the sealing condensation and containment of the root canal filling material within the root canal system. In cases where periradicular healing is incomplete, it acts as a baseline for assessment of healing in the medium and potentially long term. Imaging is important in evaluating the results of previous therapy, delayed healing, evaluating potential obstacles to retreatment, as well as surgical considerations .
3. Limitations of Conventional 2D Imaging
Intraoral radiography is based on the transmission, attenuation, and recording of X-rays on an analog film or digital receptor, and requires optimized geometric configuration of the X-ray generator, tooth, and sensor to provide an accurate projection of the tooth. The image produced is a two-dimensional (2D) representation of a three-dimensional (3D) object. If any component of the imaging chain process is compromised, the resulting image may demonstrate exposure or geometric errors  and be suboptimal. 3D characteristics such as complex dental anatomy and surrounding structures can make interpretation of 2D “shadows” difficult and can contribute to nonhealing of endodontic cases.
Success in endodontics is assessed in healing of the periapical bone adjacent to obturated canals. Goldman et al.  showed that in evaluating healing of periapical lesions using 2D periapical radiographs there was only 47% agreement between six examiners. Goldman et al.  also reported that when those same examiners evaluated the same films at two different times, they only had 19%–80% agreement between the two evaluations.
4. Cone Beam Computerized Tomography
In fields of dentistry where 3D imaging is necessary, CBCT is considered by some to be the standard of care [9–14]. CBCT is accomplished by using a rotating gantry to which an X-ray source and detector are fixed. A divergent pyramidal- or cone-shaped source of ionizing radiation is directed through the middle of the area of interest onto an area X-ray detector on the opposite side of the patient. The X-ray source and detector rotate around a fixed fulcrum within the region of interest (ROI). During the exposure sequence hundreds of planar projection images are acquired of the field of view (FOV) in an arc of at least . In this single rotation, CBCT provides precise, essentially immediate and accurate 3D radiographic images. As CBCT exposure incorporates the entire FOV, only one rotational sequence of the gantry is necessary to acquire enough data for image reconstruction. CBCT is a complementary modality for specific applications rather than a replacement for 2D imaging modalities [9–13].
The Food and Drug Administration (FDA) approved the first CBCT unit for dental use in the United States in March 8, 2001—the NewTom DVT 9000 (Quantitative Radiology srl, Verona, Italy). FDA approval for three more CBCT units quickly followed in 2003 followed for the 3D Accuitomo, (J. Morita Mfg. Corp., Kyoto, Japan) in March 6, the i-CAT (Imaging Sciences International, Hatfield, PA) in October 2, and the CB MercuRay (Hitachi, Medical Corp., Kashiwa-shi, Chiba-ken, Japan) on October 20. Since 2003, a number of other CBCT units have been FDA approved in the United States, including the Kodak 9000 3D, (Carestream/Trophy, Marne-la-Vallée, France), which is currently the highest resolution unit (Table 1). Several additional units are in various stages of development, testing, or application for FDA approval.
4.1. Types of CBCT Equipment
CBCT systems can be categorized according to the orientation of the patient during image acquisition, the scan volume irradiated, or the clinical functionality.
Depending on the system employed, maxillofacial CBCT can be performed with the patient in three possible positions: () sitting, () standing, and () supine. Equipment that requires the patient to be supine has a larger physical footprint and may not be readily accessible for patients with physical disabilities. Standing units may not be able to be adjusted to a height to accommodate wheelchair bound patients. Seated units are the most comfortable; however fixed seats may not allow ready scanning of physically disabled or wheelchair bound patients. As scan times are often similar to or greater than those used with panoramic imaging, perhaps more important than patient orientation is the head restraint mechanism used.
The dimensions of the FOV, or scan volume, are primarily dependent on the detector size and shape, beam projection geometry, and the ability to collimate the beam. The shape of the FOV can be either a cylinder or spherical (e.g., NewTom 3G). Collimation of the primary X-ray beam limits x-radiation exposure to the region of interest (ROI). Field size limitation therefore ensures that an optimal FOV can be selected for each patient based on disease presentation and the region designated to be imaged. Based on available or selected scan volume height, the use of units can be designed as follows: (1)localized region (also referred to as focused, small field or, limited field)—approximately 5 cm or less, (2)single arch—5 cm to 7 cm, (3)inter-arch—7 cm to 10 cm, (4)maxillofacial—10 cm to 15 cm, (5)craniofacial—greater than 15 cm. In general, the smaller the scan volume, the higher the spatial resolution of the image. As the earliest sign of periapical pathology is discontinuity in the lamina dura and widening of the periodontal ligament space, it is desirable that the optimal resolution of the any CBCT imaging system used in endodontics does not exceed 200 m—the average width of the periodontal ligament space. The 3D Accuitomo (J. Morita, Corporation, Kyoto, Japan)—the first of the small FOV systems—provided a resolution of 0.125 mm. At the time of publication, nominal voxel resolution varies from 0.4 mm to 0.076 mm.
Hybrid multimodal systems combine digital panoramic radiography with a relatively small-to medium-FOV CBCT system. This combination is now priced at a level similar to upper-level digital panoramic radiographic systems of the relatively recent past. Cost savings come from the fact that the cost of CBCT detectors is highly dependent on size. The ProMax 3D CBVT (Planmeca Oy, Helsinki, Finland) was the first to incorporate a small FOV 3D sensor to their ProMax digital panoramic line, which can be also be retrofitted to any of the prior ProMax digital models. Examples of other hybrid units are the Veraviewepocs 3D (J. Morita, Corporation, Kyoto, Japan), the Picasso Trio (Vatech/E. Woo Corporation, Korea), and the Kodak Dental Imaging 9000 DS (Kodak Dental Imaging/Practiceworks Atlanta, GA, USA) (Figure 1).
There are advantages beyond reduced capital costs to small FOV CBCT units for endodontic applications. First, a small FOV means that high-resolution images with a spatial resolution down to 0.076 mm isotropic voxel size can be achieved at very low exposure to ionizing radiation and without extensive reconstruction times that would be expected with larger FOV systems due to the greater file sizes to be processed. Second, a restricted FOV reduces the volume examined, and for which the practitioner is responsible to interpret. Small FOV systems concentrate on the dental arches or individual temporomandibular joints, the structures in which the average dentist is most familiar. There is less detail of the cranial cavity, paranasal sinuses, ear, and neck—structures less familiar to the average dentist. A small FOV CBCT system is undoubtedly too restrictive for maxillofacial surgeons who conduct craniofacial and orthognathic surgery or for complex implant/prosthetic situations where the jaws and both temporomandibular joints are best evaluated in toto rather than as individual components; however, third-party software is now available to “stitch” together adjacent small FOV images .
4.2. Radiation Dose Considerations
For a meaningful comparison of radiation risk, radiation exposures are converted to effective dose (E), measured in Sieverts (Sv). The Sv is a large unit; so in maxillofacial imaging milli-[10−3; mSv] or micro-[10−6; Sv] Sieverts are reported. The radiation dose to specific tissues is measured, adjusted for the amount of that tissue in the field of view, and weighted based on radiation sensitivity of the tissue. The weighted tissue/organ doses are then summed to assess Effective Dose (E). Comparisons can be performed with respect to natural background radiation.
The tissues/organs used to calculate the effective dose are specified by the International Commission on Radiological Protection (ICRP). The organs used to calculate effective dose for imaging of the head include the bone marrow, thyroid, esophagus, skin, bone surface, salivary glands, brain, and “remainder” tissues . Published effective doses for digital panoramic radiographs range from 5.5 to 22.0 Sv , while digital cephalometric radiographs have effective doses of 2.2 to 3.4 Sv . This compares with an average annual effective dose from background radiation in the United States of about 3,000 Sv (3.0 mSv).
There are a number of factors that will affect the radiation dose produced by a CBCT system: the imaging parameters used (kVp, mAs); pulsed beam versus continuous beam; amount, type, and shape of the beam filtration; the number of basis images dependent partly on use of or lesser rotations; and limitations on the size of the field of view. Factors such as beam quality and filtration are unique to a specific machine, while other factors, such as FOV, can sometimes be operator controlled. Typically, the smaller the field of view for a given system, the lower the radiation dose applied [19, 20]. Since the effective dose is computed from a weighted summation of doses to various organs, removing some organs from the path of the X-ray beam can reduce the effective dose. Since the radiation received by the thyroid gland contributes a large amount to the effective dose, limiting the beam to the maxilla instead of the whole head produces a lower effective dose.
Tables 2 and 3 provide the most recent published radiation exposures for selected CBCT units using ICRP (2007) recommendations [19–30] and compares them as multiples of digital panoramic examinations (using an average digital panoramic exposure of 14 Sv obtained from the published range of effective dose) and equivalent days of per capita background dose (based on an annual full body background exposure of 3 mSv). At the time of publication, the CBCT unit with the highest resolution and the smallest field of view (the KODAK 9000 3D) involves patient radiation exposure varying from as little as 0.4 to 2.7 digital panoramic equivalents depending on the part of the mouth studied .
4.3. Advantages of CBCT in Endodontics
Perhaps the most important advantage of CBCT in endodontics is that it demonstrates anatomic features in 3D that intraoral, panoramic, and cephalometric images cannot. CBCT units reconstruct the projection data to provide interrelational images in three orthogonal planes (axial, sagittal, and coronal). In addition because reconstruction of CBCT data is performed natively using a personal computer, data can be reoriented in their true spatial relationships.
Due to the isotropic nature of the constructed volume elements (“voxels”) constituting the volumetric dataset, image data can be sectioned nonorthogonally. Most software provides for various nonaxial 2D images in multiplanar reformation (MPR). Such MPR modes include oblique, curved planar reformation (providing “simulated” distortion free panoramic images) and serial transplanar reformation (providing cross-sections), which can be used to highlight specific anatomic regions for diverse diagnostic tasks (Figure 2). Enhancements including zoom magnification, window/level adjustments, and text or arrow annotation can be applied. Cursor-driven measurement algorithms provide the clinician with an interactive capability for real-time dimensional assessment. On-screen measurements are free from distortion and magnification.
Because acquisition occurs innately as high-resolution three-dimensional volumetric data and can be displayed as interactive images, CBCT technology provides the clinician with an unparalleled visualization of the often complex relationships and boundaries between teeth and their associated pathology and anatomic features within the alveolus and jaws such as the maxillary sinus and mandibular canal and foramen.
4.4. Limitations of CBCT in Endodontics
Despite the provision of the third dimension, the spatial resolution of CBCT images (0.4 mm to 0.076 mm or equivalent to 1.25 to 6.5 line pairs per mm−1[lp.mm−1]) is inferior to conventional film-based (approx. 20 lp.mm−1) or digital (ranging from 8–20 lp.mm−1) intraoral radiography . However, the ability of this technology to demonstrate geometrically accurate images in all three dimensions and the elimination of anatomic noise facilitates the assessment of a number of features important in endodontic diagnosis, treatment, and long-term management. The optimal resolution for CBCT images in endodontics is invariably task specific—however; most aspects of endodontics involve imaging of small structures. Liedke et al.  have recommended a minimal voxel resolution of 0.3 mm for the detection of external root resorption. Ex vivo research performed at our institution  has determined the effect of isotropic voxel dimensions on observer detection of the presence or absence of secondary canals in the mesiobuccal root of the maxillary first permanent molar. Observer interrater reliability and detection of mesiobuccal canals increased substantially with increasing resolution with more than 93% accuracy with a voxel resolution of 0.12 mm but accuracy barely over 60% with 0.4 mm resolution. The diagnosis of other subtle conditions (e.g., initial stages of apical periodontitis) involving the periodontal ligament space, which has an average dimension of 0.2 mm, also demands high resolution.
The CBCT projection geometry results in the whole volume within the FOV being irradiated with every basis image projection. Scattered radiation is produced omnidirectionally and is recorded by pixels on the cone beam CT detector but does not reflect actual attenuation of the object within a specific path of the X-ray beam. Additional recorded X-ray nonlinear attenuation is noise. This can be eliminated somewhat by algorithms such as wavelet transformation of filtered back-projection data; however, because of the use of an area detector, some of this nonlinear attenuation is recorded and contributes to image degradation when not adequately attended to by noise reduction algorithms. Remaining noise contributes to the graininess of the image which can be more pronounced in images in systems using a large FOV, especially where low signal due to restricted radiation exposure is the case.
Maxillofacial CBCT images presently lack the ability to record subtle changes in attenuation across a wide range of tissue radiodensities. In endodontics, contrast resolution might well be of importance in distinguishing the nature of periapical or sinus soft tissue contents. Three factors, inherent in the CBCT acquisition process, presently limit contrast resolution: () scattered radiation contributing to the potential for increased noise, () CBCT systems pronounced “heel effect” due to the divergence of the X-ray beam over the area detector producing nonuniformity of the incident X-ray beam, and () detector imperfections affecting linearity in response to x-radiation. These factors, and a desire to restrict dose, contribute to restricting the application of current maxillofacial CBCT imaging to the assessment of osseous structures. Work continues to develop systems capable of a wide contrast range supporting both hard tissue and soft tissue applications while still limiting dose.
CBCT images, like those from other diagnostic modalities, are susceptible to artifacts that affect image fidelity. Artifacts can be attributed to four sources : () the patient; () the scanner; () artifacts specific to the CBCT system used including partial volume averaging, undersampling, and the cone beam effect; and () X-ray beam artifacts arising from the inherent polychromatic nature of the projection X-ray beam that results in what is known as beam hardening (i.e., mean energy increases because lower energy photons are absorbed in preference to higher-energy photons). Beam hardening results in two types of artifact: () distortion of metallic structures due to differential absorption, known as a cupping artifact; and () streaks and dark bands that can appear between two dense objects. The presence of dental restorations, including apically positioned retrograde restorations, in the FOV can lead to severe streaking artifacts. As the CBCT X-ray beam is heterochromatic and has lower mean kVp energy compared to conventional CT, such artifact can be pronounced in CBCT images. In clinical endodontic practice, CBCT scanners with a limited field of view might provide clearer images as they can avoid scanning structures outside the region of interest susceptible to beam hardening (e.g., metallic restorations, dental implants).
5. CBCT Applications in Endodontics
A PUBMED search performed in May 2009 (search terms: cone beam, CBCT, endodontics, root canal, periapical) resulted in less than 30 comparative retrospective or ex vivo studies published quantifying specific clinical efficacies of CBCT imaging in endodontics. Similarly a recent review performed by the SEDENTEXCT project indicated that while several nonsystematic reviews in the literature provide a favorable perspective of the role of CBCT imaging in endodontics, only a few studies have been published that satisfy the criteria for formal systematic review .
While there are presently no definitive patient selection criteria for the use of CBCT in endodontics, the use of CBCT in endodontic diagnosis should not be avoided or ignored. One of the authors (Martin D. Levin) is a Board Certified Endodontist with a full time private practice with limited field CBCT. CBCT has been used to assist diagnosis and facilitate treatment in more than half of all patients referred to his practice for assessment and treatment of complex endodontic conditions (Figures 3 and 4).
Depending on the equipment used, CBCT exposure may subject a patient to only slightly higher radiation doses than conventional 2D imaging—or considerably more, so it is important that practitioners follow professional judgment in minimizing the radiation dose to the patient to that deemed essential for optimal diagnosis and treatment guidance. There should be justification of the exposure to the patient such that the total potential diagnostic benefits are greater than the uncertain detriment radiation exposure might cause. Published research, while admittedly sparse, indicates that CBCT has several applications in selected endodontic cases (Figures 5 and 6). The absence of high prospective randomized clinical trials underlines the need for further research on the treatment outcomes related to CBCT applications in endodontic practice. At this time CBCT should not be considered a replacement for standard digital radiographic applications. Rather, CBCT is a complementary modality for specific applications .
5.1. Preoperative Assessment
5.1.1. Tooth Morphology
The success of endodontic treatment depends on the identification of all root canals so that they can be accessed, cleaned, shaped, and obturated . The prevalence of a second mesiobuccal canal (MB2) in maxillary first molars has been reported to vary from 69% to 93% depending on the study method employed. This variability occurs in the buccolingual plane where superimposition of anatomic structures impedes detection of small structural density changes [37, 38]. Conventional radiographic techniques, at best, can only detect up to 55% of these configurations (Figure 7) . Ramamurthy et al.  found that raters evaluating different two-dimensional film modalities were rarely able to detect more than a 50% presence of MB2 canals. They found differences in detection rates with complementary metal oxide semiconductors (CMOSs), analog film, and photostimulable phosphor plates (PSP) detecting 55%, 44%, and 39% of MB2 canals, respectively. Matherne et al.  compared the ability of three board-certified endodontists to detect the number of root canals on intraoral digital (both charged-couple device and photostimulable phosphor) plate images with CBCT in 72 extracted teeth (3 equal groups of maxillary molars, mandibular premolars, and mandibular incisors). They found that on average the observers failed to detect at least one root canal in 40% of teeth using intraoral radiographs. CBCT evaluations identified an average of 3.58 root canals (RCS) per maxillary molar, 1.21 per mandibular premolar, and 1.5 per mandibular incisor. Evaluation of CCD images demonstrated an average number of 1.0 RCS per mandibular incisor, 1.0 per mandibular first premolar, and 3.1 per maxillary molar. Evaluation of PSP images demonstrated an average number of 1.3 RCS per mandibular incisor, 1.1 per mandibular first premolar, and 3.0 per maxillary molar. Baratto Filho et al.  investigated the internal morphology of extracted maxillary first molars by comparing detection rates obtained using an operating microscope and CBCT to ex vivo sections. They reported an ex vivo prevalence of a fourth canal in 67.14% of teeth and additional root canals in 92.85% of mesiobuccal roots. Clinical assessment provided slightly lower overall (53.26%) but higher (95.63%) MB2 detection rates whereas CBCT results showed the lowest overall (37.05%) detection rate. They indicated that CBCT provided a good method for the initial evaluation of maxillary first molar internal morphology but that the use of operating microscopes was optimal. Unpublished ex vivo research performed at our institution  investigated the effect of increasing voxel resolution on the detection rate of multiple observers of the MB2 on 24 maxillary first molars by CBCT. Compared to the overall prevalence of MB2 (92% prevalence), CBCT detection rates increased from 60% to 93.3% with increasing resolution suggesting that if CBCT is to be used, then resolutions in the order of 0.12 mm or less are optimal.
CBCT imaging has also been reported to characterize the high prevalence of the distolingual canal in Taiwanese individuals , highlight anomalies in the root canal system of mandibular premolars , and assist in the determination of root curvature .
5.1.2. Dental Periapical Pathosis
The most common pathologic conditions that involve teeth are the inflammatory lesions of the pulp and periapical areas (Figures 8, 9, 10, and 11). Lofthag-Hansen et al.  compared the accuracy of 3 observers using high-resolution limited FOV CBCT to intraoral radiographic paralleling technique using two images, one with a horizontal tube shift difference of about for the diagnosis of periapical pathology on 46 teeth. While CBCT and intraoral radiographs identified 53 roots with lesions, CBCT identified an additional 33 roots with lesions (62%). Observers agreed that additional clinically relevant material was provided by CBCT imaging in 32 of the 46 (69.5%) teeth imaged. Stavropoulos and Wenzel  compared CBCT (NewTom 3G) to digital- and film-based intraoral periapical radiography for the detection of periapical bone defects on 10 frozen pig mandibles by four calibrated examiners. They reported that CBCT provides greater diagnostic accuracy (61%) compared with digital (39%) and (44%) conventional radiographs. Özen et al.  performed a similar study comparing the detection of chemically induced periapical lesions by three observers using digital- and film-based conventional radiography to two CBCT systems (Iluma, Imtec Imaging, Ardmore, OK and iCAT, Imaging Sciences International, Hatfield, PA). They found that CBCT systems provided similar intra- and interobserver agreement substantially higher than either conventional radiography. They indicated that while detection rates for CBCT were higher, they did not advocate the replacement of intraoral radiography for detecting periapical lesions in routine clinical practice due to financial and dose considerations.
Estrela et al.  compared the accuracy of CBCT, panoramic and periapical radiographs from a consecutive sample of 888 imaging exams of patients with endodontic infection (1,508 teeth) in the detection of apical periodontitis (AP). While a gold standard was not available, they found the detected prevalence of AP to be significantly higher with CBCT (Figure 12). Estrela and colleagues proposed a periapical index based on cone beam-computed tomography (CBCTPAI) for identification of AP . The CBCT PAI is a 6-point (0–5) scoring system calculated from determining the largest lesional measurement in either the buccopalatal, mesio-distal, or diagonal dimension and taking into account expansion and destruction of cortical bone. Using their criteria, 3 observers applied it to 1,014 images (periapical radiographs and high resolution CBCT images) originally taken from 596 patients. They found that CBCT imaging detected 54.2% more AP lesions than intraoral radiography alone. Similar results are reported by Low et al.  who compared the preoperative consensus assessment of the apical condition of 37 premolars and 37 molars in the maxilla (156 total roots) using periapical radiography and CBCT referred for possible apical surgery and found the later method to demonstrate significantly more lesions (34%) than conventional radiography. CBCT showed significantly more findings including expansion of lesions into the maxillary sinus, sinus membrane thickening, and missed canals. Using an ex vivo model consisting of 2 mm diameter defects placed in the cancellous bone at the apices of 10 first molar teeth on six partially dentate intact human dry mandibles, Patel et al.  reported a detection rate of 24.8% and 100% for intraoral radiography and CBCT imaging respectively.
The generally higher detection rates afforded by CBCT are similar to those reported for conventional CT . This may be of clinical importance in patients who present with pain or who have poorly localized symptoms associated with an untreated or previously root treated tooth with no evidence of pathology identified by conventional imaging [54–56].
5.1.3. Root Fracture
While root fractures are less common than fractures of the crown and occur in only 7% or fewer of dental injuries [57, 58], they are difficult to diagnose accurately using conventional radiography. Numerous authors have illustrated the usefulness and importance of CBCT in the diagnosis and management in specific aspects of dento-alveolar trauma, especially root fractures (Figure 13) [59–62], luxation and/or displacement, and alveolar fracture . CBCT has found particular application for the diagnosis of root fractures. Hassan et al.  compared the accuracy of 4 observers in detecting ex vivo vertical root fractures (VRFs) on CBCT and periapical images and assessed the influence of root canal filling on fracture visibility. They found an overall higher accuracy for CBCT (0.86) scans than periapical radiographs (0.66) for detecting VRF which was slightly reduced by the presence of opaque obturation material. Similar results were reported by Kamburoğlu et al.  who compared the diagnostic accuracy of 3 oral and maxillofacial radiologists in detecting simulated horizontal root fractures on conventional radiographic (analog film, PSP and CCD-based digital) images and CBCT of 36 teeth. They found that the sensitivity of CBCT (0.92) was significantly greater than analog film (0.74), PSP (0.71), and CCD (0.68) images. Most recently Bernardes et al.  retrospectively compared conventional periapical radiographs and CBCT images for 20 patients with suspected root fractures. They found that CBCT was able to detect fractures in 18 (90%) of patients whereas conventional periapicals could only detect fractures 6 to 8 of the cases (30% to 40%) and indicated that CBCT was an excellent supplement to conventional radiography in the diagnosis of root fractures.
5.1.4. Root Resorption
The use of serial cross-sectional CT in diagnosing the size and location of external root resorption (ERR) has been well described (Figures 14 and 15) [66–68]. Similarly, several authors have presented selected cases illustrating the utility of CBCT in the detection of small lesions, localizing and differentiation the resorption from other conditions, classification of the lesion, in determining prognosis, and directing treatment (Figures 14 and 15) [54, 69–73]. The accuracy of CBCT in the detection of surface defects, while higher than conventional imaging modalities, is not perfect  and appears to increase with increasing voxel resolution of the volumetric dataset . CBCT has also been shown to have particular application in the assessment of the postorthodontic apical root resorption  and, in particular, of the roots of lateral maxillary incisors by impacted maxillary canines [75–77].
Internal root resorption (IRR) within the root canal itself is rare, usually asymptomatic, slowly progressing, and presents as a serendipitous finding on intraoral radiographic examination. The inflammatory etiology of the resorptive process is not fully understood, although IRR has been associated with a history of trauma, persistent chronic pulpitis, and as well as orthodontic treatment. It is very common that internal and external inflammatory root resorption are confused and misdiagnosed. Still, accurate assessment is essential as these conditions represent totally different pathological processes, with different etiological factors and treatment protocols. Diagnosis using conventional radiography is difficult; however, unlike external resorption, which presents with irregular radiolucency and intact root canal, internal resorption has clearly defined borders with no canal radiographically visible in the defect (Figure 16) . CBCT has been used successfully to confirm the presence of IRR and differentiate it from ERR .
5.2. Postoperative Assessment
Monitoring the healing of apical lesions is an important aspect of postoperative assessment in endodontics. Pinsky et al.  investigated the accuracy of CBCT (iCAT with 0.2 mm voxel resolution) in the detection of the simulated osseous defects of varying diameters and depths in an acrylic block and on the buccal cortex of a human mandible. They found mean accuracy for the acrylic block to be within the tolerance of the nominal resolution of the CBCT unit ( mm 0.02 (SE) mean width difference and mm 0.01 (SE) mean height difference). For the human mandible, they found differences to be slightly higher (mean width accuracy, mm (0.02 SE); mean height accuracy, mm (0.02 SE)). In addition they segmented the defect and applied and automated algorithm to calculate volume. They found that automated volume accuracy error was significantly higher ( mm3 (4 SE)) than manually derived measurements ( mm3 (2.6 SE)).
As adequacy of root canal obturation is an important determinant of endodontic success, it might be considered that CBCT is used in the initial and subsequent monitoring of the integrity of root canal fillings. Soğur et al.  compared the subjective quality of 3 radiologists and 3 endodontists using limited field CBCT, storage phosphor plate (SPP), and F-speed analog film images for the evaluation of length and homogeneity of root fillings on 17 extracted permanent mandibular incisor teeth. They found that SPP and F-speed film images were perceived as superior to the corresponding CBCT images and they reported that this may be due to the presence of streaking artifacts from the gutta percha and sealer compromising the quality of those images as regards root filling evaluations.
Endodontic surgery is often complicated in the posterior teeth by their proximity to anatomical structures. The mandibular teeth can be close to the mandibular canal while maxillary molars are often close to the maxillary sinus. CBCT imaging provides several advantages for preoperative treatment planning especially in maxillary posterior teeth with apical pathology . Rigolone et al.  first described the value of CBCT in planning for endodontic surgery. They imaged 43 maxillary first molars on 31 patients referred for retreatment and measured the mean distance of the palatine root from the external vestibular cortex (Mean; 9.73 mm) and the frequency that the maxillary sinus lateral recess lays between the roots (25%) to evaluate the ability to surgically approach the palatal root of a maxillary molar from a vestibular access as opposed to the more difficult palatal access. They concluded that CBCT may play an important role in optimizing palatine root apicoectomy via directing surgery through vestibular access. The importance of CBCT for apical surgery of teeth adjacent to the maxillary sinus has subsequently been illustrated by Nakata et al.  who presented a case report localizing the presence of a periradicular lesion to a specific root and Tsurumachi and Honda  who described the use of CBCT in localizing a fractured endodontic instrument protruding into the maxillary sinus prior to periapical surgery. Most recently Low et al.  compared the preoperative findings obtained from periapical radiography and CBCT of 2 observers in the diagnosis of posterior maxillary teeth (37 premolars and 37 molars—a total of 156 roots) referred for possible apical surgery. They found that CBCT demonstrated significantly more lesions (34%) than conventional radiography. They also reported that numerous additional clinically relevant findings were seen significantly more frequently in CBCT images including expansion of lesions into the maxillary sinus, sinus membrane thickening, and missed canals.
Conventional intraoral radiography provides clinicians with an accessible, cost effective, high-resolution imaging modality that continues to be of value in endodontic therapy. There are, however, specific situations, both pre- and postoperatively, where the understanding of spatial relationships afforded by CBCT facilitates diagnosis and influences treatment. The usefulness of CBCT imaging can no longer be disputed—CBCT is a useful task specific imaging modality and an important technology in comprehensive endodontic evaluation.
The authors gratefully acknowledge the assistance of Lakshmi S. Garladinne-Nethi, B.D.S., M.S., current graduate student in Oral and Maxillofacial Radiology, University of Connecticut, for her assistance in the preliminary stages of the manuscript. Unless otherwise indicated, CBCT data for all figures were acquired on a KODAK 9000 3D at 0.076 mm resolution and reformatted using KODAK Dental Imaging software v.220.127.116.11 (PracticeWorks, Atlanta, GA).
- O. E. Langland and R. P. Langlais, “Early pioneers of oral and maxillofacial radiology,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 80, no. 5, pp. 496–511, 1995.
- P. H. Jacobsohn and R. J. Fedran, “Making darkness visible: the discovery of X-ray and its introduction to dentistry,” The Journal of the American Dental Association, vol. 126, no. 10, pp. 1359–1367, 1995.
- A. G. Farman, C. M. Levato, and W. C. Scarfe, “3D X-ray: an update,” Inside Dentistry, vol. 3, no. 6, pp. 70–74, 2007.
- R. E. Walton, “Diagnostic imaging A. endodontic radiography,” in Ingles' Endodontics, J. I. Ingle, L. K. Bakland, and J. C. Baumgartner, Eds., p. 554, BC Decker, Hamilton, Canada, 6th edition, 2008.
- V. E. Rushton, K. Horner, and H. V. Worthington, “Screening panoramic radiology of adults in general dental practice: radiological findings,” British Dental Journal, vol. 190, no. 9, pp. 495–501, 2001.
- H.-G. Gröndahl and S. Huumonen, “Radiographic manifestations of periapical inflammatory lesions,” Endodontic Topics, vol. 8, pp. 55–67, 2004.
- M. Goldman, A. H. Pearson, and N. Darzenta, “Endodontic success—who's reading the radiograph?” Oral Surgery, Oral Medicine, Oral Pathology, vol. 33, no. 3, pp. 432–437, 1972.
- M. Goldman, A. H. Pearson, and N. Darzenta, “Reliability of radiographic interpretations,” Oral Surgery Oral Medicine and Oral Pathology, vol. 38, no. 2, pp. 287–293, 1974.
- A. G. Farman, “Image guidance: the present future of dental care,” Practical Procedures & Aesthetic Dentistry, vol. 18, no. 6, pp. 342–344, 2006.
- A. G. Farman, C. M. Levato, and W. C. Scarfe, “A primer on cone beam computed tomography,” Inside Dentistry, vol. 3, pp. 90–92, 2007.
- W. C. Scarfe, A. G. Farman, and P. Sukovic, “Clinical applications of cone-beam computed tomography in dental practice,” Journal of the Canadian Dental Association, vol. 72, no. 1, pp. 75–80, 2006.
- W. C. Scarfe and A. G. Farman, “Cone beam computed tomography: a paradigm shift for clinical dentistry,” Australasian Dental Practice, pp. 102–110, 2007.
- Y. Hayakawa, T. Sano, P. Sukovic, W. C. Scarfe, and A. G. Farman, “Cone beam computed tomography: a paradigm shift for clinical dentistry,” Nippon Dental Review, vol. 65, pp. 125–132, 2005.
- P. Kursağlu, P. Sukovic, W. C. Scarfe, and A. G. Farman, “Dental Volumetrik Bilgisayarli Tomograti,” Akademik Dental, vol. 26, pp. 30–36, 2005.
- A. G. Farman, C. Levato, W. C. Scarfe, and D. Chenin, “Cone beam CT trends towards hybrid systems and third party software utilization,” Inside Dentistry, vol. 4, no. 9, pp. 102–105, 2008.
- J. Valentin, The 2007 Recommendations of the International Commission on Radiological Protection, Publication 103, Annals of the ICRP, vol. 37, 2009.
- J. B. Ludlow, L. E. Davies-Ludlow, and S. L. Brooks, “Dosimetry of two extraoral direct digital imaging devices: NewTom cone beam CT and Orthophos Plus DS panoramic unit,” Dentomaxillofacial Radiology, vol. 32, no. 4, pp. 229–234, 2003.
- F. Gijbels, G. Sanderink, J. Wyatt, J. Van Dam, B. Nowak, and R. Jacobs, “Radiation doses of indirect and direct digital cephalometric radiography,” British Dental Journal, vol. 197, no. 3, pp. 149–152, 2004.
- J. A. Roberts, N. A. Drage, J. Davies, and D. W. Thomas, “Effective dose from cone beam CT examinations in dentistry,” British Journal of Radiology, vol. 82, no. 973, pp. 35–40, 2009.
- J. B. Ludlow, L. E. Davies-Ludlow, S. L. Brooks, and W. B. Howerton, “Dosimetry of 3 CBCT devices for oral and maxillofacial radiology: CB Mercuray, NewTom 3G and i-CAT,” Dentomaxillofacial Radiology, vol. 35, no. 4, pp. 219–226, 2006.
- E. Coppenrath, F. Draenert, U. Lechel, et al., “Cross-sectional imaging in dentomaxillofacial diagnostics: dose comparison of dental MSCT and NewTom 9000 DVT,” RöFo: Fortschritte auf dem Gebiet der Röntgenstrahlen und der Bildgebenden Verfahren, vol. 180, no. 5, pp. 396–401, 2008.
- E. Hirsch, U. Wolf, F. Heinicke, and M. A. G. Silva, “Dosimetry of the cone beam computed tomography Veraviewepocs 3D compared with the 3D Accuitomo in different fields of view,” Dentomaxillofacial Radiology, vol. 37, no. 5, pp. 268–273, 2008.
- S. Lofthag-Hansen, A. Thilander-Klang, A. Ekestubbe, E. Helmrot, and K. Grōndahl, “Calculating effective dose on a cone beam computed tomography device: 3D Accuitomo and 3D Accuitomo FPD,” Dentomaxillofacial Radiology, vol. 37, no. 2, pp. 72–79, 2008.
- M. Loubele, R. Bogaerts, E. Van Dijck, et al., “Comparison between effective radiation dose of CBCT and MSCT scanners for dentomaxillofacial applications,” European Journal of Radiology, vol. 71, no. 3, pp. 461–468, 2009.
- J. B. Ludlow and M. Ivanovic, “Comparative dosimetry of dental CBCT devices and 64-slice CT for oral and maxillofacial radiology,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 106, no. 1, pp. 930–938, 2008.
- T. Okano, Y. Harata, Y. Sugihara, et al., “Absorbed and effective doses from cone beam volumetric imaging for implant planning,” Dentomaxillofacial Radiology, vol. 38, no. 2, pp. 79–85, 2009.
- J. M. Palomo, P. S. Rao, and M. G. Hans, “Influence of CBCT exposure conditions on radiation dose,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 105, no. 6, pp. 773–782, 2008.
- M. A. G. Silva, U. Wolf, F. Heinicke, A. Bumann, H. Visser, and E. Hirsch, “Cone-beam computed tomography for routine orthodontic treatment planning: a radiation dose evaluation,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 133, no. 5, pp. 640.e1–640.e5, 2008.
- M. A. Garcia Silva, U. Wolf, F. Heinicke, K. Gründler, H. Visser, and E. Hirsch, “Effective dosages for recording Veraviewepocs dental panoramic images: analog film, digital, and panoramic scout for CBCT,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 106, no. 4, pp. 571–577, 2008.
- J. B. Ludlow, “Dosimetry of the Kodak 9000 3D Small FOV CBCT and Panoramic Unit,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 107, no. 4, p. e29, 2008.
- A. G. Farman and T. T. Farman, “A comparison of 18 different X-ray detectors currently used in dentistry,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 99, no. 4, pp. 485–489, 2005.
- G. S. Liedke, H. E. D. da Silveira, H. L. D. da Silveira, V. Dutra, and J. A. P. de Figueiredo, “Influence of voxel size in the diagnostic ability of cone beam tomography to evaluate simulated external root resorption,” Journal of Endodontics, vol. 35, no. 2, pp. 233–235, 2009.
- M. Bauman, The effect of cone beam computed tomography voxel resolution on the detection of canals in the mesiobuccal roots of permanent maxillary first molars, M.S. thesis, University of Louisville School of Dentistry, Louisville, Ky, USA, 2009.
- W. C. Scarfe and A. G. Farman, “What is cone-beam CT and how does it work?” Dental Clinics of North America, vol. 52, no. 4, pp. 707–730, 2008.
- SEDENTEXCT, “Radiation protection: cone beam CT for dental and maxillofacial radiology,” Provisional guidelines, v1.1, May 2009.
- F. J. Vertucci, “Root canal anatomy of the human permanent teeth,” Oral Surgery Oral Medicine and Oral Pathology, vol. 58, no. 5, pp. 589–599, 1984.
- F. Pineda, “Roentgenographic investigation of the mesiobuccal root of the maxillary first molar,” Oral Surgery Oral Medicine and Oral Pathology, vol. 36, no. 2, pp. 253–260, 1973.
- R. Nance, D. Tyndall, L. G. Levin, and M. Trope, “Identification of root canals in molars by tuned-aperture computed tomography,” International Endodontic Journal, vol. 33, no. 4, pp. 392–396, 2000.
- R. Ramamurthy, J. P. Scheetz, S. J. Clark, and A. G. Farman, “Effects of imaging system and exposure on accurate detection of the second mesio-buccal canal in maxillary molar teeth,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 102, no. 6, pp. 796–802, 2006.
- R. Ramamurthy, J. P. Scheetz, S. J. Clark, and A. G. Farman, “Effects of imaging system and exposure on accurate detection of the second mesio-buccal canal in maxillary molar teeth,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 102, no. 6, pp. 796–802, 2006.
- R. P. Matherne, C. Angelopoulos, J. C. Kulild, and D. Tira, “Use of cone-beam computed tomography to identify root canal systems in vitro,” Journal of Endodontics, vol. 34, no. 1, pp. 87–89, 2008.
- F. Baratto Filho, S. Zaitter, G. A. Haragushiku, E. A. de Campos, A. Abuabara, and G. M. Correr, “Analysis of the internal anatomy of maxillary first molars by using different methods,” Journal of Endodontics, vol. 35, no. 3, pp. 337–342, 2009.
- M.-G. Tu, H.-L. Huang, S.-S. Hsue, et al., “Detection of permanent three-rooted mandibular first molars by cone-beam computed tomography imaging in taiwanese individuals,” Journal of Endodontics, vol. 35, no. 4, pp. 503–507, 2009.
- B. M. Cleghorn, W. H. Christie, and C. C. S. Dong, “Anomalous mandibular premolars: a mandibular first premolar with three roots and a mandibular second premolar with a C-shaped canal system,” International Endodontic Journal, vol. 41, no. 11, pp. 1005–1014, 2008.
- C. Estrela, M. R. Bueno, M. D. Sousa-Neto, and J. D. Pécora, “Method for determination of root curvature radius using cone-beam computed tomography images,” Brazilian Dental Journal, vol. 19, no. 2, pp. 114–118, 2008.
- S. Lofthag-Hansen, S. Huumonen, K. Gröndahl, and H.-G. Gröndahl, “Limited cone-beam CT and intraoral radiography for the diagnosis of periapical pathology,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 103, no. 1, pp. 114–119, 2007.
- A. Stavropoulos and A. Wenzel, “Accuracy of cone beam dental CT, intraoral digital and conventional film radiography for the detection of periapical lesions. An ex vivo study in pig jaws,” Clinical Oral Investigations, vol. 11, no. 1, pp. 101–106, 2007.
- T. Özen, K. Kamburoğlu, A. R. I. Cebeci, S. P. Yüksel, and C. S. Paksoy, “Interpretation of chemically created periapical lesions using 2 different dental cone-beam computerized tomography units, an intraoral digital sensor, and conventional film,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 107, no. 3, pp. 426–432, 2009.
- C. Estrela, M. R. Bueno, C. R. Leles, B. Azevedo, and J. R. Azevedo, “Accuracy of cone beam computed tomography and panoramic and periapical radiography for detection of apical periodontitis,” Journal of Endodontics, vol. 34, no. 3, pp. 273–279, 2008.
- C. Estrela, M. R. Bueno, B. C. Azevedo, J. R. Azevedo, and J. D. Pécora, “A new periapical index based on cone beam computed tomography,” Journal of Endodontics, vol. 34, no. 11, pp. 1325–1331, 2008.
- K. M. T. Low, K. Dula, W. Bürgin, and T. von Arx, “Comparison of periapical radiography and limited cone-beam tomography in posterior maxillary teeth referred for apical surgery,” Journal of Endodontics, vol. 34, no. 5, pp. 557–562, 2008.
- S. Patel, A. Dawood, F. Mannocci, R. Wilson, and T. Pitt Ford, “Detection of periapical bone defects in human jaws using cone beam computed tomography and intraoral radiography,” International Endodontic Journal, vol. 42, no. 6, pp. 507–515, 2009.
- E. G. Jorge, M. Tanomaru-Filho, M. Gonçalves, and J. M. G. Tanomaru, “Detection of periapical lesion development by conventional radiography or computed tomography,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 106, no. 1, pp. e56–e61, 2008.
- T. P. Cotton, T. M. Geisler, D. T. Holden, S. A. Schwartz, and W. G. Schindler, “Endodontic applications of cone-beam volumetric tomography,” Journal of Endodontics, vol. 33, no. 9, pp. 1121–1132, 2007.
- S. Patel, A. Dawood, T. Pitt Ford, and E. Whaites, “The potential applications of cone beam computed tomography in the management of endodontic problems,” International Endodontic Journal, vol. 40, no. 10, pp. 818–830, 2007.
- K. Nakata, M. Naitoh, M. Izumi, K. Inamoto, E. Ariji, and H. Nakamura, “Effectiveness of dental computed tomography in diagnostic imaging of periradicular lesion of each root of a multirooted tooth: a case report,” Journal of Endodontics, vol. 32, no. 6, pp. 583–587, 2006.
- J. O. Andreasen and F. M. Andreasen, “Classification, etiology and epidemiology,” in Textbook and Color Atlas of Traumatic Injuries to the Teeth, J. O. Andreasen and F. M. Andreasen, Eds., pp. 151–216, Munksgaard, Copenhagen, Denmark, 3rd edition, 1994.
- M. Cvek, G. Tsilingaridis, and J. O. Andreasen, “Survival of 534 incisors after intra-alveolar root fracture in patients aged 7–17 years,” Dental Traumatology, vol. 24, no. 4, pp. 379–387, 2008.
- D. A. Tyndall and S. Rathore, “Cone-beam CT diagnostic applications: caries, periodontal bone assessment, and endodontic applications,” Dental Clinics of North America, vol. 52, no. 4, pp. 825–841, 2008.
- N. Cohenca, J. H. Simon, R. Roges, Y. Morag, and J. M. Malfaz, “Clinical indications for digital imaging in dento-alveolar trauma—part 1: traumatic injuries,” Dental Traumatology, vol. 23, no. 2, pp. 95–104, 2007.
- D. Ilgüy, M. Ilgüy, E. Fisekcioglu, and G. Bayirli, “Detection of jaw and root fractures using cone beam computed tomography: a case report,” Dentomaxillofacial Radiology, vol. 38, no. 3, pp. 169–173, 2009.
- S. Patel, “New dimensions in endodontic imaging—part 2: cone beam computed tomography,” International Endodontic Journal, vol. 42, no. 6, pp. 463–475, 2009.
- B. Hassan, M. E. Metska, A. R. Ozok, P. van der Stelt, and P. R. Wesselink, “Detection of vertical root fractures in endodontically treated teeth by a cone beam computed tomography scan,” Journal of Endodontics, vol. 35, no. 5, pp. 719–722, 2009.
- K. Kamburoğlu, A. R. İlker Cebeci, and H. G. Gröndahl, “Effectiveness of limited cone-beam computed tomography in the detection of horizontal root fracture,” Dental Traumatology, vol. 25, no. 3, pp. 256–261, 2009.
- R. A. Bernardes, I. G. de Moraes, M. A. Húngaro Duarte, B. C. Azevedo, J. R. de Azevedo, and C. M. Bramante, “Use of cone-beam volumetric tomography in the diagnosis of root fractures,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 108, no. 2, pp. 270–277, 2009.
- E. Kim, K.-D. Kim, B.-D. Roh, Y.-S. Cho, and S.-J. Lee, “Computed tomography as a diagnostic aid for extracanal invasive resorption,” Journal of Endodontics, vol. 29, no. 7, pp. 463–465, 2003.
- A. Gulsahi, K. Gulsahi, and M. Ungor, “Invasive cervical resorption: clinical and radiological diagnosis and treatment of 3 cases,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 103, no. 3, pp. e65–e72, 2007.
- H. L. D. da Silveira, H. E. D. Silveira, G. S. Liedke, C. A. Lermen, R. B. Dos Santos, and J. A. P. de Figueiredo, “Diagnostic ability of computed tomography to evaluate external root resorption in vitro,” Dentomaxillofacial Radiology, vol. 36, no. 7, pp. 393–396, 2007.
- S. Patel and A. Dawood, “The use of cone beam computed tomography in the management of external cervical resorption lesions,” International Endodontic Journal, vol. 40, no. 9, pp. 730–737, 2007.
- A. Maini, P. Durning, and N. Drage, “Resorption: within or without? The benefit of cone-beam computed tomography when diagnosing a case of an internal/external resorption defect,” British Dental Journal, vol. 204, no. 3, pp. 135–137, 2008.
- N. Cohenca, J. H. Simon, A. Mathur, and J. M. Malfaz, “Clinical indications for digital imaging in dento-alveolar trauma—part 2: root resorption,” Dental Traumatology, vol. 23, no. 2, pp. 105–113, 2007.
- S. Patel, S. Kanagasingam, and T. Pitt Ford, “External cervical resorption: a review,” Journal of Endodontics, vol. 35, no. 5, pp. 616–625, 2009.
- W. Hahn, S. Fricke-Zech, J. Fricke, et al., “Detection and size differentiation of simulated tooth root defects using flat-panel volume computerized tomography (fpVCT),” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 107, no. 2, pp. 272–278, 2009.
- A. Dudic, C. Giannopoulou, M. Leuzinger, and S. Kiliaridis, “Detection of apical root resorption after orthodontic treatment by using panoramic radiography and cone-beam computed tomography of super-high resolution,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 135, no. 4, pp. 434–437, 2009.
- L. Walker, R. Enciso, and J. Mah, “Three-dimensional localization of maxillary canines with cone-beam computed tomography,” American Journal of Orthodontics and Dentofacial Orthopedics, vol. 128, no. 4, pp. 418–423, 2005.
- D. Liu, W.-L. Zhang, Z. Zhang, Y. Wu, and X. Ma, “Localization of impacted maxillary canines and observation of adjacent incisor resorption with cone-beam computed tomography,” Oral Surgery, Oral Medicine, Oral Pathology, Oral Radiology, and Endodontology, vol. 105, no. 1, pp. 91–98, 2008.
- S. L. Hechler, “Cone-beam CT: applications in orthodontics,” Dental Clinics of North America, vol. 52, no. 4, pp. 809–823, 2008.
- K. Gulabivala and L. J. Searson, “Clinical diagnosis of internal resorption: an exception to the rule,” International Endodontic Journal, vol. 28, no. 5, pp. 255–260, 1995.
- H. M. Pinsky, S. Dyda, R. W. Pinsky, K. A. Misch, and D. Sarment, “Accuracy of three-dimensional measurements using cone-beam CT,” Dentomaxillofacial Radiology, vol. 35, no. 6, pp. 410–416, 2006.
- E. Soğur, B. G. Baksi, and H.-G. Gröndahl, “Imaging of root canal fillings: a comparison of subjective image quality between limited cone-beam CT, storage phosphor and film radiography,” International Endodontic Journal, vol. 40, no. 3, pp. 179–185, 2007.
- G. R. Young, “Contemporary management of lateral root perforation diagnosed with the aid of dental computed tomography,” Australian Endodontic Journal, vol. 33, no. 3, pp. 112–118, 2007.
- S. Patel, A. Dawood, T. Pitt Ford, and E. Whaites, “The potential applications of cone beam computed tomography in the management of endodontic problems,” International Endodontic Journal, vol. 40, no. 10, pp. 818–830, 2007.
- M. Rigolone, D. Pasqualini, L. Bianchi, E. Berutti, and S. D. Bianchi, “Vestibular surgical access to the palatine root of the superior first molar: “low-dose cone-beam” CT analysis of the pathway and its anatomic variations,” Journal of Endodontics, vol. 29, no. 11, pp. 773–775, 2003.
- T. Tsurumachi and K. Honda, “A new cone beam computerized tomography system for use in endodontic surgery,” International Endodontic Journal, vol. 40, no. 3, pp. 224–232, 2007.