Research Article | Open Access
Vanessa N. Alves, Simone S. O. Borges, Waldomiro B. Neto, Nívia M. M. Coelho, "Determination of Low Levels of Lead in Beer Using Solid-Phase Extraction and Detection by Flame Atomic Absorption Spectrometry", Journal of Analytical Methods in Chemistry, vol. 2011, Article ID 464102, 6 pages, 2011. https://doi.org/10.1155/2011/464102
Determination of Low Levels of Lead in Beer Using Solid-Phase Extraction and Detection by Flame Atomic Absorption Spectrometry
In this study, a method for the determination of low concentrations of lead in beer samples using solid-phase extraction with a flow injection analysis system and detection by flame atomic absorption spectrometry (FAAS) was developed. Moringa oleifera seeds were used as a biosorbent material. Chemical and flow variables of the online preconcentration system, such as sample pH, preconcentration flow rate, eluent flow rate, eluent concentration, particle size, and sorbent mass, were studied. The optimum extraction conditions were obtained using a sample pH of 6.0, sample flow rate of 6.0 mL min−1, 63.0 mg of sorbent mass, and 2.0 mol L−1 HNO3 at a flow rate of 2.0 mL min−1 as the eluent. With the optimized conditions, the preconcentration factor, precision, detection limit, consumption index, and sample throughput were estimated as 93, 0.3% (10.0 g L−1, ), 7.5 g L−1, 0.11 mL, and 23 samples per hour, respectively. The method developed was successfully applied to beer samples and recovery tests, with recovery ranging from 80% to 100%.
The concentration of metals in many alcoholic beverages is a significant parameter which can affect their consumption and conservation .
Metals find their way into alcoholic beverages at different stages and through various sources including the raw materials, type of brewing process and equipment, bottling, aging/storage, and adulteration .
Thus, many metals are monitored and regulated, which has resulted in the development of analytical techniques for their analysis. The determination of trace metals in beer is relevant, because they might be essential or toxic to the human body, and they can also influence the brewing process. The analysis is generally carried out by atomic absorption or emission techniques ; however, the long sample preparation times required often preclude their widespread use.
In the case of beer analysis, one of the official methods indicates the need to destroy the organic matter before analysis by flame atomic absorption spectrometry (FAAS) . This is often performed by acid digestion with H2SO4 and H2O2 on a hot plate in an open beaker, which is a time-consuming procedure prone to losses and contamination .
The digestion of the sample can be avoided for the direct analysis of beer samples. However, direct aspiration of beer causes flame fluctuations and the formation of solid deposits on the burner head . This problem can be minimized by using higher dilution of the beer samples; however, the typical sensitivities reached by FAAS do not allow the adoption of this strategy when determining minor and trace elements .
One alternative to circumvent the low sensitivity of FAAS involves a preconcentration step of the metals from the beverage matrix using either ionic resins or adsorbents before the analysis. Column techniques have been extensively utilized for metal preconcentration . These methodologies based on solid phase extraction are attractive when coupled online with the detection instrument and with the use of a high sorption capacity sorbent, such as Moringa oleifera seeds.
Moringa oleifera is the best known species of the Moringaceae family. It is a plant native to northwest India but has spread all over the world, mainly in tropical countries. Its seeds have been used for the treatment of turbid water due to their flocculation properties, attributed to a flocculating protein isolated by Gassenschmidt et al.  with a molecular mass of around 6.5 kDa and an isoelectric point above pH 10.
Araújo et al., (2010)  used M. oleifera seeds to develop a flow preconcentration system for Ag determination in aqueous solution. The optimum preconcentration conditions were obtained using a sample pH in the range of 6.0–8.0, preconcentration time of 4 min at a flow rate of 3.5 mL min−1, 0.5 mol L−1 HNO3 as the eluent at a flow rate of 4.5 mL min−1 and 35 mg of sorbent mass. With the optimized conditions, the preconcentration factor, precision, detection limit, and sample throughput were estimated as 35 (for preconcentration of 14 mL sample), 3.8% (5.0 g L−1, ), 0.22 g L−1 and 12 samples per hour, respectively. The developed method was successfully applied to mineral and tap water.
Thus, the objective of this study was to develop a methodology for an online preconcentration system, using M. oleifera seeds as a biosorbent, coupled to flame atomic absorption spectrometry, for the determination of low levels of lead in beer samples.
A Varian flame atomic absorption spectrometer, model SpectrAA 220 (Victoria, Australia), equipped with a lead hollow cathode lamp and a deuterium lamp for background correction was used for the detection of lead. The instrument was operated under the conditions recommended by the manufacturer: lamp current of 5 mA, wavelength of 217.0 nm, slit width of 0.1 nm, burner height of 17 mm, acetylene flow rate of 2.0 L min−1, and air flow rate of 13.5 L min−1.
The flow preconcentration system was constructed using a Gilson Minipuls 3 peristaltic pump (Villiers Le Bel, France) equipped with eight channels and Tygon and polyethylene tubes were used to pump the solutions through the minicolumn (60 mm × 3 mm) in the elution and preconcentration steps. A Gehaka PG1800 pH meter was used to adjust the pH of the samples and the working solutions.
2.2. Reagents, Solutions, and Samples
All working solutions were prepared with deionized water obtained from a Gehaka (São Paulo, Brazil) water purification system.
All reagents were analytical grade. Before use, laboratory glassware was kept overnight in 10% (v/v) nitric acid aqueous solution, followed by ultrasonication for 1 h and finally rinsing with deionized water. Working solutions of lead were prepared daily by appropriate dilution of a 1000 mg L−1 standard lead solution (Carlo Erba, Val de Reuil, France). The nitric acid solution used as the eluent was prepared through dilution in water of concentrated nitric acid obtained from Merck (Darmstadt, Germany).
Beer samples were purchased in local shops in Uberlândia (Minas Gerais State, Brazil) and analyzed without prior treatment.
2.3. Preparation of the Column
The Moringa seeds used to construct the minicolumn were obtained from trees cultivated in the city of Uberlândia (Minas Gerais, Brazil) and collected during January–July 2009. The seeds were separated from the pods, washed with deionized water and dried at 25°C. After drying, the seeds were crushed in a blender (Black & Decker, São Paulo, Brazil) and passed through 850 μm sieves.
The minicolumns were comprised of polyethylene tubes with an inner diameter of 3 mm and were sealed at both ends with glass wool. The minicolumn (60 mm × 3 mm) was filled with 63 mg of the seeds, and the performance was stable during all experiments.
2.4. Online Preconcentration System
The online flow system used for the development of the proposed method is shown in Figure 1. The flow system consists of a peristaltic pump equipped with Tygon tubes, four three-way solenoid valves and a minicolumn filled with biosorbent. The system was coupled to the FAAS instrument. During the preconcentration step (Figure 1(a)), valve 1 is open, and the other valves remain closed; the samples or working solutions are pumped through the minicolumn and the effluent is discharged. In the elution step (Figure 1(b)), valve 1 is closed, and valves 2, 3, and 4 are open. Thus, the eluent percolates through the minicolumn in the opposite direction to that of the sample undergoing the preconcentration step. The eluate is carried directly to the nebulization system of the FAAS instrument.
2.5. Optimization System
The optimization of the parameters affecting the sorption of Pb by the M. oleifera seeds was performed using a two-level full factorial experimental design involving six factors and a final optimization using a central composite design (CCD). All experiments were carried out in duplicate using 10.0 mL of a beer sample spiked with 10 g L−1 Pb solution. The variables studied were sample pH, adsorbent mass, eluent concentration, sample flow rate, eluent flow rate, and particle size of adsorbent.
3. Results and Discussion
3.1. Optimization Strategies
Preliminary tests were performed to investigate which factors exert significant influence on the adsorption of Pb (II) by the M. oleifera seeds. The factors selected were pH, adsorbent mass, eluent concentration, sample flow rate, eluent flow rate and particle size of adsorbent. The eluent used was HNO3.
The analytical response was taken as the integrated absorbance, and the beer sample volume used for the preconcentration was 10 mL spiked at 10 g L−1 Pb. Table 1 shows the response for each factorial design experiment.
From the results, a Pareto chart (Figure 2) was plotted to check the influence of the factors and their interactions in the system. An effect was considered significant when it was above the standard error at the 95% confidence level (P> 0.05), which is denoted by the vertical line on the graph.
As can be observed, the majority of variables appear to be statistically significant (at the 95% confidence level) within the range studied. In this study, it could be observed that eluent concentration, adsorbent mass, and pH were significant for the high level, while the elution flow rate and particle size were significant for the low level. The particle size was the variable that most influenced the system, however, due to limitations of the system and observed following the referrals, it was kept at 850 μm.
The interaction between particle size and eluent concentration and the interaction between adsorbent mass and particle size also presented a significant effect on the interactions. Nevertheless, these interactions were not taken into consideration, since they had less effect than those observed for the main variables.
The effect of the preconcentration flow rate was not significant, and this variable was kept at 6.0 mL min−1.
To refine the eluent concentration, adsorbent mass, eluent flow rate, and pH, the optimization was carried out using the response surface methodology in order to obtain the critical values through application of the CCD. Table 2 shows the design matrix used to construct the response surface.
The response data were used to generate response surfaces for the system studied (Figures 3(a) and 3(b)). Maximum points were obtained for all surface responses and the critical concentration values for the factors investigated adopted in further experiments were pH 6.0, sorbent mass 63.0 mg, and 2.0 mol L−1 HNO3 at a flow rate of 2.0 mL min−1 as the eluent.
3.2. Analytical Features
With the optimized system, the method was evaluated through the main analytical features. The preconcentration factor was calculated as the ratio between the slopes of the calibration curves obtained with and without the preconcentration step  and was 93.0. The detection limit was calculated as three times the standard deviation of 15 independent measurements of a blank sample divided by the slope of the calibration curve. The detection limit was calculated as 7.5 g L−1. A satisfactory correlation coefficient was obtained (0.9949) for the analytical signal observed experimentally in the range of 5 to 50 g L−1, where the regression equation for lead determination is Abs = 1.085 [Pb2+] + 0.018. The repeatability of the proposed method was assessed by performing seven consecutive preconcentration steps at a concentration level of 10 g L−1 lead and expressing the result in terms of relative standard deviation. The value of 0.27% was obtained, demonstrating excellent repeatability.
In the present study, the method proposed for determining lead in beer samples was described and compared according to the detection technique used. The most important details of the published procedures for lead preconcentration, in terms of kind of sample, are presented in Table 3. To compare the method studied in this paper, preconcentration factor and sample volume have been taken into account, and it does not require the use of complexing agents.
|PF—preconcentration factor; LOD—limit of detection; CPE—cloud point extraction; SPE—solid phase extraction; SV—sample volume; APDC—ammonium pyrrolidine dithiocarbamate.|
3.3. Application of the Method and Recovery Tests
The proposed method was applied to the beer samples. In all samples, the analyte concentration was below the detection limit of the method. Thus, in order to assess the analyte recovery, all samples were spiked at concentration levels from 0 to 50 g L−1, and analytical curves were constructed in order to compare the slopes. The results obtained are shown in Table 4, where it can be seen that there is no difference between the recovery values for the samples, indicating that the analyte was quantitatively retained and eluted for all of the samples evaluated.
The procedure developed using SPE with detection by FAAS allowed the direct determination of Pb at the level of g L−1 in beer samples. Due to the preconcentration step, it was possible obtain a low limit of detection. The optimization of the preconcentration system using a factorial design and the response surface methodology enabled the optimization of the procedure to be carried out with a reduced number of experiments, thus requiring a lower quantity of reagents. The proposed procedure may be applied as a quality control tool to determine other elements present in low concentrations in beer samples.
The authors are grateful for financial support from Conselho Nacional de Desenvolvimento Científico e Tecnológico (CNPq), Fundação de Amparo à Pesquisa do Estado de Minas Gerais (FAPEMIG), and Coordenação de Aperfeiçoamento de Pessoal de Nível Superior (CAPES).
- J. G. Ibanez, A. Carreon-Alvarez, M. Barcena-Soto, and N. Casillas, “Metals in alcoholic beverages: a review of sources, effects, concentrations, removal, speciation, and analysis,” Journal of Food Composition and Analysis, vol. 21, no. 8, pp. 672–683, 2008.
- P. Pohl, “What do metals tell us about wine?” Trends in Analytical Chemistry, vol. 26, no. 9, pp. 941–949, 2007.
- K. Pyrzynska, “Chemical speciation and fractionation of metals in wine,” Chemical Speciation and Bioavailability, vol. 19, no. 1, pp. 1–8, 2007.
- C. A. Watson, Official and Standardized Methods of Analysis, Royal Society of Chemistry, Cambridge, UK, 3rd edition, 1994.
- D. Bellido-Milla, A. Oñate-Jaén, J. M. Palacios-Santander, D. Palacios-Tejero, and M. P. Hernández-Artiga, “Beer digestions for metal determination by atomic spectrometry and residual organic matter,” Microchimica Acta, vol. 144, no. 1–3, pp. 183–190, 2004.
- D. Bellido-Milla, J. M. Moreno-Perez, and M. P. Hernández-Artiga, “Differentiation and classification of beers with flame atomic spectrometry and molecular absorption spectrometry and sample preparation assisted by microwaves,” Spectrochimica acta, Part B, vol. 55, no. 7, pp. 855–864, 2000.
- C. C. Nascentes, M. Y. Kamogawa, K. G. Fernandes, M. A. Z. Arruda, A. R. A. Nogueira, and J. A. Nóbrega, “Direct determination of Cu, Mn, Pb, and Zn in beer by thermospray flame furnace atomic absorption spectrometry,” Spectrochimica Acta—Part B, vol. 60, no. 5, pp. 749–753, 2005.
- M. Tuzen, O. D. Uluozlu, C. Usta, and M. Soylak, “Biosorption of copper(II), lead(II), iron(III) and cobalt(II) on Bacillus sphaericus-loaded Diaion SP-850 resin,” Analytica Chimica Acta, vol. 581, no. 2, pp. 241–246, 2007.
- U. Gassenschmidt, K. D. Jany, B. Tauscher, and H. Niebergall, “Isolation and characterization of a flocculating protein from Moringa oleifera Lam,” Biochimica et Biophysica Acta, vol. 1243, no. 3, pp. 477–481, 1995.
- C. S. T. Araújo, V. N. Alves, H. C. Rezende, and N. M. M. Coelho, “Development of a flow system for the determination of low concentrations of silver using Moringa oleifera seeds as biosorbent and flame atomic absorption spectrometry,” Microchemical Journal, vol. 96, no. 1, pp. 82–85, 2010.
- J. N. Bianchin, R. Mior, E. Martendal, J. S. Carletto, and E. Carasek, “Multivariate optimization and application of silica gel chemically modified with Niobium (V) oxide for the determination of Ni (II) in aqueous matrices in an on-line system,” Ecletica Quimica, vol. 33, no. 4, pp. 25–32, 2008.
- D. Citak and M. Tuzen, “A novel preconcentration procedure using cloud point extraction for determination of lead, cobalt and copper in water and food samples using flame atomic absorption spectrometry,” Food and Chemical Toxicology, vol. 48, no. 5, pp. 1399–1404, 2010.
- E. Melek, M. Tuzen, and M. Soylak, “Flame atomic absorption spectrometric determination of cadmium(II) and lead(II) after their solid phase extraction as dibenzyldithiocarbamate chelates on Dowex Optipore V-493,” Analytica Chimica Acta, vol. 578, no. 2, pp. 213–219, 2006.
- M. Tuzen and M. Soylak, “Multi-element coprecipitation for separation and enrichment of heavy metal ions for their flame atomic absorption spectrometric determinations,” Journal of Hazardous Materials, vol. 162, no. 2-3, pp. 724–729, 2009.
- M. Tuzen, K. O. Saygi, and M. Soylak, “Solid phase extraction of heavy metal ions in environmental samples on multiwalled carbon nanotubes,” Journal of Hazardous Materials, vol. 152, no. 2, pp. 632–639, 2008.
Copyright © 2011 Vanessa N. Alves et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.