Research Article  Open Access
Zeming Jian, Yifan Huang, Lei Nie, Mengran Liu, Guojun Zhang, "Research on “CylinderFourBeam” Microstructure Improvement of MEMS Bionic Vector Hydrophone", Journal of Sensors, vol. 2021, Article ID 5580390, 11 pages, 2021. https://doi.org/10.1155/2021/5580390
Research on “CylinderFourBeam” Microstructure Improvement of MEMS Bionic Vector Hydrophone
Abstract
The existing MEMS bionic vector hydrophone has the problems of lowsensitivity and narrowworking band, and the sensitivity and working bandwidth cannot be improved simultaneously by changing the single microstructural parameter. In this paper, the MEMS bionic vector hydrophone microstructural parameters (length, width and height of cantilever, side length of the center block, height and radius of the rigid cylinder) have been optimized simultaneously to obtain a higher sensitivity at the almost same working bandwidth. Firstly, through the mechanical analysis of the microstructure, the objective function and feasible region are established to optimize the parameters of the microstructure, and a set of optimized parameters is obtained. Secondly, the optimized structure is verified by ANSYS simulation, and then, the optimized fourbeam structure is fabricated by the MEMS manufacturing technology. Finally, these two kinds of hydrophones (the previous one and the optimized one) are produced, and their performance tests are carried out. The testing results show that the performances of the optimized hydrophone have been greatly improved, exhibiting a receiving sensitivity of −181.2 dB@1 kHz (increasing by 6.5 dB, 0 dB reference 1 V/μ Pa), the frequency response ranging from 20 Hz to 1 kHz which is the same working bandwidth as before, and a good dipole directivity. The optimization researches in this paper provide a method and idea for the performance improvement of the following MEMS vector hydrophone.
1. Introduction
With the continuous consumption of land resources, the huge resources in the deep ocean will become precious and provide the abundant substances that human beings need to live by including the energy, mine, and biology [1, 2]. Sonar system is the key equipment to obtain information from the ocean. And vector hydrophone as an important part of the sonar detection devices can measure vibrating acceleration of water particle and has the advantage of noise cancellation due to its directional pattern [3–6]. Faced with the increasingly serious underwater security situation, it urgently needs to adopt new technologies and methods to meet the requirements of engineering application. MEMS technology, due to the characteristic of 3 M (miniaturization, multiplicity and microelectronics), shows great vitality [7–10]. To this end, scholars expect to resolve current problems of large volume, poor consistency, and high cost through MEMS technology.
MEMS bionic vector hydrophone based on the bionic principle, piezoresistive principle, and MEMS technology, invented by the North University of China in 2007, is shown in Figure 1 [11], which has advantages of small volume, good consistency, and wonderful low frequency effect [11, 12].
However, the hydrophone has the problems of low sensitivity and narrow working bandwidth [13, 14]. In recent years, a lot of efforts have been put into the MEMS vector hydrophone to improve its sensitivity. The “Lollipopshaped” structure MEMS vector hydrophone was produced by Liu et.al [15], and a stress centralized structure MEMS vector hydrophone was presented by Zhang et al. [16]. Although these two structures, respectively, improved the sensitivity of the hydrophones, it greatly reduced their working bandwidth. The sensitivity of the hydrophone has been also improved by replacing the previous cylinder structure with five different new structures, which are, respectively, the cupshaped MEMS vector hydrophone designed by Xu et al. [17], the twocomponent cilia cylinder MEMS vector hydrophone proposed by Xu et al. [18], the cilia cluster MEMS vector hydrophone developed by Zhang et al. [19], the dumbbellshaped ciliary MEMS vector hydrophone reported by Ji et al. [20], and the hollow cilium cylinder vector hydrophone presented by Yang et al. [21]. However, they increased varying degrees of the difficulty of the hydrophone fabricating process and reduced the consistency of the hydrophone production. And, in reference [21], they did not take into account that the hollow cilium cylinder structure is prone to bubble forming, which has destructive effects on the performance of the hydrophone.
Furthermore, all of the above studies only improved the sensitivity of MEMS hydrophones by changing the cylinder structure or fourbeam microstructure alone, without optimizing and improving both. In this paper, through theoretical analysis and derivation, the maximum stress on the beam and the natural frequency of the microstructure are opposite trend changing with each parameter (length, width and height of cantilever, side length of the center block, height and radius of the rigid cylinder) of the microstructure. However, the influence weight of each parameter on the maximum stress and the natural frequency is different. Therefore, the parameters of cylinder structure and fourbeam microstructure are optimized simultaneously, aiming to further improve the sensitivity of the hydrophone and retain the same working bandwidth.
2. Working Principles
MEMS bionic vector hydrophone microstructure includes four beams, a rigid cylinder fixed in the middle of the four beams, and eight equal strain pizeoresistors in the edge of four beams (as shown in Figure 2) [22]. Figure 2(a) is the threedimensional model; Figure 2(b) is the twodimensional plan. Eight equal strain pizeoresistors constitute two Wheatstone bridges (as shown in Figure 3).
(a) The threedimensional model
(b) The twodimensional plan
When sound waves cause vibration of rigid cylinder and make four beams deformed, the stress on four beams will cause the resistance change of the pizeoresistors. Thus, we can get the sound information from the outputs ( and ) of the Wheatstone bridge.
The output voltage of the first Wheatstone bridges can be expressed as When and ,
For the Ptype strain pizeoresistors [22, 23],
Therefore, formula (3) can be approximated by: where is component of longitudinal stress; is the input voltage, and . From formula (4), we can conclude that the output () of the Wheatstone bridge is proportional to the stress on the beam.
3. Optimization Design
3.1. Microstructure Analysis
The crosssectional view of the MEMS bionic vector hydrophone microstructure is shown in Figure 4, where is the length of the beam, is the thickness of the beam, 2a is the side length of the center block, is the height of the rigid cylinder, and is the height of the rigid cylinder.
The mathematical model analysis of the hydrophone microstructure is based on the following assumptions: (a) compared with rigid cylinder, the effective mass of the fourbeamcenter block is very small, which can be ignored in the analysis process; (b) the fourbeam and the center block is an elastic structure; (c) the rigid cylinder and the frame of the fourbeamcenter block are rigid. When the rigid cylinder of the hydrophone microstructure is subjected to uniformly distributed loads in the direction. will produce two components in the center block of the microstructure, that is, the horizontal force along the direction and the moment () around the direction. Establish the mechanical analysis model of the microstructure (as shown in Figures 5–7). Figure 5 shows deformation when subjected to uniformly distributed loads . Figure 6 shows the force analysis of the center block, and Figure 7 shows the stress of a single cantilever beam.
Based on the principles of material mechanics and elastic mechanics [23], for any point on the cantilever beam, there are:
The vertical upward force at point can be expressed as:
In Figure 7, the center point of the center block is selected as the research object. Formula (7) can be written as:
By substituting formula (6) into (7), it can be obtained that
Substituting formulas (6) and (8) into (5), moment applied at any point on the beam can be expressed as:
Therefore, the stress of any point on the cantilever beam under the action of is: wherein, is the bending section modulus of the beam, and is the width of the beam.
The stress at any point of a single cantilever beam under the action of is:
Thus, the stress at any point of a single cantilever beam under the combined action of and is
It can be known from formula (12) that the maximal stress on the beam is on the both ends. That is, is maximal (as shown in formula (13)).
According to material mechanics theory, the stiffness of the microstructure on the or direction can be calculated which can be expressed as [23, 24]: where E is the Young’s modulus of silicon material.
The nature frequency of the microstructure is can be expressed as: where is the mass of the rigid cylinder. is the stiffness of the microstructure and .
From formulas (14) and (15), it can be concluded that the natural frequency of the microstructure and the maximum stress on the cantilever beam vary with the microstructural parameters (length, width and height of cantilever, side length of the center block, height and radius of the rigid cylinder) (as shown in Figures 8–13).
From Figures 8–13, it can be seen that the microstructural parameters have different weight effects on the natural frequency of the microstructure and the maximum stress on the beam. And, 2a and have almost the same weight effect on the maximum stress and the natural frequency; and have a greater weight effect on the maximum stress than the natural frequency; and have a greater weight effect on the natural frequency than the maximum stress. Therefore, MEMS vector hydrophone microstructural parameters can be designed reasonably to obtain the maximum stress, and the natural frequency hardly changes.
3.2. Optimization Design
The optimal solution is any feasible solution that makes the objective function obtain the optimal value (maximum or minimum value) in the feasible region.
The maximum stress and nature frequency , respectively, can be seen as the multivariate functions of the microstructural parameters () (as shown in formulas (16) and (17)).
The first set of values () as shown in Table 1 is commonly used as the microstructure parameters. According to the actual processing conditions and requirements, the ranges of the variable parameters are set as follows: (1) , (2) , (3) , (4) , (5) , (6) , and (7) . is regarded as the feasible zone constituted by the above seven constraint conditions, and is viewed as the objective function. Searching the maximum of in by MATLAB, the optimal solution is obtained when acquire the max (as shown in the second set of values of Table 1).

4. Simulation Analysis and Fabrication
4.1. Simulation Analysis
The finite element models of the previous microstructure and the optimized microstructure have been, respectively, established by ANSYS software. The material property parameters are shown in Table 2. The modal analysis and static analysis have been carried out. The nature frequency of the microstructures can be obtained by modal diagrams (as shown in Figure 14). And the stress distribution curves on the beams are shown in Figure 15.

(a) Previous microstructure
(b) Optimized microstructure
(a) Previous microstructure
(b) Optimized microstructure
It can be seen from Figures 14 and 15 that the nature frequency of the optimized microstructure is 1170 Hz, which is close to that (1242 Hz) of previous microstructure, and the maximum stress on the beam of the optimized microstructure is 160669 Pa, which is more than twice as much as that (72325 Pa) of previous microstructure. That is to say, the sensitivity of the optimized microstructure has been doubled, and the nature frequency almost remains unchanged. Thus, the optimization design of the microstructure is feasible.
4.2. Fabrication
The fourbeam microstructure is fabricated by standard MEMS technology which mainly includes photolithography, oxidation, etching, ion implantation, and sputtering [25, 26]. The main process is shown in Figure 16. (a) An Ntype 4inch SOI (12 μm device layer, 2 μm buried oxide layer, and 400 μm substrate layer) with crystal plane (100) and crystal orientation <100> is prepared. (b) The SOI is oxidized at 950°C to form 1000 Å SiO_{2} on both sides, and the front SiO_{2} is etched by ICP to form the piezoresistor windows. The boron ions with 100 keV energy and surface concentration are implanted, and then anneal to form the piezoresistors. (c) The SOI is reoxidized, and the front SiO_{2} is etched by ICP to form ohmic contact region. The denser boron ions with 100 keV energy and surface concentration are implanted, and then, the ohmic contacts are formed by annealing. (d) 1100 Å Si_{3}N_{4} is deposited on both sides of the SOI by PECVD. The Si_{3}N_{4} and SiO_{2} on the back side of SOI are etched by RIE, and then, etch the backside silicon using KOH and selfstop at the SiO_{2} layer. (e) 500 Å Cr and 1000 Å Au are deposited on the front side and etched by iodine water to realize the connection of the Wheatstone bridge. (f) Si_{3}N_{4} and SiO_{2} on the front side are etched by RIE, and SiO_{2} on the back side is etched by RIE, and then, the cantilever structure is released.
(a) SOI
(b) Pizeoresistor
(c) Ohmic contact
(d) Etching back substrate
(e) Metal wire
(f) Fourbeam structure
The fourbeam microstructure (MEMS chip) is fabricated as shown in Figure 17. The high aspect ratio microstructure cannot be processed through the current domestic MEMS processing lines, so it is necessary to bond the fourbeam microstructure and the rigid cylinder. Then, the two hydrophones (previous one and optimized one) have been fabricated (as shown in Figure 18).
5. Performance Test
The calibration tests for these two hydrophones without package have been, respectively, performed in the standing wave barrel with the method of comparison [27]. During testing, one of these two hydrophones (tested hydrophone) and a standard hydrophone are placed in the standing wave barrel filled with silicone oil. The standard hydrophone is the RS100 piezoelectric sound pressure hydrophone with sensitivity of 180 dB (0 dB ref 1 V/μ Pa) and bandwidth of 20 Hz~100 kHz used as a reference hydrophone for the calibration test. The sensitivity of the tested hydrophone at different frequency can be acquired by comparing the output voltage values of the two hydrophones (tested hydrophone and standard hydrophone). The schematic diagram of the calibration devices is shown in Figure 19, and the calibration system is shown in Figure 20. During the testing, data are recorded in onethirdoctave frequency points from 20 Hz to 1 kHz. When the sensitivity of or direction is calibrated, it must be made sure that the acoustic wave propagation direction of the standing wave barrel is paralleled to the corresponding maximum sensitivity direction.
The direction is as an example in this paper. The sensitivity of the tested hydrophone can be expressed as [27]: where is the voltage sensitivity of the standard hydrophone; and are the open output voltage of the tested hydrophone and the standard hydrophone, respectively; is the depth of the tested hydrophone; is the depth of the standard hydrophone; is wave number. The sensitivity testing results of these two hydrophones are shown in Figure 21.
Then, the directivity pattern of the optimized hydrophone is measured by the same equipment as shown in Figure 20. At the tested frequency, rotate the optimized hydrophone from 0° to 360° and record the hydrophone’s outputs () at each degree. The recorded data should be normalized by formula (19). where is the maximum of the outputs. The directivity pattern is obtained by plotting the normalized data () in polar coordinates. The directivity pattern at 315 Hz is depicted in Figure 22.
From Figure 21, it can be obtained that the sensitivity of the optimized hydrophone is 181.2 dB @1 kHz (0 dB reference 1 V/μ Pa), which is about 6.5 dB higher than that of the previous hydrophone, and they all range in working bandwidth from 20 Hz to 1000 Hz. Figure 22 shows that the optimized hydrophone has a good directional pattern in the form of an “8” shape and excellent symmetry property. Therefore, the optimized microstructure can improve the sensitivity of the hydrophone and maintain the same working bandwidth with the previous microstructure.
6. Conclusion
To better realize the engineering application of the MEMS vector hydrophone, this paper introduces the optimal designs of the hydrophone microstructure to further improve its sensitivity without changing working bandwidth. The mechanics analysis of the microstructure is carried out, and the objective function and feasible region are established to calculate the optimal solution of the microstructure. And the optimal designs have been proved through ANSYS simulation.
Then, the optimized microstructure has been fabricated by the standard MEMS technology. Finally, the two hydrophones (previous one and optimized one) have been fabricated and tested in standing wave barrel to verify the feasibility of the optimization. The test results show that the receiving sensitivity of the optimized hydrophone reaches up to −181.2 dB@1 kHz (0 dB reference 1 V/μ Pa), improved 6.5 dB, and whose working bandwidth is 201000 Hz without changed. Moreover, the optimized hydrophone has superior “8” shape directivity and concave point depth. Thus, the performance of the optimized hydrophone has been greatly improved compared with that of the previous hydrophone. The optimization provides new ideas to improve the performance of MEMS bionic vector hydrophone and lays the further foundation for its engineering application.
Data Availability
The data used to support the findings of this study are available from the corresponding author upon request.
Conflicts of Interest
The authors declare that there is no conflict of interest regarding the publication of this paper.
Acknowledgments
This work was funded by the National Natural Science Foundation of China (Grant nos. 51805154 and 51975191) and Scientific Research Foundation of Hubei University of Technology (GCRC2020010).
References
 S. L. Fan, Thinking of China's Sea Power and Ocean Strategy, [Ph.D. thesis], Central Party School, 2007.
 D. H. Liu, C. C. Lian, F. M. Liu, C. J. Wang, R. X. Ji, and X. X. Li, “Some views on ocean strategic layout of the Atlantic Ocean for China,” Ocean Development and Management, vol. 33, no. 5, pp. 3–7, 2016. View at: Google Scholar
 S. T. Yang, The Research of DeepWater Vector Hydrophone, [Ph.D. thesis], Harbin Institute of Technology, 2010.
 K. Kim, Investigation of an Underwater Acoustic Vector Sensor, [Ph.D. thesis], The Pennsylvania State University, 2002.
 G. P. Zhang, C. Zheng, and W. S. Lin, “Steering acoustic intensity estimator using a single acoustic vector hydrophone,” Journal of Sensors, vol. 2018, Article ID 8526092, 10 pages, 2018. View at: Publisher Site  Google Scholar
 M. Liu, L. Nie, G. Zhang, W. Zhang, and J. Zou, “Realization of a composite MEMS hydrophone without leftright ambiguity,” Sensors and Actuators A: Physical, vol. 272, pp. 231–241, 2018. View at: Publisher Site  Google Scholar
 P. Amiri, Z. Kordrostami, and K. Hassanli, “Design of a MEMS bionic vector hydrophone with piezogated MOSFET readout,” Microelectronics Journal, vol. 98, article 104748, p. 7, 2020. View at: Publisher Site  Google Scholar
 A. Hossain and A. Mian, “Fourterminal square piezoresistive sensors for MEMS pressure sensing,” Journal of Sensors, vol. 2017, Article ID 6954875, 11 pages, 2017. View at: Publisher Site  Google Scholar
 J. Kim, S. Pyo, and Y. Roh, “Analysis of a thicknessshear mode vibrator for the accelerometer in vector hydrophones,” Sensors and Actuators, A: Physical, vol. 266, pp. 9–14, 2017. View at: Publisher Site  Google Scholar
 B. A. Ganji, M. S. Nateri, and M. Dardel, “Design and modeling of a novel high sensitive MEMS piezoelectric vector hydrophone,” Microsystem Technologies, vol. 24, no. 4, pp. 2085–2095, 2018. View at: Publisher Site  Google Scholar
 C. Xue, S. Chen, W. Zhang, B. Zhang, G. Zhang, and H. Qiao, “Design, fabrication, and preliminary characterization of a novel mems bionic vector hydrophone,” Microelectronics Journal, vol. 38, no. 1011, pp. 1021–1026, 2007. View at: Publisher Site  Google Scholar
 C. Xue, Z. Tong, B. Zhang, and W. Zhang, “A novel vector hydrophone based on the piezoresistive effect of resonant tunneling diode,” IEEE Sensors Journal, vol. 8, no. 4, pp. 401402, 2008. View at: Publisher Site  Google Scholar
 M. Liu, G. Zhang, X. Song, Y. Liu, and W. Zhang, “Design of the monolithic integrated array MEMS hydrophone,” IEEE Sensors Journal, vol. 16, no. 4, pp. 989–995, 2016. View at: Publisher Site  Google Scholar
 Q. Xu, G. Zhang, Y. Zhao et al., “New insight into contradictive relationship between sensitivity and working bandwidth of cilium MEMS bionic vector hydrophone,” Journal of Micromechanics and Microengineering, vol. 29, no. 11, article 115016, 2019. View at: Publisher Site  Google Scholar
 Y. Liu, R. Wang, G. Zhang et al., ““Lollipopshaped” highsensitivity microelectromechanical systems vector hydrophone based on parylene encapsulation,” Journal of Applied Physics, vol. 118, no. 4, article 044501.1, 2015. View at: Publisher Site  Google Scholar
 G. Zhang, J. Ding, W. Xu et al., “Design and optimization of stress centralized MEMS vector hydrophone with high sensitivity at low frequency,” Mechanical Systems and Signal Processing, vol. 104, pp. 607–618, 2018. View at: Publisher Site  Google Scholar
 W. Xu, Y. Liu, G. Zhang et al., “Development of cupshaped microelectromechanical systemsbased vector hydrophone,” Journal of Applied Physics, vol. 120, no. 12, article 124502, 2016. View at: Publisher Site  Google Scholar
 Q. Xu, G. Zhang, J. Ding et al., “Design and implementation of twocomponent cilia cylinder MEMS vector hydrophone,” Sensors and Actuators, A: Physical, vol. 277, pp. 142–149, 2018. View at: Publisher Site  Google Scholar
 L. Zhang, Q. Xu, G. Zhang et al., “Design and fabrication of a multipurpose cilia cluster MEMS vector hydrophone,” Sensors and Actuators, A: Physical, vol. 296, pp. 331–339, 2019. View at: Publisher Site  Google Scholar
 S. Ji, L. Zhang, W. Zhang et al., “Design and realization of dumbbellshaped ciliary MEMS vector hydrophone,” Sensors and Actuators, A: Physical, vol. 311, p. 9, 2020. View at: Publisher Site  Google Scholar
 X. Yang, Q. Xu, G. Zhang et al., “Design and implementation of hollow cilium cylinder MEMS vector hydrophone,” Measurement, vol. 168, article 108309, 2021. View at: Publisher Site  Google Scholar
 K. V. Meena, R. Mathew, J. Leelavathi, and A. R. Sankar, “Performance comparison of a single element piezoresistor with a halfactive Wheatstone bridge for miniaturized pressure sensors,” Measurement, vol. 111, pp. 340–350, 2017. View at: Publisher Site  Google Scholar
 S. Chen, Research of MEMS Bionic Vector Hydrophone Based on Silicon, [Ph.D. thesis], The North University of China, 2008.
 Z. M. Jian, G. J. Zhang, M. R. Liu, and W. D. Zhang, “Microstructure optimization design of the MEMS bionic vector hydrophone,” Micronanoelectronic Technology, vol. 51, no. 9, pp. 576–582, 2014. View at: Google Scholar
 H. Guo, X. Li, Q. Zhu et al., “Imaging nanodefects of metal waveguides using the microwave cavity interference enhancement method,” Nanotechnology, vol. 31, no. 45, article 455203, 2020. View at: Publisher Site  Google Scholar
 M. Liu, L. Nie, S. Li et al., “Array MEMS vector hydrophone oriented at different direction angles,” Sensors, vol. 19, no. 19, article 4282, 2019. View at: Publisher Site  Google Scholar
 J. X. Fan, The Study of Vector Hydrophone Calibration Equipment, [M.S. thesis], Harbin Engineering University, 2007.
Copyright
Copyright © 2021 Zeming Jian et al. This is an open access article distributed under the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.